Although a wide variety of graphite tube designs have been used over the years, the dimensions nowadays are typically 20–25 mm in length and 5–6 mm inner diameter. With this technique liquid/dissolved, solid and gaseous samples may be analyzed directly. A measured volume (typically 10–50 μL) or a weighed mass (typically around 1 mg) of a solid sample are introduced into the graphite tube and subject to a temperature program. This typically consists of stages, such as drying – the solvent is evaporated; pyrolysis – the majority of the matrix constituents are removed; atomization – the analyte element is released to the gaseous phase; and cleaning – eventual residues in the graphite tube are removed at high temperature.