In ET AAS a transient signal is generated, the area of which is directly proportional to the mass of analyte (not its concentration) introduced into the graphite tube. This technique has the advantage that any kind of sample, solid, liquid or gaseous, can be analyzed directly. Its sensitivity is 2–3 orders of magnitude higher than that of flame AAS, so that determinations in the low μg L−1 range (for a typical sample volume of 20 μL) and ng g−1 range (for a typical sample mass of 1 mg) can be carried out. It shows a very high degree of freedom from interferences, so that ET AAS might be considered the most robust technique available nowadays for the determination of trace elements in complex matrices.