전문가 상담

기술 지원

테스트 및 측정

모델번호로 검색: 예제: 34401A, E4440A

1-25 / 32

ENA 6 시리즈는 주파수 스윕할 때 swept sweep 모드를 지원합니까?

FAQ 2013-04-25

Is it possible on the Agilent ENA series Network Analyzers to over-ride the output source blanking at every sweep end?

FAQ 2012-10-17


FAQ 2012-10-17

4294A에서 16048A/D/E를 사용할 수 있나요?
No, the 4294A adapter set up fuction only supports 16048G/H cable to extend calibration plane./ 아니요, 4294A 아답터 setup 기능은 교정평면 확장을 위해 16048 G/H 케이블만 지원한다.

FAQ 2012-02-02


FAQ 2012-01-11

What are the implications of swapping test heads between different Agilent 4287A Impedance Analyzers?
The Agilent 4287A Impedance Analyzer must be used with the furnished test head as a set because it is an indivisible portion of the calibrated 4287A.

FAQ 2011-12-19

On the ENA / ENA-L, what occurs if I select "order-based display" in sweep mode?
Measurement points of different segments are plotted in constant intervals.

FAQ 2011-10-07


FAQ 2011-10-07

On the ENA, how can I make CW sweep (Fixed frequency) without turning off the ENA's output signal each time a sweep finishes?
Turn off the "Init Src Ctrl" feature, then hold trigger.

FAQ 2011-10-07

애질런트 ENA-시리즈 E5071C 네트워크 분석기에서 글자 크기를 바꿀 수 있는가?
예, E5071C 네트워크 분석기에서는 디스플레이 주석 글자 크기를 바꿀 수 있습니다.

FAQ 2011-10-07

When I save a Touchstone file with the ENA / ENA-L, which calculation will be reflected in the Touchstone file?
All calculations are reflected in the Touchstone file except for the mixed-mode conversion.

FAQ 2011-10-07

Does the touchstone file generated by the ENA / ENA-L include the electrical delay?
The touchstone files do not include electrical delay. Please use port extension.

FAQ 2011-10-07

For Cable Testing should I use a TDR, a network analyzer, or both?
The cable specifications will determine the test equipment requirements...

FAQ 2011-10-07

Using the VBA program, can the ENA / ENA-L control the handler interface to control other peripherals?

FAQ 2011-10-07

How can the S21 to transmission impedance (Zt) conversion equation on the 4395A Operation Manual be derived?
The equivalent circuit model of transmission measurement is as follows.

FAQ 2011-06-03

Why don't the E4991A and 4287A have Load compensation function?
Performing Load compensation in addition to Open/Short compensation is equivalent to performing Open/Short/ Load calibration at DUT contacts of test fixture (by using the working standard devices).

FAQ 2010-07-02


FAQ 2010-07-02

Why is Sweep Delay Time of the 4294A prolonged when DC bias is set to ON?
The DC bias source inside the 4294A requires a certain settling time before starting a sweep measurement.

FAQ 2010-07-02

What are primary differences between the E4991A and 4291B?
The major difference is frequency range. E4991A: 3 GHz, 4291B: 1.8 GHz.

FAQ 2010-07-02

What are the frequency limitations on the various probes utilized with the Agilent 85070E Dielectric Probe Kit and why?

FAQ 2009-05-28

벡터 네트워크 분석기와 관련하여 “스위프 속도”, “cycle time”과 “time per data point” 간의 차이점은 무엇입니까?

FAQ 2009-02-24

Why is "OVERLOAD" message sometimes displayed in Network Analyzer mode of the 4395A? How can I avoid this error message?
The OVERLOAD message on the 4395A Network Analyzer mode is often displayed when Sweep Start frequency is lower than 5 kHz and a wide IFBW is selected.

FAQ 2008-11-17

애질런트 4395A/4396B 결합 분석기를 노이즈 형식의 스펙트럼 분석기 모드에서 사용하고 있습니다. 단위는 V(전압)으로 설정되어 있는데 마커는 ‘nV / Hz’로 표시되어 있습니다. 표시된 단위는 이 모드에서 오류가 발생한 것입니까?

FAQ 2008-07-15

Our 4395A displays "Caution: CONT switching may damage Mech switch.". How should I repair?
The 4395A does not need repair.

FAQ 2008-06-27

What are the criteria to be applied when choosing the instrument for C-V measurements from the 4294A and E4980A?
It depends on the size and the thickness of the gate oxide film of the device.

FAQ 2008-05-26

1 2 다음