Once the treatment planner has made an initial plan, the treatment planning system calculates the required monitor units to deliver a prescribed dose to a specific area, and the distribution of dose in the body this will create. The dose distribution in the patient is dependent on the anatomy and beam modifiers such as wedges, specialized collimation, field sizes, tumor depth, etc. The information from a prior CT scan of the patient allows more accurate modelling of the behaviour of the radiation as it travels through the patient's tissues. Different dose calculation models are available, including pencil beam, convolution-superposition and monte carlo simulation, with precision versus computation time being the relevant trade-off.