It is important to realise that the 468 specification covers both weighted and 'unweighted' (using a 22 Hz to 22 kHz 18 dB/octave bandpass filter) measurement and that both use a very special quasi-peak rectifier with carefully devised dynamics (A-weighting uses RMS detection for no particular reason). Rather than having a simple 'integration time' this detector requires implementation with two cascaded 'peak followers' each with different attack time-constants carefully chosen to control the response to both single and repeating tone-bursts of various durations. This ensures that measurements on impulsive noise take proper account of our reduced hearing sensitivity to short bursts. This quasi-peak measurement is also called psophometric weighting.