JYTOP SP800 LCD Infusion Pump Accurate flow rate control Unique door design Alarm

Sale price $746.00 Regular price $1,340.00

Shipping calculated at checkout.

SP800 Infusion Pump is a product combining the modern electronic technology and modern nursing, it can control the infusion rate and volume featuring in long time, high-accuracy, large range and continuous, and it has fully met the different requirements of modern clinical medical in various occasions.

1)Unique door design.
2)LCD display.
3)Enter KVO state after alarm.
4)Accurate flow rate control.
5)Adjustable flow rate.
6)Built-in rechargeable battery.
7)Alarm functions: occlusion, air bubbles, door open, completion, KVO completion, low battery, battery exhausted, power interruption, no operation, system failure, near completion, standby end, system abnormality, uninstalled IV set, etc.
8)Optional eight languages: Chinese, English, Spanish, French, Russian, Turkish, Portuguese and German.
9)The infusion rate can be changed during infusing.

Flow rate: 0.1~1500 mL/h, increment: 0.1ml/h
Bolus rate: 0.1~1500 mL/h
Volume to be injected: 0.1~9999ml; 0.1 ml~999.9 ml, increment: 0.1ml; 1000~9999ml, increment: 1ml
Accumulated volume: 0.1 ml~9999 ml
Exhaust flow rate: 600 ml/h
KVO rate: 0.5 ml/h, unadjustable
Accuracy: ±5 %
Occlusion pressure:
Low grade: 150±125mmHg(20±16.7kPa)
Middle grade: 525±125mmHg(70±16.7kPa)
High grade: 900±180mmHg(120±24.0kPa) 
Air bubble size: 1~5 grades(50, 100, 250, 500, 800 ul), sensitivity for single air bubble: 50 ul
Accumulated air bubble: 0.1~4.0ml/h
Display information: infusion rate, total volume, accumulated volume and battery power.
Classification: Class I, type CF applied part
Waterproof degree: IP34
Bed No.: 1~999
Power supply: AC100 V-240 V(50/60 Hz, 0.4-0.14A)
DC10-16 V （2.25-1.5A）
Battery: rechargeable battery, DC7.4V, 2600 mAh; it can continuously work for more than four hours with the injection rate of 25 ml/h after fully charged.
Working environment: 
Temperature: 5 ℃~40 ℃
Atmospheric pressure: 570 hPa~1060 hPa
Relative humidity: 15%~95% %, non-condensing
Storage environment:
Temperature: -20 ℃~+60 ℃ 
Atmospheric pressure: 500 hPa~1060 hPa
Relative humidity:10 %~95 %, non-condensing 
No corrosive gas 
Well ventilation

One versatile socket
One power cord
One user manual

Physical characteristic
Dimension: 146 mm × 193 mm × 95 mm(L×W×H)
Weight: about 1.5 kg