Electrospray ionization has also been achieved at pressures as low as 25 torr and termed subambient pressure ionization with nanoelectrospray (SPIN) based upon a two-stage ion funnel interface developed by Richard D. Smith and coworkers. The SPIN implementation provided increased sensitivity due to the use of ion funnels that helped confine and transfer ions to the lower pressure region of the mass spectrometer. Nanoelectrospray emitter is made out of a fine capillary with a small aperture about 1–3 micrometer. For sufficient conductivity this capillary is usually sputter-coated with conductive material, e.g. gold. Nanoelectrospray ionization consumes only a few microliters of a sample and forms smaller droplets. Operation at low pressure was particularly effective for low flow rates where the smaller electrospray droplet size allowed effective desolvation and ion formation to be achieved. As a result, the researchers were later able to demonstrate achieving an excess of 50% overall ionization utilization efficiency for transfer of ions from the liquid phase, into the gas phase as ions, and through the dual ion funnel interface to the mass spectrometer.