Details of the fabrication and resulting microstructure of the glass membrane of the pH electrode are maintained as trade secrets by the manufacturers. However, certain aspects of design are published. Glass is a solid electrolyte, for which alkali-metal ions can carry current. The pH-sensitive glass membrane is generally spherical to simplify the manufacture of a uniform membrane. These membranes are up to 0.4 millimeters in thickness, thicker than original designs, so as to render the probes durable. The glass has silicate chemical functionality on its surface, which provides binding sites for alkali-metal ions and hydrogen ions from the solutions. This provides an ion-exchange capacity in the range of 10−6 to 10−8 mol/cm2. Selectivity for hydrogen ions (H+) arises from a balance of ionic charge, volume requirements versus other ions, and the coordination number of other ions. Electrode manufacturers have developed compositions that suitably balance these factors, most notably lithium glass.