The first step of analysis is to filter the raw data to remove very high frequency data (called "micro-roughness") since it can often be attributed to vibrations or debris on the surface. Filtering out the micro-roughness at a given cut-off threshold also allows to bring closer the roughness assessment made using profilometers having different stylus ball radius e.g. 2 µm and 5 µm radii. Next, the data is separated into roughness, waviness and form. This can be accomplished using reference lines, envelope methods, digital filters, fractals or other techniques. Finally, the data is summarized using one or more roughness parameters, or a graph. In the past, surface finish was usually analyzed by hand. The roughness trace would be plotted on graph paper, and an experienced machinist decided what data to ignore and where to place the mean line. Today, the measured data is stored on a computer, and analyzed using methods from signal analysis and statistics.