Hollow cathode lamps (HCL) are the most common radiation source in LS AAS. Inside the sealed lamp, filled with argon or neon gas at low pressure, is a cylindrical metal cathode containing the element of interest and an anode. A high voltage is applied across the anode and cathode, resulting in an ionization of the fill gas. The gas ions are accelerated towards the cathode and, upon impact on the cathode, sputter cathode material that is excited in the glow discharge to emit the radiation of the sputtered material, i.e., the element of interest. In the majority of cases single element lamps are used, where the cathode is pressed out of predominantly compounds of the target element. Multi-element lamps are available with combinations of compounds of the target elements pressed in the cathode. Multi element lamps produce slightly less sensitivity than single element lamps and the combinations of elements have to be selected carefully to avoid spectral interferences. Most multi-element lamps combine a handful of elements, e.g.: 2 - 8. Atomic Absorption Spectrometers can feature as few as 1-2 hollow cathode lamp positions or in automated multi-element spectrometers, a 8-12 lamp positions may be typically available.