<reponame>sidneycadot/pydwf
"""The |pydwf.core.api.digital_out| module implements a single class: |DigitalOut|."""

from typing import Tuple, List

from pydwf.core.dwf_device_subapi import AbstractDwfDeviceSubApi

from pydwf.core.auxiliary.typespec_ctypes import typespec_ctypes
from pydwf.core.auxiliary.enum_types import (DwfTriggerSource, DwfTriggerSlope, DwfDigitalOutOutput, DwfDigitalOutType,
                                             DwfState, DwfDigitalOutIdle)
from pydwf.core.auxiliary.constants import RESULT_SUCCESS
from pydwf.core.auxiliary.exceptions import PyDwfError


class DigitalOut(AbstractDwfDeviceSubApi):
    """The |DigitalOut| class provides access to the digital output (pattern generator) instrument of a
    |DwfDevice:link|.

    Attention:
        Users of |pydwf| should not create instances of this class directly.

        It is instantiated during initialization of a |DwfDevice| and subsequently assigned to its
        public |digitalOut:link| attribute for access by the user.
    """

    # pylint: disable=too-many-public-methods

    ####################################################################################################################
    #                                                                                                                  #
    #                                               INSTRUMENT CONTROL                                                 #
    #                                                                                                                  #
    ####################################################################################################################

    def reset(self) -> None:
        """Reset the |DigitalOut| instrument.

        Raises:
            DwfLibraryError: An error occurred while executing the *reset* operation.
        """
        result = self.lib.FDwfDigitalOutReset(self.hdwf)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def configure(self, start: bool) -> None:
        """Start or stop the |DigitalOut| instrument.

        Parameters:
            start (int): Whether to start/stop the instrument.

        Raises:
            DwfLibraryError: An error occurred while executing the *configure* operation.
        """
        result = self.lib.FDwfDigitalOutConfigure(self.hdwf, start)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def status(self) -> DwfState:
        """Return the status of the |DigitalOut| instrument.

        This method performs a status request to the |DigitalOut| instrument and receives its response.

        Returns:
            DwfState: The status of the |DigitalOut| instrument.

        Raises:
            DwfLibraryError: An error occurred while executing the *status* operation.
        """
        c_status = typespec_ctypes.DwfState()
        result = self.lib.FDwfDigitalOutStatus(self.hdwf, c_status)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        status_ = DwfState(c_status.value)
        return status_

    ####################################################################################################################
    #                                                                                                                  #
    #                                                    CHANNEL COUNT                                                 #
    #                                                                                                                  #
    ####################################################################################################################

    def count(self) -> int:
        """Get the |DigitalOut| instrument channel (digital pin) count.

        Returns:
            int: The number of digital output channels.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_channel_count = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutCount(self.hdwf, c_channel_count)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        channel_count = c_channel_count.value
        return channel_count

    ####################################################################################################################
    #                                                                                                                  #
    #                                                 STATE MACHINE SETTINGS                                           #
    #                                                                                                                  #
    ####################################################################################################################

    def waitInfo(self) -> Tuple[float, float]:
        """Get the |DigitalOut| instrument range for the |Wait:link| state duration, in seconds.

        Returns:
            Tuple[float, float]: A tuple containing the minimal and maximal configurable |Wait:link| state duration,
            in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_wait_duration_min = typespec_ctypes.c_double()
        c_wait_duration_max = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutWaitInfo(self.hdwf, c_wait_duration_min, c_wait_duration_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        wait_duration_min = c_wait_duration_min.value
        wait_duration_max = c_wait_duration_max.value
        return (wait_duration_min, wait_duration_max)

    def waitSet(self, wait_duration: float) -> None:
        """Set the |DigitalOut| instrument |Wait:link| state duration, in seconds.

        Parameters:
            wait_duration (float): Digital-out |Wait:link| state duration, in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutWaitSet(self.hdwf, wait_duration)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def waitGet(self) -> float:
        """Get |DigitalOut| instrument |Wait:link| state duration, in seconds.

        Returns:
            float: The |Wait:link| state duration, in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_wait_duration = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutWaitGet(self.hdwf, c_wait_duration)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        wait_duration = c_wait_duration.value
        return wait_duration

    def runInfo(self) -> Tuple[float, float]:
        """Get the |DigitalOut| instrument range for the |Running:link| state duration, in seconds.

        Returns:
            Tuple[float, float]: A tuple containing the minimal and maximal |Running:link| state duration, in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_run_duration_min = typespec_ctypes.c_double()
        c_run_duration_max = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutRunInfo(self.hdwf, c_run_duration_min, c_run_duration_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        run_duration_min = c_run_duration_min.value
        run_duration_max = c_run_duration_max.value
        return (run_duration_min, run_duration_max)

    def runSet(self, run_duration: float) -> None:
        """Set the |DigitalOut| instrument |Running:link| state duration, in seconds.

        Parameters:
            run_duration: The |Running:link| state duration, in seconds.

                The value 0 is special; it means *forever*.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutRunSet(self.hdwf, run_duration)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def runGet(self) -> float:
        """Get the |DigitalOut| instrument |Running:link| state duration, in seconds.

        Returns:
            float: The |Running:link| state duration, in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_run_duration = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutRunGet(self.hdwf, c_run_duration)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        run_duration = c_run_duration.value
        return run_duration

    def runStatus(self) -> int:
        """Get the |DigitalOut| instrument |Running:link| state duration time left,
        in clock cycles.

        This value is internally expressed as an integer with 48-bit resolution,
        and is measured in integer clock cycles. The C API returns it as a
        double-precision floating point number, to avoid using 64-bit integers.

        Use the :py:meth:`internalClockInfo` method to retrieve the clock frequency.

        Returns:
            int: The number of clock cycles until the nest state transition of the
            |DigitalOut| instrument's state machine.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_run_status = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutRunStatus(self.hdwf, c_run_status)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

        if not c_run_status.value.is_integer():
            raise PyDwfError("Bad c_run_status value.")

        run_status = int(c_run_status.value)
        return run_status

    def repeatTriggerSet(self, repeat_trigger_flag: bool) -> None:
        """Specify if each |DigitalOut| pulse sequence run should wait for its own trigger.

        Parameters:
            repeat_trigger_flag (bool): If True, not only the first, both also every successive run of the
                pulse output sequence will wait until it receives a trigger.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutRepeatTriggerSet(self.hdwf, repeat_trigger_flag)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def repeatTriggerGet(self) -> bool:
        """Get if each |DigitalOut| pulse sequence run should wait for its own trigger.

        Returns:
            bool: If True, not only the first, both also every successive run of the pulse output sequence
            will wait until it receives a trigger.
        """
        c_repeat_trigger = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutRepeatTriggerGet(self.hdwf, c_repeat_trigger)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        repeat_trigger_flag = bool(c_repeat_trigger.value)
        return repeat_trigger_flag

    def repeatInfo(self) -> Tuple[int, int]:
        """Get the |DigitalOut| minimal and maximal repeat count for pulse-sequence runs.

        Returns:
            Tuple[int, int]: A tuple containing the minimal and maximal repeat count for
            digital-out pulse-sequence runs.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_repeat_min = typespec_ctypes.c_unsigned_int()
        c_repeat_max = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutRepeatInfo(self.hdwf, c_repeat_min, c_repeat_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        repeat_min = c_repeat_min.value
        repeat_max = c_repeat_max.value
        return (repeat_min, repeat_max)

    def repeatSet(self, repeat: int) -> None:
        """Set the |DigitalOut| repeat count for pulse-sequence runs.

        Parameters:
            repeat (int): Repeat count.
                The value 0 is special; it means *forever*.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutRepeatSet(self.hdwf, repeat)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def repeatGet(self) -> int:
        """Set the |DigitalOut| repeat count for pulse-sequence runs.

        Returns:
            int: Repeat count.
            The value 0 is special; it means *forever*.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_repeat = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutRepeatGet(self.hdwf, c_repeat)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        repeat = c_repeat.value
        return repeat

    def repeatStatus(self) -> int:
        """Get the |DigitalOut| count of repeats remaining for the currently active output sequence.

        This number counts down as a digital output sequence is active.

        Returns:
            int: The repeat count status.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_repeat_status = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutRepeatStatus(self.hdwf, c_repeat_status)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        repeat_status = c_repeat_status.value
        return repeat_status

    ####################################################################################################################
    #                                                                                                                  #
    #                                                 TRIGGER CONFIGURATION                                            #
    #                                                                                                                  #
    ####################################################################################################################

    def triggerSourceInfo(self) -> List[DwfTriggerSource]:
        """Get the valid |DigitalOut| trigger sources.

        Warning:
            **This method is obsolete.**

            Use the generic DeviceControl.triggerInfo() method instead.

        Returns:
            List[DwfTriggerSource]: A list of valid the trigger sources.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_trigger_source_bitset = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutTriggerSourceInfo(self.hdwf, c_trigger_source_bitset)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        trigger_source_bitset = c_trigger_source_bitset.value
        trigger_source_list = [trigger_source for trigger_source in DwfTriggerSource
                               if trigger_source_bitset & (1 << trigger_source.value)]
        return trigger_source_list

    def triggerSourceSet(self, trigger_source: DwfTriggerSource) -> None:
        """Set the |DigitalOut| trigger source.

        Parameters:
            trigger_source (DwfTriggerSource): The trigger source to be configured.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutTriggerSourceSet(self.hdwf, trigger_source.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def triggerSourceGet(self) -> DwfTriggerSource:
        """Get the currently selected instrument trigger source.

        Returns:
            DwfTriggerSource: The currently selected instrument trigger source.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_trigger_source = typespec_ctypes.DwfTriggerSource()
        result = self.lib.FDwfDigitalOutTriggerSourceGet(self.hdwf, c_trigger_source)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        trigger_source = DwfTriggerSource(c_trigger_source.value)
        return trigger_source

    def triggerSlopeSet(self, trigger_slope: DwfTriggerSlope) -> None:
        """Select the |DigitalOut| instrument trigger slope.

        Parameters:
            trigger_slope (DwfTriggerSlope): The trigger slope to be selected.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutTriggerSlopeSet(self.hdwf, trigger_slope.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def triggerSlopeGet(self) -> DwfTriggerSlope:
        """Get the currently configured |DigitalOut| instrument trigger slope.

        Returns:
            DwfTriggerSlope: The currently selected |DigitalOut| instrument trigger slope.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_trigger_slope = typespec_ctypes.DwfTriggerSlope()
        result = self.lib.FDwfDigitalOutTriggerSlopeGet(self.hdwf, c_trigger_slope)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        trigger_slope = DwfTriggerSlope(c_trigger_slope.value)
        return trigger_slope

    ####################################################################################################################
    #                                                                                                                  #
    #                                                    OUTPUT SETTINGS                                               #
    #                                                                                                                  #
    ####################################################################################################################

    def enableSet(self, channel_index: int, enable_flag: bool) -> None:
        """Enable or disable a |DigitalOut| channel (pin).

        Parameters:
            channel_index (int): The digital pin to enable or disable.
            enable_flag (bool): Whether to enable or disable the digital output.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutEnableSet(self.hdwf, channel_index, enable_flag)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def enableGet(self, channel_index: int) -> bool:
        """Check if a specific |DigitalOut| channel (pin) is enabled for output.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            bool: Whether the digital pin is enabled as an output.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_enable_flag = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutEnableGet(self.hdwf, channel_index, c_enable_flag)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        enable_flag = bool(c_enable_flag.value)
        return enable_flag

    def outputInfo(self, channel_index: int) -> List[DwfDigitalOutOutput]:
        """Get valid |DigitalOut| output choices (e.g. Push/Pull, tristate).

        Returns:
            List[DwfDigitalOutOutput]: A list of valid output settings.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_output_bitset = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutOutputInfo(self.hdwf, channel_index, c_output_bitset)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        output_bitset = c_output_bitset.value
        output_list = [output for output in DwfDigitalOutOutput if output_bitset & (1 << output.value)]
        return output_list

    def outputSet(self, channel_index: int, output_value: DwfDigitalOutOutput) -> None:
        """Set |DigitalOut| output choice (e.g. Push/Pull, tristate).

        Parameters:
            channel_index (int): The digital pin.
            output_value (DwfDigitalOutOutput): The digital output setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutOutputSet(self.hdwf, channel_index, output_value.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def outputGet(self, channel_index: int) -> DwfDigitalOutOutput:
        """Get currently configured |DigitalOut| output (e.g. Push/Pull, tristate).

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            DwfDigitalOutOutput: The digital output setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_output_value = typespec_ctypes.DwfDigitalOutOutput()
        result = self.lib.FDwfDigitalOutOutputGet(self.hdwf, channel_index, c_output_value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        output_value = DwfDigitalOutOutput(c_output_value.value)
        return output_value

    def typeInfo(self, channel_index: int) -> List[DwfDigitalOutType]:
        """Get a list of valid |DigitalOut| output channel types.

        Returns:
            List[DwfDigitalOutType]: A list of valid digital output channel types.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_type_bitset = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutTypeInfo(self.hdwf, channel_index, c_type_bitset)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        type_bitset = c_type_bitset.value
        type_list = [type_ for type_ in DwfDigitalOutType if type_bitset & (1 << type_.value)]
        return type_list

    def typeSet(self, channel_index: int, output_type: DwfDigitalOutType) -> None:
        """Select the |DigitalOut| output channel type.

        Parameters:
            channel_index (int): The digital pin.
            output_type (DwfDigitalOutType): The digital output channel type.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutTypeSet(self.hdwf, channel_index, output_type.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def typeGet(self, channel_index: int) -> DwfDigitalOutType:
        """Get the currently selected |DigitalOut| output channel type.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            DwfDigitalOutType: The digital output channel type.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_output_type = typespec_ctypes.DwfDigitalOutType()
        result = self.lib.FDwfDigitalOutTypeGet(self.hdwf, channel_index, c_output_type)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        output_type = DwfDigitalOutType(c_output_type.value)
        return output_type

    def idleInfo(self, channel_index: int) -> List[DwfDigitalOutIdle]:
        """Get valid |DigitalOut| idle output values.

        Returns:
            List[DwfDigitalOutIdle]: A list of valid idle output values.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_idle_bitset = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutIdleInfo(self.hdwf, channel_index, c_idle_bitset)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        idle_bitset = c_idle_bitset.value
        idle_list = [idle for idle in DwfDigitalOutIdle if idle_bitset & (1 << idle.value)]
        return idle_list

    def idleSet(self, channel_index: int, idle_mode: DwfDigitalOutIdle) -> None:
        """Set the |DigitalOut| idle output value.

        Parameters:
            channel_index (int): The digital pin.
            idle_mode (DwfDigitalOutIdle): The idle output value.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutIdleSet(self.hdwf, channel_index, idle_mode.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def idleGet(self, channel_index: int) -> DwfDigitalOutIdle:
        """Get the currently configured idle output value.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            DwfDigitalOutIdle: The currently configured idle output value.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_idle_mode = typespec_ctypes.DwfDigitalOutIdle()
        result = self.lib.FDwfDigitalOutIdleGet(self.hdwf, channel_index, c_idle_mode)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        idle_mode = DwfDigitalOutIdle(c_idle_mode.value)
        return idle_mode

    ####################################################################################################################
    #                                                                                                                  #
    #                                               OUTPUT PATTERN TIMING DEFINITION                                   #
    #                                                                                                                  #
    ####################################################################################################################

    def internalClockInfo(self) -> float:
        """Get the |DigitalOut| instrument clock frequency.

        Returns:
            float: The digital-out clock frequency, in Hz.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_internal_clock_frequency = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutInternalClockInfo(self.hdwf, c_internal_clock_frequency)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        internal_clock_frequency = c_internal_clock_frequency.value
        return internal_clock_frequency

    def dividerInfo(self, channel_index: int) -> Tuple[int, int]:
        """Get the |DigitalOut| divider value range.

        Returns:
            Tuple[int, int]: The range of valid divider settings.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_divider_min = typespec_ctypes.c_unsigned_int()
        c_divider_max = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutDividerInfo(
            self.hdwf,
            channel_index,
            c_divider_min,
            c_divider_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        divider_min = c_divider_min.value
        divider_max = c_divider_max.value
        return (divider_min, divider_max)

    def dividerSet(self, channel_index: int, divider: int) -> None:
        """Set the |DigitalOut| divider value.

        Parameters:
            channel_index (int): The digital pin.
            divider (int): The divider setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutDividerSet(self.hdwf, channel_index, divider)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def dividerGet(self, channel_index: int) -> int:
        """Get the currently configured |DigitalOut| divider value.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            int: The divider setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_divider = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutDividerGet(self.hdwf, channel_index, c_divider)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        divider = c_divider.value
        return divider

    def dividerInitSet(self, channel_index: int, divider_init: int) -> None:
        """Set the |DigitalOut| initial divider value.

        Parameters:
            channel_index (int): The digital pin.
            divider_init (int): The divider init setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutDividerInitSet(self.hdwf, channel_index, divider_init)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def dividerInitGet(self, channel_index: int) -> int:
        """Get the currently configured |DigitalOut| initial divider value.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            int: The divider init setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_divider_init = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutDividerInitGet(self.hdwf, channel_index, c_divider_init)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        divider_init = c_divider_init.value
        return divider_init

    def counterInfo(self, channel_index: int) -> Tuple[int, int]:
        """Get the |DigitalOut| counter value range.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            Tuple[int, int]: The range of valid counter settings.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_counter_min = typespec_ctypes.c_unsigned_int()
        c_counter_max = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutCounterInfo(
            self.hdwf,
            channel_index,
            c_counter_min,
            c_counter_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        counter_min = c_counter_min.value
        counter_max = c_counter_max.value
        return (counter_min, counter_max)

    def counterSet(self, channel_index: int, low_count: int, high_count: int) -> None:
        """Set the |DigitalOut| counter durations for both the low and high signal output levels.

        Parameters:
            channel_index (int): The digital pin.
            low_count (int): The number of cycles the signal should be Low.
            high_count (int): The number of cycles the signal should be High.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutCounterSet(self.hdwf, channel_index, low_count, high_count)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def counterGet(self, channel_index: int) -> Tuple[int, int]:
        """Get the |DigitalOut| counter durations for both the low and high signal output levels.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            Tuple[int, int]: The number of cycles the signal should be Low, High.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_low_count = typespec_ctypes.c_unsigned_int()
        c_high_count = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutCounterGet(self.hdwf, channel_index, c_low_count, c_high_count)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        low_count = c_low_count.value
        high_count = c_high_count.value
        return (low_count, high_count)

    def counterInitSet(self, channel_index: int, high: bool, counter_init: int) -> None:
        """Set the |DigitalOut| initial signal value and initial counter value.

        Parameters:
            channel_index (int): The digital pin.
            high (bool): Whether to start High (True) or Low (False).
            counter_init (int): The initial counter setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutCounterInitSet(
            self.hdwf,
            channel_index,
            high, counter_init)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def counterInitGet(self, channel_index: int) -> Tuple[bool, int]:
        """Get the |DigitalOut| initial signal value and initial counter value.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            Tuple [bool, int]: Whether to start High (True) or Low (False),
            and the initial counter setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_high = typespec_ctypes.c_int()
        c_counter_init = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutCounterInitGet(
            self.hdwf,
            channel_index, c_high,
            c_counter_init)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        high = bool(c_high.value)
        counter_init = c_counter_init.value
        return (high, counter_init)

    ####################################################################################################################
    #                                                                                                                  #
    #                                                      DATA PLAYBACK                                               #
    #                                                                                                                  #
    ####################################################################################################################

    def dataInfo(self, channel_index: int) -> int:
        """Return the maximum buffer size for the specified |DigitalOut| channel,
        i.e., the number of custom data bits.

        Parameters:
            channel_index (int): the channel for which to obtain the data bits count.

        Returns:
            int: The number of custom data bits that can be specified for the channel.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_max_data_bits = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutDataInfo(self.hdwf, channel_index, c_max_data_bits)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        max_data_bits = c_max_data_bits.value
        return max_data_bits

    def dataSet(self, channel_index: int, bits: str, tristate: bool = False) -> None:
        """Set the |DigitalOut| arbitrary output data.

        This function also sets the counter initial, low and high value, according to the
        number of bits. The data-bits are sent out in LSB-first order.

        Parameters:
            channel_index (int): the channel for which to set the output data.
            bits (str): The bits, as a string.
            tristate (bool): Whether to interpret the string as a tristate signal.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """

        if tristate:
            bits = bits.replace('1', '11').replace('0', '01').replace('Z', '00')

        # For tristate output, the count of bits is the total number of output values (I/O)
        # and output enable (OE) bits, which should be an even number.

        count_of_bits = len(bits)

        octets = []
        while len(bits) > 0:
            octet_str = bits[:8]
            octet = int(octet_str[::-1], 2)
            octets.append(octet)
            bits = bits[8:]

        octets_as_bytes = bytes(octets)

        result = self.lib.FDwfDigitalOutDataSet(self.hdwf, channel_index, octets_as_bytes, count_of_bits)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def playDataSet(self, bits: str, bits_per_sample: int, count_of_samples: int) -> None:
        """Set the |DigitalOut| playback data.

        The output can de PushPull, OpenDrain, or OpenSource. Threestate data is not supported.

        Note:
            The DWF documentation explicitly states that this function is supported by the Digital Discovery.
            (So, by implication, it's probably not supported on anything else.)

        Parameters:
            bits (str): string of '0' and '1' characters. Its length should be (bits_per_sample * count_of_samples).
            bits_per_sample (int): Bits per sample, should be 1, 2, 4, 8, or 16.
            count_of_samples (int): Number of samples.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """

        if len(bits) != (bits_per_sample * count_of_samples):
            raise PyDwfError("Unexpected number of bits.")

        octets = []
        while len(bits) > 0:
            octet_str = bits[:8]
            octet = int(octet_str[::-1], 2)
            octets.append(octet)
            bits = bits[8:]

        octets_as_bytes = bytes(octets)

        result = self.lib.FDwfDigitalOutPlayDataSet(
            self.hdwf,
            octets_as_bytes,
            bits_per_sample,
            count_of_samples)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def playRateSet(self, playback_rate: float) -> None:
        """Set the |DigitalOut| playback rate, in Hz.

        Note:
            The DWF documentation explicitly states that this function is supported by the Digital Discovery.
            (So, by implication, it's probably not supported on anything else.)

        Parameters:
            playback_rate (float): The playback rate, in Hz.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutPlayRateSet(self.hdwf, playback_rate)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

"""The |pydwf.core.api.digital_out| module implements a single class: |DigitalOut|."""

from typing import Tuple, List

from pydwf.core.dwf_device_subapi import AbstractDwfDeviceSubApi

from pydwf.core.auxiliary.typespec_ctypes import typespec_ctypes
from pydwf.core.auxiliary.enum_types import (DwfTriggerSource, DwfTriggerSlope, DwfDigitalOutOutput, DwfDigitalOutType,
                                             DwfState, DwfDigitalOutIdle)
from pydwf.core.auxiliary.constants import RESULT_SUCCESS
from pydwf.core.auxiliary.exceptions import PyDwfError


class DigitalOut(AbstractDwfDeviceSubApi):
    """The |DigitalOut| class provides access to the digital output (pattern generator) instrument of a
    |DwfDevice:link|.

    Attention:
        Users of |pydwf| should not create instances of this class directly.

        It is instantiated during initialization of a |DwfDevice| and subsequently assigned to its
        public |digitalOut:link| attribute for access by the user.
    """

    # pylint: disable=too-many-public-methods

    ####################################################################################################################
    #                                                                                                                  #
    #                                               INSTRUMENT CONTROL                                                 #
    #                                                                                                                  #
    ####################################################################################################################

    def reset(self) -> None:
        """Reset the |DigitalOut| instrument.

        Raises:
            DwfLibraryError: An error occurred while executing the *reset* operation.
        """
        result = self.lib.FDwfDigitalOutReset(self.hdwf)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def configure(self, start: bool) -> None:
        """Start or stop the |DigitalOut| instrument.

        Parameters:
            start (int): Whether to start/stop the instrument.

        Raises:
            DwfLibraryError: An error occurred while executing the *configure* operation.
        """
        result = self.lib.FDwfDigitalOutConfigure(self.hdwf, start)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def status(self) -> DwfState:
        """Return the status of the |DigitalOut| instrument.

        This method performs a status request to the |DigitalOut| instrument and receives its response.

        Returns:
            DwfState: The status of the |DigitalOut| instrument.

        Raises:
            DwfLibraryError: An error occurred while executing the *status* operation.
        """
        c_status = typespec_ctypes.DwfState()
        result = self.lib.FDwfDigitalOutStatus(self.hdwf, c_status)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        status_ = DwfState(c_status.value)
        return status_

    ####################################################################################################################
    #                                                                                                                  #
    #                                                    CHANNEL COUNT                                                 #
    #                                                                                                                  #
    ####################################################################################################################

    def count(self) -> int:
        """Get the |DigitalOut| instrument channel (digital pin) count.

        Returns:
            int: The number of digital output channels.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_channel_count = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutCount(self.hdwf, c_channel_count)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        channel_count = c_channel_count.value
        return channel_count

    ####################################################################################################################
    #                                                                                                                  #
    #                                                 STATE MACHINE SETTINGS                                           #
    #                                                                                                                  #
    ####################################################################################################################

    def waitInfo(self) -> Tuple[float, float]:
        """Get the |DigitalOut| instrument range for the |Wait:link| state duration, in seconds.

        Returns:
            Tuple[float, float]: A tuple containing the minimal and maximal configurable |Wait:link| state duration,
            in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_wait_duration_min = typespec_ctypes.c_double()
        c_wait_duration_max = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutWaitInfo(self.hdwf, c_wait_duration_min, c_wait_duration_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        wait_duration_min = c_wait_duration_min.value
        wait_duration_max = c_wait_duration_max.value
        return (wait_duration_min, wait_duration_max)

    def waitSet(self, wait_duration: float) -> None:
        """Set the |DigitalOut| instrument |Wait:link| state duration, in seconds.

        Parameters:
            wait_duration (float): Digital-out |Wait:link| state duration, in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutWaitSet(self.hdwf, wait_duration)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def waitGet(self) -> float:
        """Get |DigitalOut| instrument |Wait:link| state duration, in seconds.

        Returns:
            float: The |Wait:link| state duration, in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_wait_duration = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutWaitGet(self.hdwf, c_wait_duration)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        wait_duration = c_wait_duration.value
        return wait_duration

    def runInfo(self) -> Tuple[float, float]:
        """Get the |DigitalOut| instrument range for the |Running:link| state duration, in seconds.

        Returns:
            Tuple[float, float]: A tuple containing the minimal and maximal |Running:link| state duration, in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_run_duration_min = typespec_ctypes.c_double()
        c_run_duration_max = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutRunInfo(self.hdwf, c_run_duration_min, c_run_duration_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        run_duration_min = c_run_duration_min.value
        run_duration_max = c_run_duration_max.value
        return (run_duration_min, run_duration_max)

    def runSet(self, run_duration: float) -> None:
        """Set the |DigitalOut| instrument |Running:link| state duration, in seconds.

        Parameters:
            run_duration: The |Running:link| state duration, in seconds.

                The value 0 is special; it means *forever*.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutRunSet(self.hdwf, run_duration)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def runGet(self) -> float:
        """Get the |DigitalOut| instrument |Running:link| state duration, in seconds.

        Returns:
            float: The |Running:link| state duration, in seconds.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_run_duration = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutRunGet(self.hdwf, c_run_duration)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        run_duration = c_run_duration.value
        return run_duration

    def runStatus(self) -> int:
        """Get the |DigitalOut| instrument |Running:link| state duration time left,
        in clock cycles.

        This value is internally expressed as an integer with 48-bit resolution,
        and is measured in integer clock cycles. The C API returns it as a
        double-precision floating point number, to avoid using 64-bit integers.

        Use the :py:meth:`internalClockInfo` method to retrieve the clock frequency.

        Returns:
            int: The number of clock cycles until the nest state transition of the
            |DigitalOut| instrument's state machine.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_run_status = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutRunStatus(self.hdwf, c_run_status)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

        if not c_run_status.value.is_integer():
            raise PyDwfError("Bad c_run_status value.")

        run_status = int(c_run_status.value)
        return run_status

    def repeatTriggerSet(self, repeat_trigger_flag: bool) -> None:
        """Specify if each |DigitalOut| pulse sequence run should wait for its own trigger.

        Parameters:
            repeat_trigger_flag (bool): If True, not only the first, both also every successive run of the
                pulse output sequence will wait until it receives a trigger.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutRepeatTriggerSet(self.hdwf, repeat_trigger_flag)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def repeatTriggerGet(self) -> bool:
        """Get if each |DigitalOut| pulse sequence run should wait for its own trigger.

        Returns:
            bool: If True, not only the first, both also every successive run of the pulse output sequence
            will wait until it receives a trigger.
        """
        c_repeat_trigger = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutRepeatTriggerGet(self.hdwf, c_repeat_trigger)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        repeat_trigger_flag = bool(c_repeat_trigger.value)
        return repeat_trigger_flag

    def repeatInfo(self) -> Tuple[int, int]:
        """Get the |DigitalOut| minimal and maximal repeat count for pulse-sequence runs.

        Returns:
            Tuple[int, int]: A tuple containing the minimal and maximal repeat count for
            digital-out pulse-sequence runs.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_repeat_min = typespec_ctypes.c_unsigned_int()
        c_repeat_max = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutRepeatInfo(self.hdwf, c_repeat_min, c_repeat_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        repeat_min = c_repeat_min.value
        repeat_max = c_repeat_max.value
        return (repeat_min, repeat_max)

    def repeatSet(self, repeat: int) -> None:
        """Set the |DigitalOut| repeat count for pulse-sequence runs.

        Parameters:
            repeat (int): Repeat count.
                The value 0 is special; it means *forever*.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutRepeatSet(self.hdwf, repeat)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def repeatGet(self) -> int:
        """Set the |DigitalOut| repeat count for pulse-sequence runs.

        Returns:
            int: Repeat count.
            The value 0 is special; it means *forever*.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_repeat = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutRepeatGet(self.hdwf, c_repeat)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        repeat = c_repeat.value
        return repeat

    def repeatStatus(self) -> int:
        """Get the |DigitalOut| count of repeats remaining for the currently active output sequence.

        This number counts down as a digital output sequence is active.

        Returns:
            int: The repeat count status.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_repeat_status = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutRepeatStatus(self.hdwf, c_repeat_status)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        repeat_status = c_repeat_status.value
        return repeat_status

    ####################################################################################################################
    #                                                                                                                  #
    #                                                 TRIGGER CONFIGURATION                                            #
    #                                                                                                                  #
    ####################################################################################################################

    def triggerSourceInfo(self) -> List[DwfTriggerSource]:
        """Get the valid |DigitalOut| trigger sources.

        Warning:
            **This method is obsolete.**

            Use the generic DeviceControl.triggerInfo() method instead.

        Returns:
            List[DwfTriggerSource]: A list of valid the trigger sources.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_trigger_source_bitset = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutTriggerSourceInfo(self.hdwf, c_trigger_source_bitset)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        trigger_source_bitset = c_trigger_source_bitset.value
        trigger_source_list = [trigger_source for trigger_source in DwfTriggerSource
                               if trigger_source_bitset & (1 << trigger_source.value)]
        return trigger_source_list

    def triggerSourceSet(self, trigger_source: DwfTriggerSource) -> None:
        """Set the |DigitalOut| trigger source.

        Parameters:
            trigger_source (DwfTriggerSource): The trigger source to be configured.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutTriggerSourceSet(self.hdwf, trigger_source.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def triggerSourceGet(self) -> DwfTriggerSource:
        """Get the currently selected instrument trigger source.

        Returns:
            DwfTriggerSource: The currently selected instrument trigger source.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_trigger_source = typespec_ctypes.DwfTriggerSource()
        result = self.lib.FDwfDigitalOutTriggerSourceGet(self.hdwf, c_trigger_source)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        trigger_source = DwfTriggerSource(c_trigger_source.value)
        return trigger_source

    def triggerSlopeSet(self, trigger_slope: DwfTriggerSlope) -> None:
        """Select the |DigitalOut| instrument trigger slope.

        Parameters:
            trigger_slope (DwfTriggerSlope): The trigger slope to be selected.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutTriggerSlopeSet(self.hdwf, trigger_slope.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def triggerSlopeGet(self) -> DwfTriggerSlope:
        """Get the currently configured |DigitalOut| instrument trigger slope.

        Returns:
            DwfTriggerSlope: The currently selected |DigitalOut| instrument trigger slope.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_trigger_slope = typespec_ctypes.DwfTriggerSlope()
        result = self.lib.FDwfDigitalOutTriggerSlopeGet(self.hdwf, c_trigger_slope)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        trigger_slope = DwfTriggerSlope(c_trigger_slope.value)
        return trigger_slope

    ####################################################################################################################
    #                                                                                                                  #
    #                                                    OUTPUT SETTINGS                                               #
    #                                                                                                                  #
    ####################################################################################################################

    def enableSet(self, channel_index: int, enable_flag: bool) -> None:
        """Enable or disable a |DigitalOut| channel (pin).

        Parameters:
            channel_index (int): The digital pin to enable or disable.
            enable_flag (bool): Whether to enable or disable the digital output.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutEnableSet(self.hdwf, channel_index, enable_flag)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def enableGet(self, channel_index: int) -> bool:
        """Check if a specific |DigitalOut| channel (pin) is enabled for output.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            bool: Whether the digital pin is enabled as an output.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_enable_flag = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutEnableGet(self.hdwf, channel_index, c_enable_flag)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        enable_flag = bool(c_enable_flag.value)
        return enable_flag

    def outputInfo(self, channel_index: int) -> List[DwfDigitalOutOutput]:
        """Get valid |DigitalOut| output choices (e.g. Push/Pull, tristate).

        Returns:
            List[DwfDigitalOutOutput]: A list of valid output settings.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_output_bitset = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutOutputInfo(self.hdwf, channel_index, c_output_bitset)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        output_bitset = c_output_bitset.value
        output_list = [output for output in DwfDigitalOutOutput if output_bitset & (1 << output.value)]
        return output_list

    def outputSet(self, channel_index: int, output_value: DwfDigitalOutOutput) -> None:
        """Set |DigitalOut| output choice (e.g. Push/Pull, tristate).

        Parameters:
            channel_index (int): The digital pin.
            output_value (DwfDigitalOutOutput): The digital output setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutOutputSet(self.hdwf, channel_index, output_value.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def outputGet(self, channel_index: int) -> DwfDigitalOutOutput:
        """Get currently configured |DigitalOut| output (e.g. Push/Pull, tristate).

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            DwfDigitalOutOutput: The digital output setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_output_value = typespec_ctypes.DwfDigitalOutOutput()
        result = self.lib.FDwfDigitalOutOutputGet(self.hdwf, channel_index, c_output_value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        output_value = DwfDigitalOutOutput(c_output_value.value)
        return output_value

    def typeInfo(self, channel_index: int) -> List[DwfDigitalOutType]:
        """Get a list of valid |DigitalOut| output channel types.

        Returns:
            List[DwfDigitalOutType]: A list of valid digital output channel types.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_type_bitset = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutTypeInfo(self.hdwf, channel_index, c_type_bitset)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        type_bitset = c_type_bitset.value
        type_list = [type_ for type_ in DwfDigitalOutType if type_bitset & (1 << type_.value)]
        return type_list

    def typeSet(self, channel_index: int, output_type: DwfDigitalOutType) -> None:
        """Select the |DigitalOut| output channel type.

        Parameters:
            channel_index (int): The digital pin.
            output_type (DwfDigitalOutType): The digital output channel type.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutTypeSet(self.hdwf, channel_index, output_type.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def typeGet(self, channel_index: int) -> DwfDigitalOutType:
        """Get the currently selected |DigitalOut| output channel type.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            DwfDigitalOutType: The digital output channel type.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_output_type = typespec_ctypes.DwfDigitalOutType()
        result = self.lib.FDwfDigitalOutTypeGet(self.hdwf, channel_index, c_output_type)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        output_type = DwfDigitalOutType(c_output_type.value)
        return output_type

    def idleInfo(self, channel_index: int) -> List[DwfDigitalOutIdle]:
        """Get valid |DigitalOut| idle output values.

        Returns:
            List[DwfDigitalOutIdle]: A list of valid idle output values.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_idle_bitset = typespec_ctypes.c_int()
        result = self.lib.FDwfDigitalOutIdleInfo(self.hdwf, channel_index, c_idle_bitset)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        idle_bitset = c_idle_bitset.value
        idle_list = [idle for idle in DwfDigitalOutIdle if idle_bitset & (1 << idle.value)]
        return idle_list

    def idleSet(self, channel_index: int, idle_mode: DwfDigitalOutIdle) -> None:
        """Set the |DigitalOut| idle output value.

        Parameters:
            channel_index (int): The digital pin.
            idle_mode (DwfDigitalOutIdle): The idle output value.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutIdleSet(self.hdwf, channel_index, idle_mode.value)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def idleGet(self, channel_index: int) -> DwfDigitalOutIdle:
        """Get the currently configured idle output value.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            DwfDigitalOutIdle: The currently configured idle output value.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_idle_mode = typespec_ctypes.DwfDigitalOutIdle()
        result = self.lib.FDwfDigitalOutIdleGet(self.hdwf, channel_index, c_idle_mode)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        idle_mode = DwfDigitalOutIdle(c_idle_mode.value)
        return idle_mode

    ####################################################################################################################
    #                                                                                                                  #
    #                                               OUTPUT PATTERN TIMING DEFINITION                                   #
    #                                                                                                                  #
    ####################################################################################################################

    def internalClockInfo(self) -> float:
        """Get the |DigitalOut| instrument clock frequency.

        Returns:
            float: The digital-out clock frequency, in Hz.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_internal_clock_frequency = typespec_ctypes.c_double()
        result = self.lib.FDwfDigitalOutInternalClockInfo(self.hdwf, c_internal_clock_frequency)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        internal_clock_frequency = c_internal_clock_frequency.value
        return internal_clock_frequency

    def dividerInfo(self, channel_index: int) -> Tuple[int, int]:
        """Get the |DigitalOut| divider value range.

        Returns:
            Tuple[int, int]: The range of valid divider settings.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_divider_min = typespec_ctypes.c_unsigned_int()
        c_divider_max = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutDividerInfo(
            self.hdwf,
            channel_index,
            c_divider_min,
            c_divider_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        divider_min = c_divider_min.value
        divider_max = c_divider_max.value
        return (divider_min, divider_max)

    def dividerSet(self, channel_index: int, divider: int) -> None:
        """Set the |DigitalOut| divider value.

        Parameters:
            channel_index (int): The digital pin.
            divider (int): The divider setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutDividerSet(self.hdwf, channel_index, divider)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def dividerGet(self, channel_index: int) -> int:
        """Get the currently configured |DigitalOut| divider value.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            int: The divider setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_divider = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutDividerGet(self.hdwf, channel_index, c_divider)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        divider = c_divider.value
        return divider

    def dividerInitSet(self, channel_index: int, divider_init: int) -> None:
        """Set the |DigitalOut| initial divider value.

        Parameters:
            channel_index (int): The digital pin.
            divider_init (int): The divider init setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutDividerInitSet(self.hdwf, channel_index, divider_init)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def dividerInitGet(self, channel_index: int) -> int:
        """Get the currently configured |DigitalOut| initial divider value.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            int: The divider init setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_divider_init = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutDividerInitGet(self.hdwf, channel_index, c_divider_init)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        divider_init = c_divider_init.value
        return divider_init

    def counterInfo(self, channel_index: int) -> Tuple[int, int]:
        """Get the |DigitalOut| counter value range.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            Tuple[int, int]: The range of valid counter settings.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_counter_min = typespec_ctypes.c_unsigned_int()
        c_counter_max = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutCounterInfo(
            self.hdwf,
            channel_index,
            c_counter_min,
            c_counter_max)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        counter_min = c_counter_min.value
        counter_max = c_counter_max.value
        return (counter_min, counter_max)

    def counterSet(self, channel_index: int, low_count: int, high_count: int) -> None:
        """Set the |DigitalOut| counter durations for both the low and high signal output levels.

        Parameters:
            channel_index (int): The digital pin.
            low_count (int): The number of cycles the signal should be Low.
            high_count (int): The number of cycles the signal should be High.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutCounterSet(self.hdwf, channel_index, low_count, high_count)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def counterGet(self, channel_index: int) -> Tuple[int, int]:
        """Get the |DigitalOut| counter durations for both the low and high signal output levels.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            Tuple[int, int]: The number of cycles the signal should be Low, High.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_low_count = typespec_ctypes.c_unsigned_int()
        c_high_count = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutCounterGet(self.hdwf, channel_index, c_low_count, c_high_count)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        low_count = c_low_count.value
        high_count = c_high_count.value
        return (low_count, high_count)

    def counterInitSet(self, channel_index: int, high: bool, counter_init: int) -> None:
        """Set the |DigitalOut| initial signal value and initial counter value.

        Parameters:
            channel_index (int): The digital pin.
            high (bool): Whether to start High (True) or Low (False).
            counter_init (int): The initial counter setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutCounterInitSet(
            self.hdwf,
            channel_index,
            high, counter_init)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def counterInitGet(self, channel_index: int) -> Tuple[bool, int]:
        """Get the |DigitalOut| initial signal value and initial counter value.

        Parameters:
            channel_index (int): The digital pin.

        Returns:
            Tuple [bool, int]: Whether to start High (True) or Low (False),
            and the initial counter setting.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_high = typespec_ctypes.c_int()
        c_counter_init = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutCounterInitGet(
            self.hdwf,
            channel_index, c_high,
            c_counter_init)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        high = bool(c_high.value)
        counter_init = c_counter_init.value
        return (high, counter_init)

    ####################################################################################################################
    #                                                                                                                  #
    #                                                      DATA PLAYBACK                                               #
    #                                                                                                                  #
    ####################################################################################################################

    def dataInfo(self, channel_index: int) -> int:
        """Return the maximum buffer size for the specified |DigitalOut| channel,
        i.e., the number of custom data bits.

        Parameters:
            channel_index (int): the channel for which to obtain the data bits count.

        Returns:
            int: The number of custom data bits that can be specified for the channel.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        c_max_data_bits = typespec_ctypes.c_unsigned_int()
        result = self.lib.FDwfDigitalOutDataInfo(self.hdwf, channel_index, c_max_data_bits)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
        max_data_bits = c_max_data_bits.value
        return max_data_bits

    def dataSet(self, channel_index: int, bits: str, tristate: bool = False) -> None:
        """Set the |DigitalOut| arbitrary output data.

        This function also sets the counter initial, low and high value, according to the
        number of bits. The data-bits are sent out in LSB-first order.

        Parameters:
            channel_index (int): the channel for which to set the output data.
            bits (str): The bits, as a string.
            tristate (bool): Whether to interpret the string as a tristate signal.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """

        if tristate:
            bits = bits.replace('1', '11').replace('0', '01').replace('Z', '00')

        # For tristate output, the count of bits is the total number of output values (I/O)
        # and output enable (OE) bits, which should be an even number.

        count_of_bits = len(bits)

        octets = []
        while len(bits) > 0:
            octet_str = bits[:8]
            octet = int(octet_str[::-1], 2)
            octets.append(octet)
            bits = bits[8:]

        octets_as_bytes = bytes(octets)

        result = self.lib.FDwfDigitalOutDataSet(self.hdwf, channel_index, octets_as_bytes, count_of_bits)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def playDataSet(self, bits: str, bits_per_sample: int, count_of_samples: int) -> None:
        """Set the |DigitalOut| playback data.

        The output can de PushPull, OpenDrain, or OpenSource. Threestate data is not supported.

        Note:
            The DWF documentation explicitly states that this function is supported by the Digital Discovery.
            (So, by implication, it's probably not supported on anything else.)

        Parameters:
            bits (str): string of '0' and '1' characters. Its length should be (bits_per_sample * count_of_samples).
            bits_per_sample (int): Bits per sample, should be 1, 2, 4, 8, or 16.
            count_of_samples (int): Number of samples.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """

        if len(bits) != (bits_per_sample * count_of_samples):
            raise PyDwfError("Unexpected number of bits.")

        octets = []
        while len(bits) > 0:
            octet_str = bits[:8]
            octet = int(octet_str[::-1], 2)
            octets.append(octet)
            bits = bits[8:]

        octets_as_bytes = bytes(octets)

        result = self.lib.FDwfDigitalOutPlayDataSet(
            self.hdwf,
            octets_as_bytes,
            bits_per_sample,
            count_of_samples)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()

    def playRateSet(self, playback_rate: float) -> None:
        """Set the |DigitalOut| playback rate, in Hz.

        Note:
            The DWF documentation explicitly states that this function is supported by the Digital Discovery.
            (So, by implication, it's probably not supported on anything else.)

        Parameters:
            playback_rate (float): The playback rate, in Hz.

        Raises:
            DwfLibraryError: An error occurred while executing the operation.
        """
        result = self.lib.FDwfDigitalOutPlayRateSet(self.hdwf, playback_rate)
        if result != RESULT_SUCCESS:
            raise self.dwf.exception()
The |pydwf.core.api.digital_out| module implements a single class: |DigitalOut|. The |DigitalOut| class provides access to the digital output (pattern generator) instrument of a     |DwfDevice:link|.      Attention:         Users of |pydwf| should not create instances of this class directly.          It is instantiated during initialization of a |DwfDevice| and subsequently assigned to its         public |digitalOut:link| attribute for access by the user. # pylint: disable=too-many-public-methods #################################################################################################################### #                                                                                                                  # #                                               INSTRUMENT CONTROL                                                 # #                                                                                                                  # #################################################################################################################### Reset the |DigitalOut| instrument.          Raises:             DwfLibraryError: An error occurred while executing the *reset* operation. Start or stop the |DigitalOut| instrument.          Parameters:             start (int): Whether to start/stop the instrument.          Raises:             DwfLibraryError: An error occurred while executing the *configure* operation. Return the status of the |DigitalOut| instrument.          This method performs a status request to the |DigitalOut| instrument and receives its response.          Returns:             DwfState: The status of the |DigitalOut| instrument.          Raises:             DwfLibraryError: An error occurred while executing the *status* operation. #################################################################################################################### #                                                                                                                  # #                                                    CHANNEL COUNT                                                 # #                                                                                                                  # #################################################################################################################### Get the |DigitalOut| instrument channel (digital pin) count.          Returns:             int: The number of digital output channels.          Raises:             DwfLibraryError: An error occurred while executing the operation. #################################################################################################################### #                                                                                                                  # #                                                 STATE MACHINE SETTINGS                                           # #                                                                                                                  # #################################################################################################################### Get the |DigitalOut| instrument range for the |Wait:link| state duration, in seconds.          Returns:             Tuple[float, float]: A tuple containing the minimal and maximal configurable |Wait:link| state duration,             in seconds.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| instrument |Wait:link| state duration, in seconds.          Parameters:             wait_duration (float): Digital-out |Wait:link| state duration, in seconds.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get |DigitalOut| instrument |Wait:link| state duration, in seconds.          Returns:             float: The |Wait:link| state duration, in seconds.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the |DigitalOut| instrument range for the |Running:link| state duration, in seconds.          Returns:             Tuple[float, float]: A tuple containing the minimal and maximal |Running:link| state duration, in seconds.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| instrument |Running:link| state duration, in seconds.          Parameters:             run_duration: The |Running:link| state duration, in seconds.                  The value 0 is special; it means *forever*.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the |DigitalOut| instrument |Running:link| state duration, in seconds.          Returns:             float: The |Running:link| state duration, in seconds.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the |DigitalOut| instrument |Running:link| state duration time left,         in clock cycles.          This value is internally expressed as an integer with 48-bit resolution,         and is measured in integer clock cycles. The C API returns it as a         double-precision floating point number, to avoid using 64-bit integers.          Use the :py:meth:`internalClockInfo` method to retrieve the clock frequency.          Returns:             int: The number of clock cycles until the nest state transition of the             |DigitalOut| instrument's state machine.          Raises:             DwfLibraryError: An error occurred while executing the operation. Specify if each |DigitalOut| pulse sequence run should wait for its own trigger.          Parameters:             repeat_trigger_flag (bool): If True, not only the first, both also every successive run of the                 pulse output sequence will wait until it receives a trigger.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get if each |DigitalOut| pulse sequence run should wait for its own trigger.          Returns:             bool: If True, not only the first, both also every successive run of the pulse output sequence             will wait until it receives a trigger. Get the |DigitalOut| minimal and maximal repeat count for pulse-sequence runs.          Returns:             Tuple[int, int]: A tuple containing the minimal and maximal repeat count for             digital-out pulse-sequence runs.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| repeat count for pulse-sequence runs.          Parameters:             repeat (int): Repeat count.                 The value 0 is special; it means *forever*.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| repeat count for pulse-sequence runs.          Returns:             int: Repeat count.             The value 0 is special; it means *forever*.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the |DigitalOut| count of repeats remaining for the currently active output sequence.          This number counts down as a digital output sequence is active.          Returns:             int: The repeat count status.          Raises:             DwfLibraryError: An error occurred while executing the operation. #################################################################################################################### #                                                                                                                  # #                                                 TRIGGER CONFIGURATION                                            # #                                                                                                                  # #################################################################################################################### Get the valid |DigitalOut| trigger sources.          Warning:             **This method is obsolete.**              Use the generic DeviceControl.triggerInfo() method instead.          Returns:             List[DwfTriggerSource]: A list of valid the trigger sources.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| trigger source.          Parameters:             trigger_source (DwfTriggerSource): The trigger source to be configured.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the currently selected instrument trigger source.          Returns:             DwfTriggerSource: The currently selected instrument trigger source.          Raises:             DwfLibraryError: An error occurred while executing the operation. Select the |DigitalOut| instrument trigger slope.          Parameters:             trigger_slope (DwfTriggerSlope): The trigger slope to be selected.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the currently configured |DigitalOut| instrument trigger slope.          Returns:             DwfTriggerSlope: The currently selected |DigitalOut| instrument trigger slope.          Raises:             DwfLibraryError: An error occurred while executing the operation. #################################################################################################################### #                                                                                                                  # #                                                    OUTPUT SETTINGS                                               # #                                                                                                                  # #################################################################################################################### Enable or disable a |DigitalOut| channel (pin).          Parameters:             channel_index (int): The digital pin to enable or disable.             enable_flag (bool): Whether to enable or disable the digital output.          Raises:             DwfLibraryError: An error occurred while executing the operation. Check if a specific |DigitalOut| channel (pin) is enabled for output.          Parameters:             channel_index (int): The digital pin.          Returns:             bool: Whether the digital pin is enabled as an output.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get valid |DigitalOut| output choices (e.g. Push/Pull, tristate).          Returns:             List[DwfDigitalOutOutput]: A list of valid output settings.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set |DigitalOut| output choice (e.g. Push/Pull, tristate).          Parameters:             channel_index (int): The digital pin.             output_value (DwfDigitalOutOutput): The digital output setting.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get currently configured |DigitalOut| output (e.g. Push/Pull, tristate).          Parameters:             channel_index (int): The digital pin.          Returns:             DwfDigitalOutOutput: The digital output setting.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get a list of valid |DigitalOut| output channel types.          Returns:             List[DwfDigitalOutType]: A list of valid digital output channel types.          Raises:             DwfLibraryError: An error occurred while executing the operation. Select the |DigitalOut| output channel type.          Parameters:             channel_index (int): The digital pin.             output_type (DwfDigitalOutType): The digital output channel type.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the currently selected |DigitalOut| output channel type.          Parameters:             channel_index (int): The digital pin.          Returns:             DwfDigitalOutType: The digital output channel type.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get valid |DigitalOut| idle output values.          Returns:             List[DwfDigitalOutIdle]: A list of valid idle output values.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| idle output value.          Parameters:             channel_index (int): The digital pin.             idle_mode (DwfDigitalOutIdle): The idle output value.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the currently configured idle output value.          Parameters:             channel_index (int): The digital pin.          Returns:             DwfDigitalOutIdle: The currently configured idle output value.          Raises:             DwfLibraryError: An error occurred while executing the operation. #################################################################################################################### #                                                                                                                  # #                                               OUTPUT PATTERN TIMING DEFINITION                                   # #                                                                                                                  # #################################################################################################################### Get the |DigitalOut| instrument clock frequency.          Returns:             float: The digital-out clock frequency, in Hz.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the |DigitalOut| divider value range.          Returns:             Tuple[int, int]: The range of valid divider settings.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| divider value.          Parameters:             channel_index (int): The digital pin.             divider (int): The divider setting.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the currently configured |DigitalOut| divider value.          Parameters:             channel_index (int): The digital pin.          Returns:             int: The divider setting.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| initial divider value.          Parameters:             channel_index (int): The digital pin.             divider_init (int): The divider init setting.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the currently configured |DigitalOut| initial divider value.          Parameters:             channel_index (int): The digital pin.          Returns:             int: The divider init setting.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the |DigitalOut| counter value range.          Parameters:             channel_index (int): The digital pin.          Returns:             Tuple[int, int]: The range of valid counter settings.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| counter durations for both the low and high signal output levels.          Parameters:             channel_index (int): The digital pin.             low_count (int): The number of cycles the signal should be Low.             high_count (int): The number of cycles the signal should be High.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the |DigitalOut| counter durations for both the low and high signal output levels.          Parameters:             channel_index (int): The digital pin.          Returns:             Tuple[int, int]: The number of cycles the signal should be Low, High.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| initial signal value and initial counter value.          Parameters:             channel_index (int): The digital pin.             high (bool): Whether to start High (True) or Low (False).             counter_init (int): The initial counter setting.          Raises:             DwfLibraryError: An error occurred while executing the operation. Get the |DigitalOut| initial signal value and initial counter value.          Parameters:             channel_index (int): The digital pin.          Returns:             Tuple [bool, int]: Whether to start High (True) or Low (False),             and the initial counter setting.          Raises:             DwfLibraryError: An error occurred while executing the operation. #################################################################################################################### #                                                                                                                  # #                                                      DATA PLAYBACK                                               # #                                                                                                                  # #################################################################################################################### Return the maximum buffer size for the specified |DigitalOut| channel,         i.e., the number of custom data bits.          Parameters:             channel_index (int): the channel for which to obtain the data bits count.          Returns:             int: The number of custom data bits that can be specified for the channel.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| arbitrary output data.          This function also sets the counter initial, low and high value, according to the         number of bits. The data-bits are sent out in LSB-first order.          Parameters:             channel_index (int): the channel for which to set the output data.             bits (str): The bits, as a string.             tristate (bool): Whether to interpret the string as a tristate signal.          Raises:             DwfLibraryError: An error occurred while executing the operation. # For tristate output, the count of bits is the total number of output values (I/O) # and output enable (OE) bits, which should be an even number. Set the |DigitalOut| playback data.          The output can de PushPull, OpenDrain, or OpenSource. Threestate data is not supported.          Note:             The DWF documentation explicitly states that this function is supported by the Digital Discovery.             (So, by implication, it's probably not supported on anything else.)          Parameters:             bits (str): string of '0' and '1' characters. Its length should be (bits_per_sample * count_of_samples).             bits_per_sample (int): Bits per sample, should be 1, 2, 4, 8, or 16.             count_of_samples (int): Number of samples.          Raises:             DwfLibraryError: An error occurred while executing the operation. Set the |DigitalOut| playback rate, in Hz.          Note:             The DWF documentation explicitly states that this function is supported by the Digital Discovery.             (So, by implication, it's probably not supported on anything else.)          Parameters:             playback_rate (float): The playback rate, in Hz.          Raises:             DwfLibraryError: An error occurred while executing the operation.