The principle of silver/silver chloride sensors operation is the conversion of ion current at the surface of human tissues to electron current to be delivered through the lead wire to the instrument to read. An important part of the operation is electrolyte gel which is applied between the electrode and tissues. The gel contains free chloride ions such that the charge can be carried through the electrolyte, therefore the electrolyte can be considered as conductive for ion current as the human tissues. When the ion current exists, the silver atoms in the electrode oxidize and discharge cations to the electrolyte and the electrons carry charge through the lead wire. At the same time, the chloride ions which are anions in the electrolyte travel toward the electrode and they are reduced as they bond with silver of the electrode resulting in silver chloride and free electrons to deliver to the lead wire. The reaction allows current to pass from electrolyte to electrode and the electron current passes through the lead wire for the instrument to read.