The high-voltage rf spark source is the one that was used in commercial SSMS instruments due to its ability to ionize both conducting and non-conducting materials. Typically, samples are physically incorporated into two conductive electrodes between which an intermittent (1 MHz) high-voltage (50-100 kV using a Tesla transformer) electric spark is produced, ionizing the material at the tips of the pin-shaped electrodes. When the pulsed current is applied to the electrodes under ultra-high vacuum, a spark discharge plasma occurs in the spark gap in which ions are generated via electron impact. Within the discharge plasma, the sample evaporates, atomizes, and ionizes via electron impact. The total ion current may be optimized by adjusting the distance between the electrodes. This mode of ionization can be used to ionize conducting, semi-conducting, and non-conducting samples.