To flush the machine, first remove or disable the vaporizers and then flush the machine with 10 L/min or greater fresh gas flow rate for at least 20 minutes. While flushing the machine the ventilator should be set to periodically ventilate a new breathing circuit. The soda lime should also be replaced. After machine preparation, anesthesia should be induced and maintained with non-triggering agents. The time required to flush a machine varies for different machines and volatile anesthetics. This prevention technique was optimized to prepare older generation anesthesia machines. Modern anesthetic machines have more rubber and plastic components which provide a reservoir for volatile anesthetics, and should be flushed for 60 minutes.