Automated chilled mirror devices provide significantly more repeatable results, but these measurements can be affected by contaminants that may compromise the mirror's surface. In many instances it is important to incorporate an effective filtration system that prepares the gas for analysis. On the other hand, filtration may alter the gas composition slightly and filter elements are subject to clogging and saturation. Advances in technology have led to analyzers that are less affected by contaminants and certain devices can also measure the dew point of water that may be present in the gas. One recent innovation is the use of spectroscopy to determine the nature of the condensate at dewpoint. Another device user laser interferometry to register extremely tenuous amounts of condensation. It is asserted that these technologies are less affected by interference from contaminants. Another source of error is the speed of the cooling of the mirror and the measurement of the temperature of the mirror when the condensation is detected. This error can be minimized by controlling the cooling speed, or having a fast condensation detection system.