A typical filar micrometer consists of a reticle that has two fine parallel wires or threads that can be moved by the observer using a micrometer screw mechanism. The wires are placed in the focal image plane of the eyepiece so they remain sharply superimposed over the object under observation, while the micrometer motion moves the wires across the focal plane. Other designs employ a fixed reticle, against which one wire or a second reticle moves. By rotating the eyepiece assembly in the eyetube, the measurement axis can be aligned to match the orientation of the two points of observation.