Keysight (Agilent) N9320B REPAIR and Keysight (Agilent) N9320B CALIBRATION

Custom-Cal has a high success rate in the repair of the Keysight (Agilent) N9320B. A calibration by Custom-Cal is performed by engineers with extensive OEM experience. We have the expertise and the necessary standards to perform the Keysight (Agilent) N9320B Calibration, onsite calibration may be available. We specialize in quick turnaround times and we can handle expedited deliveries upon request.

Need another RF Spectrum Analyzer?
Shop for Comparable Products on our
 used RF Spectrum Analyzer page or new RF Spectrum Analyzer page

   Keysight (Agilent) N9320B   Description / Specification:    
Keysight (Agilent) N9320B 9 kHz to 3.0 GHz RF Spectrum Analyzer

The Keysight N9320B 3 GHz RF Spectrum Analyzer is a cost-effective tool for automated test projects. It features a minimum non-zero span sweep time of 10 ms, a resolution bandwidth 10 Hz to 1 MHz and overall amplitude accuracy of ± 0.5 dB typical. The N9320B has LAN, USB, GPIB connectivity and robust measurement features.

Range: 9 kHz to 3 GHz AC coupled, 100 kHz to 3 GHz Preamp on.
Resolution: 1 Hz.
Residual FM: ≤ 100 Hz p-p in 100 ms nominal (RBW = 1 kHz, VBW = 1 kHz).
Frequency readout accuracy (start, stop, center, marker);
Marker resolution: (freq span)/(number of sweep point -1).
Uncertainty: ± (freq indication x freq reference uncertainty + 1% x span + 20% x resolution bandwidth + marker resolution).
Sweep point: 461, fixed.
Marker frequency counter;
Resolution: 1 Hz, 10 Hz, 100 Hz, 1 kHz Selectable.
Accuracy: ± [(marker freq x freq reference uncertainty ) + (counter resolution)].
Frequency span (FFT and swept mode);
Range: 0 Hz (zero span), 100 Hz to 3.0 GHz.
Resolution: 1 Hz.
Accuracy: ± span/(swept points -1).
Resolution bandwidth (RBW);
Range: (-3 dB bandwidth) 10 Hz to 1 MHz, in 1-3-10 sequence.
Accuracy: ± 5% nominal.
Resolution filter shape factor: < 5:1 nominal.

Measurement range 10 MHz to 3 GHz: Displayed average noise level (DANL) to +30 dBm.
Input attenuator range: 0 to 70 dB, in 1 dB steps.
Maximum damage level;
Average continuous power: ≤ +37 dBm (Input attenuator setting ≥ 10 dB, 3 minutes maximum).
Peak pulse power: ≤ +50 dBm (100 W) (For < 10 μs pulse width, < 1% duty cycle, and input attenuation ≥ 40 dB)
DC voltage: 50 VDC maximum.

N9320B-PA3, 3 GHz preamplifier.
N9320B-TG3, 3 GHz tracking generator.
N9320B-G01, GPIB interface.
N9320B-AMA, AM/FM demodulation metrics.
N9320B-EMF, EMI filter.
N9320B-DMA, ASK/FSK demodulation metrics.






Please contact us for your Keysight (Agilent) N9320B REPAIR and/or Keysight (Agilent) N9320B 9 kHz to 3.0 GHz RF Spectrum Analyzer CALIBRATION

I'm interested in: 
Calibration Type (Select one):

Repair Information:
 Serial Number:

 Is there any physical damage?  
 If yes, describe:

What is the description of the problem?

How/when did the problem start?

From (Email):*
Contact Name:*
Address 1:
Address 2:
State/Territory/Province: *
Zip/Postal Code: