iCO Smokerlyzer Breath CO Monitor from Bedfont

The iCO™ Smokerlyzer® is the world's first Smokerlyzer® breath carbon monoxide (CO) monitor that connects to your mobile device. Taking a CO breath test with the iCO™ Smokerlyzer® is quick and easy to complete. Your personal iCO™ enables you to test your CO levels anytime, anywhere.

What is it?

The iCO™ Smokerlyzer® plugs into the headphone slot of your smartphone/tablet and is used in conjunction with the Smokerlyzer® app. Instant readings can be downloaded to a personal profile, allowing users to track their progress, and be motivated to quit smoking, anytime, anywhere. What's more, progress can be shared with smoking cessation advisors, fellow quitters, friends and family.

Why is it important?

The iCO Smokerlyzer monitor measures the amount of poisonous carbon monoxide (CO) on a person’s breath via a simple and non-invasive breath test. This shows how much someone has been smoking and evaluates their nicotine dependency.

The iCO Smokerlyzer offers a new way for smokers to monitor their habit and provide encouragement for them to quit smoking without being in direct contact with a doctor or smoking cessation advisors.

The iCO Smokerlyzer an innovative piece of technology for users to track their progress to a smoke-free life, the iCO Smokerlyzer is user friendly, fits in your pocket and provides constant support to those who are trying to quit smoking.

Benefits to the user

  • A pocket sized, personal use, CO monitor
  • Connects to app on your smartphone or tablet
  • The Smokerlyzer® app records your readings and evaluates your nicotine dependency
  • You can send your results via the app to your doctor/smoking cessation advisor/family or friend

Why the iCO™ Smokerlyzer®?

The iCO™ Smokerlyzer® offers a unique way to aid smoking cessation, take control of your smoking habit and monitor your CO levels as a visual aid and validation of your journey to a smokeless life.

Product Specification

Concentration range 0-150ppm
Detection principle Electrochemical sensor
Repeatability <±5%
Accuracy <±3ppm/15%*
T90 response time <30 seconds
Operating temperature 15-35ºC
Storage/transport temperature 0-40ºC
Operating/storage/transport pressure Atmospheric ±10%
Operating humidity 15-90% HR (non-condensing)
Storage/transport humidity 0-95%
Sensor operating life Approx. 200 tests /3 years**
Sensor sensitivity 1ppm
Sensor drift <5% per annum
Dimensions Approx. 28mm diameter x 115mm long
Weight Approx. 35g
Materials Polycarbonate/ABS blend
H2 cross interference ≤6%

*Whichever is greater

**Whichever comes first