2018 New Pet ICU Pet Incubator MSLDW01

2018 New Pet ICU Pet Incubator MSLDW01

Availability: In stock

Model No:MSLDW01 Brand:Medsinglong Service:Medsinglong after-sales support. Contact us: Cindy@medmsl.com Keywords:puppy incubator


2018 New Pet ICU | Young Pet Incubator MSLDW01

puppy incubator

MSLDW 01 (Pet ICU/Young Pet Incubator) is a newly developed pet ICU and young pet incubator developed to serve an enormous need and fill a vacancy in the domestic pet care industry. This moderately sized piece of pet medical equipment will easily fit into most veterinarians’ practices and pet hospitals. Its characteristics and compact structural design functions perfectly server its purpose. The mechanism adopts "dual duct and double circulation convective thermal regulation" to create an excellent and bacteria-free therapeutic environment and young pet incubation environment similar to that provided pet parent (accompanied by purified air and proper temperature and humidity control).

In many countries, our Company is the first to introduce computer technology and auxiliary medical devices for pets in the area of ICU control systems. This allows for more accurate temperature control of ICU, and brings greater convenience for clinical use. The machine is equipped with an intelligent alarm device for five faults sensor, extreme temperature, excessive moisture, indoor carbon dioxide concentration, lack-of-water in the tank and power supply Interruption). This alarm system is tested and proven to be safe and reliable.

incubator for sale

Furthermore, it is equipped with a unique cabinet structure and double-layer thermostat cover, which can prevent loss of heat and radiant heat when nursing the pets. In addition, a temperature and humidity display device is provided, measuring current temperature with the set temperature clearly visible. The heating control adopts dual duct and double circulation convection heating technology. This prevents rapid change in temperature minimizing the shock that can be caused. It ensures that temperature of the ICU can rise evenly and gently. The Thermostat System adopts PID control technology, which can solve the problem of inertia temperature error comprehensively, thus the temperature control is more accurate and more energy efficient.

puppy incubator

2018 New Pet ICU | Young Pet Incubator MSLDW01 Advantages

1. Other brand only have the heater and ventilation fan on one side (right handed), we have double ventilation and heating system on both sides.

2. The temperature control tolerance in our unit is about 0.8 degrees, which could be 5 degrees in the other one.

3. We have three stages of water filtering system for the humidifier, when others only have a very simple filter inside.

4. We are the first one in the world to have quiet running built-in nebulizer.

5. We have the 254nm ultra-violet sterilization light inside.

6. We have Carbon Dioxide concentration detector, which does not exist in any other competitors'.

7. We have high concentration negative ion generating system.

8. All the components are made in plug-in way, so it's very easy to fix if there's any problem in case.

9. We have the specialized port for transfusion and oxygen.

pet brooder icu

2018 New Pet ICU | Young Pet Incubator MSLDW01 Specification

1) Power supply voltage: 220V±10%/50Hz±2%

2) Input power: ≤400VA

3) Ambient temperature: 10°C ~ 35°C

4) Temperature control range

vet hospital

Cabinet temperature: 15°C ~ 38°C (can be up to 39°C by special operations)

5) Temperature fluctuation: ≤0.8°C

6) Average temperature of monitoring chamber: ≤1.0°C

7) Cabinet temperature control accuracy: ≤±0.5°C

8) Temperature rise time: 5min ~ 20min

9) Noises in monitoring room: ≤30dB

10) Whole machine earth leakage current: ≤0.5 mA (normal state) ≤1 mA (single fault state)

11) Withstand voltage: 1500V/50Hz, lasted one minute without breakdown and flashover.

12) Ambient conditions:

animal hospital

① Transportation and storage:

a. Ambient temperature: -10°C~40°C

b. Relative humidity: ≤95%

② Operating conditions:

a. Ambient temperature: 18°C ~ 30°C

b. Relative humidity: 30% ~ 75%

puppy incubator for sale

Contact Information

Contact Information

  • *
  • *
  • *
  • *
  • *
  • *
PDF Download
test content

1. How long is the warranty?
2 years.

2. Can I choose other language?
It depends. Most machines have multiple language.

3. What certificate do you have?
Generally, most machine have CE, ISO, FSC. Some have FDA.

4. How long is the delivery?
About 4 days

5. How can I pay?
Paypal, Western Union, T/T, L/C, Trade Assurance, Money Gram etc. are accepted.

1. Can the medical incinerator burn dead body?
No, It is a non-hazardous treatment device commonly used in medical waste, such as blood cotton, needles, and disposable syringes

2. Can a medical incinerator use gasoline?
No, only diesel or natural gas liquefied gas.

What type of gas does the body cabinet compressor use?
Secop's compressors use R134a refrigerant. The refrigerant is universal and easy to buy. The refrigerant used in domestic compressors is r12, which is easy to purchase in the domestic market.

4. Does the body cabinet need to be installed before using?
No need to install. Plug it in directly. But if it is shipped by air, the refrigerant needs to be pumped out and refilled before use.

5. Is there an analysis segment for the human body analyzers MSLCA01, MSLCA02, MSLCA03, MSLCA04?
The MSLCA01 Body Analyzer does not have this function, MSLCA02, MSLCA03, MSLCA04 have the function of segmental fat, segmental moisture analysis, edema coefficient analysis, detailed body water analysis, intracellular water, extracellular water, and body water status.

6. Can the body analyzer be OEM?
It is possible when it reaches a certain amount. At present, we are all shipping from neutral machines.

7. What languages ​​does the body analyzer support?
MSLCA01 supports French, Italian, German, Portuguese, Spanish, Russian, English, Simplified Chinese, and Traditional Chinese. MSLCA02, MSLCA03, MSLCA04 support German, English, Simplified Chinese, Traditional Chinese. Lhe language needs to be confirmed in advance.

8. Can an endoscope camera connect to a monitor?
Yes, you can choose different size: 15 inches, 21 inches, 24 inches and 26 inches.

9. How often does one need to change oil in vacuum pump?

The vacuum pump oil is changed once every six months, and the pump oil is changed for 200 hours according to the frequency of use. If the catch water is overloaded, there will be a small amount of water to the vacuum pump part and then the cold trap, so the pump oil must be changed.

10. How high the degrees can the MSLFV02 refrigerator do?
Cold trap -40 degrees, plate temperature -35 degrees. This model is used for food and vegetables.

11. Can endoscopes add an extra camera?
Yes, we generally do not advise customers to purchase two cameras because the camera is the most important component of the entire device, so the price is relatively high. Of course, if customers want to buy, it is ok.

Relevant articles
test content
Inquire Now

Contact Information

Contact Information

  • *
  • *
  • *
  • *
  • *
  • *

Product Tags
