Bourdon tubes measure gauge pressure, relative to ambient atmospheric pressure, as opposed to absolute pressure; vacuum is sensed as a reverse motion. Some aneroid barometers use Bourdon tubes closed at both ends (but most use diaphragms or capsules, see below). When the measured pressure is rapidly pulsing, such as when the gauge is near a reciprocating pump, an orifice restriction in the connecting pipe is frequently used to avoid unnecessary wear on the gears and provide an average reading; when the whole gauge is subject to mechanical vibration, the entire case including the pointer and indicator card can be filled with an oil or glycerin. Tapping on the face of the gauge is not recommended as it will tend to falsify actual readings initially presented by the gauge. The Bourdon tube is separate from the face of the gauge and thus has no effect on the actual reading of pressure. Typical high-quality modern gauges provide an accuracy of ±2% of span, and a special high-precision gauge can be as accurate as 0.1% of full scale.