direct ec european parliament council march measur instrument text eea relev european parliament council european union regard treati establish european commun articl thereof regard propos commiss regard opinion european econom social committe act accord procedur laid articl treati number measur instrument cover specif direct adopt basi council direct eec juli approxim law member state relat common provis measur instrument method metrolog control specif direct technic outdat repeal replac independ direct reflect spirit council resolut approach technic harmonis standard correct traceabl measur instrument varieti measur task respond reason public interest public health safeti order protect environ consum levi tax duti fair trade directli indirectli affect daili life citizen wai requir legal control measur instrument legal metrolog control lead barrier free movement measur instrument provis concern member state proof conform accept commun legal metrolog control requir conform perform requir perform requir measur instrument meet provid high level protect conform assess provid high level confid member state gener rule prescrib legal metrolog control legal metrolog control prescrib measur instrument compli common perform requir principl option introduc direct member state exercis decid regul instrument cover direct applic extent unfair competit respons manufactur complianc requir direct specif state perform measur instrument sensit environ electromagnet environ immun measur instrument electromagnet interfer form integr part direct immun requir council direct eec approxim law member state relat electromagnet compat appli commun legisl essenti requir imped technic progress prefer perform requir provis remov technic barrier trade follow council resolut approach technic harmonis standard order account differ climat condit level consum protect appli nation level essenti requir give rise establish environment accuraci class order eas task prove conform essenti requir enabl conform assess desir harmonis standard harmonis standard drawn privat law bodi retain statu mandatori text end european committe standardis cen european committe electrotechn standardis cenelec european telecommun standard institut etsi recognis compet bodi adopt harmonis standard accord gener guidelin cooper commiss european standardis bodi sign novemb technic perform specif internation agre norm document compli part full essenti requir laid direct case internation agre norm document altern harmonis standard specif condit give rise presumpt conform conform essenti requir laid direct provid specif suppli european technic standard internation agre norm document european technic standard internation agre norm document option conform assess assembl respect provis direct assembl trade separ independ instrument exercis conform assess undertaken independ instrument concern state art measur technolog subject constant evolut lead conform assess categori measur assembl procedur choic procedur equival stringenc procedur adopt requir council decis eec juli modul phase conform assess procedur rule affix quot ce quot mark intend technic harmonis direct derog made modul order reflect specif aspect metrolog control provis made quot ce quot mark affix fabric process continu develop measur technolog concern express stakehold certif stress ensur consist conform assess procedur industri product request council resolut adopt novemb member state imped place market put measur instrument carri quot ce quot mark supplementari metrolog mark accord provis direct member state action prevent compli measur instrument market put adequ cooper compet author member state ensur commun wide effect object manufactur inform ground neg decis respect product legal remedi manufactur offer possibl exercis right obtain entri forc direct reason transit period nation specif nation requir interfer provis direct quot put quot measur implement direct adopt accord council decis ec june lai procedur exercis implement power confer commiss activ measur instrument commmitte includ proper consult repres interest parti direct eec eec eec eec eec concern meter defin annex mi direct eec eec eec eec eec eec repeal adopt direct articl scope direct appli devic system measur function defin instrument specif annex water meter mi ga meter volum convers devic mi activ electr energi meter mi heat meter mi measur system continu dynam measur quantiti liquid water mi automat weigh instrument mi taximet mi materi measur mi dimension measur instrument mi exhaust ga analys mi articl member state prescrib measur instrument mention articl measur task reason public interest public health public safeti public order protect environ protect consum levi tax duti fair trade justifi member state prescrib commun reason therefor commiss member state articl object direct establish requir devic system refer articl satisfi view market put task mention articl direct specif direct respect requir electromagnet immun sens articl direct eec direct eec continu appli regard emiss requir articl definit purpos direct quot measur instrument quot mean devic system measur function cover articl quot assembl quot mean hardwar devic mention specif annex function independ make measur instrument assembl compat measur instrument compat quot legal metrolog control quot mean control measur task intend field applic measur instrument reason public interest public health public safeti public order protect environ levi tax duti protect consum fair trade quot manufactur quot mean natur legal person respons conform measur instrument direct view place market put purpos quot place market quot mean make time commun instrument intend end user reward free charg quot put quot mean instrument intend end user purpos intend quot authoris repres quot mean natur legal person establish commun authoris manufactur write act behalf task mean direct quot harmonis standard quot mean technic specif adopt cen cenelec etsi jointli organis request commiss pursuant direct ec european parliament council june lai procedur provis inform field technic standard regul rule inform societi servic prepar accord gener guidelin agre commiss european standard organis quot norm document quot mean document technic specif adopt organis international de métrologi légale oiml subject procedur stipul articl articl applic assembl specif annex exist lai essenti requir assembl provis direct appli mutati mutandi assembl assembl measur instrument assess independ separ purpos establish conform articl essenti requir assess conform measur instrument meet essenti requir laid annex relev instrument specif annex member state requir need correct instrument inform refer annex relev instrument specif annex provid offici languag member state instrument market conform measur instrument essenti requir assess accord articl articl conform mark conform measur instrument provis direct presenc quot ce quot mark supplementari metrolog mark articl quot ce quot mark supplementari metrolog mark affix respons manufactur mark affix instrument fabric process justifi affix mark measur instrument deceiv parti mean form quot ce quot mark supplementari metrolog mark prohibit mark affix measur instrument provid visibl legibl quot ce quot mark supplementari metrolog mark reduc measur instrument subject measur adopt direct cover aspect requir affix quot ce quot mark mark instrument question presum conform requir direct case public refer direct offici journal european union document notic instruct requir direct accompani measur instrument articl place market put member state imped reason cover direct place market put measur instrument carri quot ce quot mark supplementari metrolog mark accord articl member state measur ensur measur instrument market put satisfi requir direct member state requir measur instrument satisfi provis govern put justifi local climat condit case member state choos upper lower temperatur limit tabl annex addit humid condit condens condens intend locat open close accuraci class defin measur instrument instrument specif annex head quot put quot accuraci class specif applic case member state determin accuraci class specif applic class defin subject allow accuraci class territori case fall measur instrument accuraci class owner choos trade fair exhibit demonstr member state prevent show instrument conform direct provid visibl sign conform avail place market put brought conform articl conform assess conform assess measur instrument relev essenti requir carri applic choic manufactur conform assess procedur list instrument specif annex manufactur provid technic document specif instrument group instrument set articl conform assess modul make procedur annex record correspond relat conform assess drawn offici languag member state notifi bodi carri conform assess procedur establish languag accept bodi articl technic document technic document render design manufactur oper measur instrument intellig permit assess conform requir direct technic document suffici detail ensur definit metrolog characterist reproduc metrolog perform produc instrument properli adjust intend mean integr instrument technic document includ relev assess identif type instrument gener descript instrument conceptu design manufactur draw plan compon assembl circuit manufactur procedur ensur consist product applic descript electron devic draw diagram flow diagram logic gener softwar inform explain characterist oper descript explan understand paragraph includ oper instrument list standard norm document refer articl appli full part descript solut adopt meet essenti requir standard norm document refer articl appli result design calcul examin test result demonstr type instrument compli requir direct declar rate oper condit environment disturb durabl specif ga water heat meter liquid water ec type examin certif ec design examin certif respect instrument part ident design manufactur seal mark appli manufactur condit compat interfac assembl relev articl notif member state notifi member state commiss bodi jurisdict design carri task pertain conform assess modul refer articl identif number commiss accord paragraph articl kind measur instrument bodi design addit relev instrument accuraci class measur rang measur technolog instrument characterist limit scope notif member state appli criteria set articl design bodi bodi meet criteria laid nation standard transpos relev harmonis standard refer publish offici journal european union presum meet criteria member state publish refer nation standard member state introduc nation legisl task mention articl retain design notifi bodi task relat instrument member state notifi bodi ensur bodi continu meet criteria set articl withdraw notif find bodi longer meet criteria forthwith inform member state commiss withdraw bodi notifi identif number commiss commiss publish list notifi bodi inform respect scope notif refer paragraph offici journal european union seri ensur list date articl criteria satisfi design bodi member state appli criteria design bodi accord articl bodi director staff involv conform assess task design manufactur supplier instal user measur instrument inspect authoris repres addit directli involv design manufactur market mainten instrument repres parti engag activ preced criterion preclud possibl exchang technic inform manufactur bodi purpos conform assess bodi director staff involv conform assess task free pressur induc financi induc influenc judgement result conform assess person group person interest result assess conform assess carri highest degre profession integr requisit compet field metrolog bodi subcontract specif task ensur subcontractor meet requir direct articl bodi relev document assess subcontractor qualif work carri direct dispos notifi author bodi capabl carri conform assess task design task carri bodi behalf respons dispos staff access facil carri proper manner technic administr task entail conform assess bodi staff sound technic vocat train cover conform assess task bodi design satisfactori knowledg rule govern task carri adequ experi task requisit abil draw certif record report demonstr task carri imparti bodi director staff guarante remuner bodi depend result task carri remuner bodi director staff depend number task carri result task bodi civil liabil insur civil liabil cover member state concern nation law bodi director staff bound observ profession secreci regard inform obtain perform duti pursuant direct vi vi author member state design articl harmonis standard norm document member state presum conform essenti requir refer annex relev instrument specif annex respect measur instrument compli element nation standard implement european harmonis standard measur instrument correspond element european harmonis standard refer respect publish offici journal european union seri measur instrument compli part element nation standard refer subparagraph member state presum conform essenti requir element nation standard instrument compli member state publish refer nation standard refer subparagraph member state presum conform essenti requir refer annex relev instrument specif annex respect measur instrument compli part norm document list refer articl refer respect publish offici journal european union seri measur instrument compli part norm document refer subparagraph member state presum conform essenti requir norm element instrument compli member state publish refer norm document refer subparagraph manufactur choos technic solut compli essenti requir refer annex relev instrument specif annex mi mi addit benefit presumpt conform manufactur correctli appli solut mention relev european harmonis standard part norm document list refer paragraph member state presum complianc test mention point articl test programm perform accord relev document mention paragraph test result ensur complianc essenti requir articl stand committe member state commiss consid european harmonis standard refer articl fulli meet essenti requir refer annex relev instrument specif annex member state commiss bring matter stand committe set articl direct ec give reason committe deliv opinion delai light committe opinion commiss inform member state withdraw refer nation standard public refer subparagraph articl articl measur instrument committe commiss assist measur instrument committe refer made paragraph articl decis ec appli regard provis articl thereof refer made paragraph articl decis ec appli regard provis articl thereof period laid articl decis ec set month committe adopt rule procedur commiss ensur relev inform envisag measur refer articl made interest parti due time articl function measur instrument committe request member state initi commiss act accord procedur refer articl measur identifi norm document drawn oiml list part thereof complianc rise presumpt conform essenti requir direct publish refer norm document list refer point offici journal european union seri request member state initi commiss act accord procedur refer articl measur amend instrument specif annex mi mi respect maximum permiss error mpe accuraci class rate oper condit critic chang valu disturb member state commiss consid norm document refer publish offici journal european union seri accord paragraph fulli meet essenti requir refer annex relev instrument specif annex member state commiss bring matter measur instrument committe give reason commiss act accord procedur refer articl inform member state withdraw refer norm document concern public offici journal member state measur consult interest parti nation level oiml work relat scope direct articl mark quot ce quot mark refer articl consist symbol quot ce quot design laid paragraph annex decis eec quot ce quot mark mm high supplementari metrolog mark consist capit letter quot quot digit year affix surround rectangl height rectangl equal height quot ce quot mark supplementari metrolog mark immedi follow quot ce quot mark identif number notifi bodi concern refer articl prescrib conform assess procedur follow quot ce quot mark supplementari metrolog mark measur instrument consist set devic assembl oper mark affix instrument main devic measur instrument small sensit carri quot ce quot mark supplementari metrolog mark mark carri packag accompani document requir direct quot ce quot mark supplementari metrolog mark indel identif number notifi bodi concern indel destruct remov mark visibl easili access articl market surveil administr cooper member state measur ensur measur instrument subject legal metrolog control compli applic provis direct market put compet author member state assist fulfil oblig carri market surveil compet author exchang inform extent instrument examin compli provis direct result examin ec type examin design examin certif annex issu notifi bodi addit amend withdraw relat certif issu qualiti system approv issu notifi bodi inform qualiti system refus withdrawn evalu report establish notifi bodi demand author member state ensur inform relat certif qualiti system approv made bodi notifi member state inform member state commiss compet author design exchang inform articl safeguard claus member state establish part measur instrument model bear quot ce quot mark supplementari metrolog mark satisfi essenti requir relat metrolog perform set direct correctli instal accord manufactur instruct measur withdraw instrument market prohibit restrict market prohibit restrict decid measur member state account systemat incident natur complianc member state establish complianc systemat natur immedi inform commiss measur indic reason decis commiss enter consult parti concern commiss find measur member state concern justifi immedi inform member state thereof member state compet member state action person affix mark inform commiss member state thereof complianc attribut shortcom standard norm document commiss consult parti concern bring matter committe refer articl commiss find measur member state concern justifi immedi inform member state thereof manufactur concern authoris repres commiss ensur member state inform progress outcom procedur articl unduli fix mark member state establish quot ce quot mark supplementari metrolog mark affix unduli manufactur authoris repres oblig make instrument conform provis quot ce quot mark supplementari metrolog mark cover articl end infring condit impos member state infring persist member state measur restrict prohibit place market instrument question ensur withdrawn market prohibit restrict accord procedur laid articl articl decis entail refus restrict decis pursuant direct entail withdraw market measur instrument prohibit restrict place market put instrument state exact ground base decis notifi forthwith parti concern time inform legal remedi law member state concern time limit remedi subject articl repeal direct repeal octob prejudic articl council direct eec juli approxim law member state relat ga meter direct eec juli approxim law member state relat meter liquid water direct eec octob approxim law member state relat ancillari equip meter liquid water direct eec novemb approxim law member state relat materi measur length direct eec decemb approxim law member state relat cold water meter concern meter defin annex mi direct direct eec june approxim law member state relat continu totalis weigh machin direct eec novemb approxim law member state relat electr energi meter direct eec decemb approxim law member state relat taximet direct eec april approxim law member state relat measur system liquid water direct eec decemb approxim law member state relat automat checkweigh weight grade machin direct eec septemb approxim law member state relat hot water meter articl transit provis derog articl member state permit measur task prescrib legal control measur instrument place market put measur instrument satisfi rule applic octob expiri valid type approv measur instrument case type approv indefinit valid period maximum ten year octob articl transposit member state adopt publish law regul administr provis compli direct april forthwith inform commiss thereof member state adopt measur refer direct accompani refer occas offici public method make refer laid member state member state appli provis octob member state commun commiss text provis nation law adopt field cover direct articl revis claus european parliament council invit commiss report april implement direct inter alia basi report provid member state submit propos amend european parliament council invit commiss evalu conform assess procedur industri product properli appli propos amend order ensur consist certif articl entri forc direct enter forc dai public offici journal european union articl addresse direct address member state strasbourg march european parliament presid cox council presid roch oj oj oj opinion european parliament juli oj council common posit juli oj posit european parliament decemb publish offici journal decis council februari oj direct amend regul ec oj oj oj direct amend direct eec oj oj oj oj oj direct amend direct ec oj oj direct amend commiss direct eec oj oj oj direct amend act access oj direct amend commiss direct eec oj oj oj oj oj oj direct amend commiss direct eec oj oj oj annex essenti requir measur instrument provid high level metrolog protect order parti affect confid result measur design manufactur high level qualiti respect measur technolog secur measur data requir met measur instrument set supplement specif instrument requir annex mi mi provid detail aspect gener requir solut adopt pursuit requir account intend instrument forese misus thereof definit measurand measurand quantiti subject measur influenc quantiti influenc quantiti quantiti measurand affect result measur rate oper condit rate oper condit valu measurand influenc quantiti make normal work condit instrument disturb influenc quantiti limit requir rate oper condit measur instrument influenc quantiti disturb influenc quantiti rate oper condit critic chang critic chang chang measur result consid undesir materi measur materi measur devic intend reproduc suppli perman manner valu quantiti direct sale trade transact direct sale measur result serv basi price pai parti involv transact relat measur consum parti requir similar level protect parti transact accept measur result time place climat environ climat environ condit measur instrument cope climat differ member state rang temperatur limit defin util util regard supplier electr ga heat water requir allow error rate oper condit absenc disturb error measur exce maximum permiss error mpe laid instrument specif requir state instrument specif annex mpe express bilater deviat true measur rate oper condit presenc disturb perform requir laid instrument specif requir instrument intend perman continu electromagnet field permit perform radiat electromagnet field amplitud modul test mpe manufactur climat mechan electromagnet environ instrument intend power suppli influenc quantiti affect accuraci take account requir laid instrument specif annex climat environ manufactur upper temperatur limit lower temperatur limit valu tabl annex mi mi instrument design condens condens humid intend locat instrument open close tabl gt tabl gt mechan environ classifi class class appli instrument locat vibrat shock low signific instrument fasten light support structur subject neglig vibrat shock transmit local blast pile drive activ slam door class appli instrument locat signific high level vibrat shock transmit machin pass vehicl vicin adjac heavi machin conveyor belt class appli instrument locat level vibrat shock high high instrument mount directli machin conveyor belt influenc quantiti consid relat mechan environ vibrat mechan shock electromagnet environ classifi class laid instrument specif annex class appli instrument locat electromagnet disturb found residenti commerci light industri build class appli instrument locat electromagnet disturb found industri build class appli instrument suppli batteri vehicl instrument compli requir addit requir voltag reduct caus energis starter motor circuit intern combust engin load dump transient occur event discharg batteri disconnect engin run influenc quantiti consid relat electromagnet environ voltag interrupt short voltag reduct voltag transient suppli line signal line electrostat discharg radio frequenc electromagnet field conduct radio frequenc electromagnet field suppli line signal line surg suppli line signal line influenc quantiti consid voltag variat main frequenc variat power frequenc magnet field quantiti influenc signific accuraci instrument carri test envisag direct paragraph appli basic rule test determin error essenti requir verifi relev influenc quantiti instrument specif annex essenti requir appli influenc quantiti appli effect evalu separ influenc quantiti constant refer metrolog test carri applic influenc quantiti whichev condit correspond normal oper statu instrument influenc quantiti occur ambient humid climat oper environ instrument intend damp heat steadi state condens damp heat cyclic condens test damp heat cyclic test condens import penetr vapour acceler effect breath condit condens humid factor damp heat steadi state reproduc applic measurand locat user condit result close agreement success measur differ measur result small compar mpe repeat applic measurand condit measur result close agreement success measur differ measur result small compar mpe discrimin sensit measur instrument suffici sensit discrimin threshold suffici low intend measur task durabl measur instrument design maintain adequ stabil metrolog characterist period time estim manufactur provid properli instal maintain manufactur instruct environment condit intend reliabl measur instrument design reduc effect defect lead inaccur measur result presenc defect obviou suitabl measur instrument featur facilit fraudul possibl unintent misus minim measur instrument suitabl intend take account practic work condit requir unreason demand user order obtain correct measur result error util measur instrument flow current control rang unduli bias measur instrument design measur valu measurand constant time measur instrument insensit small fluctuat measurand action measur instrument robust materi construct suitabl condit intend measur instrument design control measur task instrument market put special equip softwar control part instrument test procedur oper manual measur instrument softwar function measur function softwar critic metrolog characterist identifi inadmiss influenc softwar protect corrupt metrolog characterist measur instrument influenc inadmiss connect devic featur connect devic remot devic commun measur instrument hardwar compon critic metrolog characterist design secur secur measur foreseen provid evid intervent softwar critic metrolog characterist identifi secur softwar identif easili provid measur instrument evid intervent reason period time measur data softwar critic measur characterist metrolog import paramet store transmit adequ protect accident intent corrupt util measur instrument displai total quantiti suppli displai total quantiti suppli deriv partial refer basi payment reset inform born accompani instrument measur instrument bear inscript manufactur mark inform respect accuraci applic inform respect condit measur capac measur rang ident mark number ec type examin certif ec design examin certif inform addit devic provid metrolog result compli provis direct legal metrolog control instrument dimens small sensit composit bear relev inform packag accompani document requir provis direct suitabl mark instrument accompani inform oper simplic measur instrument make unnecessari inform easili understand includ relev rate oper condit mechan electromagnet environ class upper lower temperatur limit condens open close locat instruct instal mainten repair permiss adjust instruct correct oper special condit condit compat interfac assembl measur instrument group ident measur instrument locat util measur necessarili requir individu instruct manual instrument specif annex scale interv measur form integ unit measur symbol shown close numer materi measur mark nomin scale accompani unit measur unit measur symbol accord provis commun legisl unit measur symbol mark inscript requir requir clear eras unambigu transfer indic result indic result mean displai hard copi indic result clear unambigu accompani mark inscript inform user signific result easi read present result permit normal condit addit indic shown provid confus metrolog control indic case hard copi print record easili legibl eras measur instrument direct sale trade transact design present measur result parti transact instal intend critic case direct sale ticket provid consum ancillari devic compli requir direct bear restrict inform measur instrument intend util measur purpos remot read case fit metrolog control displai access tool consum read displai measur result serv basi price pai process data conclud trade transact measur instrument util measur instrument record durabl mean measur result accompani inform identifi transact measur repeat measur instrument intend absenc trade parti addition durabl proof measur result inform identifi transact request time measur conclud conform evalu measur instrument design readi evalu conform requir direct annex declar conform base intern product control quot declar conform base intern product control quot conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern satisfi requir direct technic document manufactur establish technic document articl document enabl assess conform instrument requir direct relev assess cover design manufactur oper instrument manufactur technic document dispos nation author ten year instrument manufactur manufactur manufactur measur ensur conform manufactur instrument requir direct written declar conform manufactur affix quot ce quot mark supplementari metrolog mark measur instrument satisfi requir direct declar conform drawn instrument model dispos nation author ten year instrument manufactur identifi instrument drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres manufactur establish commun authoris repres oblig contain paragraph respons person place instrument market annex declar conform base intern product control product test notifi bodi quot declar conform base intern product control product test notifi bodi quot conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern satisfi requir direct technic document manufactur establish technic document articl document enabl assess conform instrument requir direct relev assess cover design manufactur oper instrument manufactur technic document dispos nation author ten year instrument manufactur manufactur manufactur measur ensur conform manufactur instrument requir direct product check notifi bodi chosen manufactur carri product check carri interv determin order verifi qualiti intern check product take account inter alia technolog complex instrument quantiti product adequ sampl final product notifi bodi place market examin test identifi relev document refer articl equival test carri check conform instrument requir direct absenc relev document notifi bodi concern decid test carri case relev number instrument sampl conform accept qualiti level notifi bodi measur written declar conform manufactur affix quot ce quot mark supplementari metrolog mark respons notifi bodi refer paragraph identif number measur instrument satisfi requir direct declar conform drawn instrument model dispos nation author ten year instrument manufactur identifi model instrument drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres manufactur establish commun authoris repres oblig contain paragraph respons person place instrument market annex type examin quot type examin quot part conform assess procedur notifi bodi examin technic design measur instrument ensur declar technic design meet requir direct type examin carri manner notifi bodi decid manner specimen requir examin specimen repres product envisag complet measur instrument examin specimen repres product envisag critic part measur instrument assess adequaci technic design part measur instrument examin technic document support evid refer paragraph assess adequaci technic design measur instrument examin technic document support evid refer paragraph examin specimen applic type examin lodg manufactur notifi bodi choic applic includ address manufactur applic lodg authoris repres address addit written declar applic lodg notifi bodi technic document articl document enabl assess conform instrument requir direct relev assess cover design manufactur oper instrument specimen repres product envisag requir notifi bodi support evid adequaci technic design part measur instrument specimen requir support evid mention relev document appli relev document refer articl appli full includ result test carri laboratori manufactur test laboratori behalf respons notifi bodi specimen examin technic document verifi specimen manufactur conform identifi element design accord relev provis relev document refer articl element design appli relev provis document carri examin test carri check manufactur chosen appli solut relev document refer articl appli correctli carri examin test carri check manufactur chosen appli solut relev document refer articl solut adopt manufactur meet essenti requir direct agre applic locat examin test carri part measur instrument examin technic document support evid assess adequaci technic design part measur instrument manufactur process examin technic document assur manufactur adequ mean ensur consist product notifi bodi draw evalu report record activ undertaken accord paragraph outcom prejudic articl notifi bodi releas content report full part agreement manufactur technic design meet requir direct appli measur instrument notifi bodi issu ec type examin certif manufactur certif address manufactur authoris repres conclus examin condit valid data identif instrument certif annex attach certif annex relev inform conform evalu servic control conform manufactur instrument evalu examin type reproduc metrolog perform properli adjust mean content includ metrolog characterist type instrument measur requir ensur integr instrument seal identif softwar inform element identif instrument check visual extern conform type specif inform verifi characterist manufactur instrument case assembl inform ensur compat assembl measur instrument certif valid ten year date issu renew subsequ period ten year notifi bodi establish evalu report regard dispos member state design manufactur inform notifi bodi hold technic document ec type examin certif modif instrument affect conform instrument essenti requir condit valid certif modif requir addit approv form addit origin ec type examin certif notifi bodi immedi inform member state design ec type examin certif annex issu addit amend relat certif issu notifi bodi immedi inform member state design withdraw ec type examin certif notifi bodi hold technic file includ document submit manufactur period end valid certif manufactur copi ec type examin certif annex addit technic document year measur instrument manufactur manufactur authoris repres lodg applic refer paragraph carri oblig mention paragraph manufactur establish commun authoris repres oblig make technic document request respons person design manufactur annex declar conform type base intern product control quot declar conform type base intern product control quot part conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern conform type ec type examin certif satisfi requir direct manufactur manufactur measur ensur conform manufactur instrument type ec type examin certif requir direct written declar conform manufactur affix quot ce quot mark supplementari metrolog mark measur instrument conform type ec type examin certif satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi instrument model drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres manufactur establish commun authoris repres oblig mention paragraph respons person place instrument market annex declar conform type base intern product control product test notifi bodi quot declar conform type base intern product control product test notifi bodi quot part conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern conform type ec type examin certif satisfi requir direct manufactur manufactur measur ensur conform manufactur instrument type ec type examin certif requir direct product check notifi bodi chosen manufactur carri product check carri interv determin order verifi qualiti intern check product take account inter alia technolog complex instrument quantiti product adequ sampl final product notifi bodi place market examin test identifi relev document refer articl equival test carri check conform product type ec type examin certif requir direct absenc relev document notifi bodi concern decid test carri case relev number instrument sampl conform accept qualiti level notifi bodi measur written declar conform manufactur affix quot ce quot mark supplementari metrolog mark respons notifi bodi refer paragraph identif number measur instrument conform type ec type examin certif satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi model instrument drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres manufactur establish commun authoris repres oblig mention paragraph respons person place instrument market annex declar conform type base qualiti assur product process quot declar conform type base qualiti assur product process quot part conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern conform type ec type examin certif satisfi requir direct manufactur manufactur oper approv qualiti system product final product inspect test measur instrument concern paragraph subject surveil paragraph qualiti system manufactur lodg applic assess qualiti system notifi bodi choic applic includ relev inform instrument categori envisag document qualiti system technic document approv type copi ec type examin certif qualiti system ensur complianc instrument type ec type examin certif requir direct element requir provis adopt manufactur document systemat orderli manner form written polici procedur instruct qualiti system document permit consist interpret qualiti programm plan manual record adequ descript qualiti object organis structur respons power manag regard product qualiti manufactur qualiti control qualiti assur techniqu process systemat action examin test carri manufactur frequenc carri qualiti record inspect report test data calibr data qualif report personnel concern mean monitor achiev requir product qualiti effect oper qualiti system notifi bodi assess qualiti system determin satisfi requir refer paragraph presum conform requir respect qualiti system compli specif nation standard implement relev harmonis standard moment refer publish addit experi qualiti manag system audit team possess experi relev field metrolog instrument technolog knowledg applic requir direct evalu procedur includ inspect visit manufactur premis decis notifi manufactur notif conclus examin reason assess decis manufactur undertak fulfil oblig aris qualiti system approv maintain remain adequ effici manufactur notifi bodi approv qualiti system inform intend chang qualiti system notifi bodi evalu modif propos decid chang qualiti system satisfi requir refer paragraph assess requir notifi decis manufactur notif conclus examin reason assess decis surveil respons notifi bodi purpos surveil make manufactur fulfil oblig aris approv qualiti system manufactur notifi bodi entranc inspect purpos locat manufactur inspect test storag provid inform qualiti system document qualiti record inspect report test data calibr data qualif report personnel concern notifi bodi carri period audit make manufactur maintain appli qualiti system provid audit report manufactur addition notifi bodi pai unexpect visit manufactur visit notifi bodi carri product test carri verifi qualiti system function correctli provid manufactur visit report test carri test report written declar conform manufactur affix quot ce quot mark supplementari metrolog mark respons notifi bodi refer paragraph identif number measur instrument conform type ec type examin certif satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi model instrument drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user manufactur year instrument manufactur dispos nation author document refer paragraph indent chang refer paragraph approv decis report notifi bodi refer paragraph notifi bodi period make member state design list qualiti system approv issu refus immedi inform member state design withdraw qualiti system approv authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres annex declar conform base qualiti assur product process quot declar conform base qualiti assur product process quot conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern satisfi requir direct technic document manufactur establish technic document articl document enabl assess conform instrument requir direct relev assess cover design oper instrument manufactur technic document dispos nation author year instrument manufactur manufactur manufactur oper approv qualiti system product final product inspect test measur instrument concern paragraph subject surveil paragraph qualiti system manufactur lodg applic assess qualiti system notifi bodi choic applic includ relev inform instrument categori envisag document qualiti system technic document refer paragraph qualiti system ensur complianc instrument requir direct element requir provis adopt manufactur document systemat orderli manner form written polici procedur instruct qualiti system document permit consist interpret qualiti programm plan manual record adequ descript qualiti object organis structur respons power manag regard product qualiti manufactur qualiti control qualiti assur techniqu process systemat action examin test carri manufactur frequenc carri qualiti record inspect report test data calibr data qualif report personnel concern mean monitor achiev requir product qualiti effect oper qualiti system notifi bodi assess qualiti system determin satisfi requir refer paragraph presum conform requir respect qualiti system compli specif nation standard implement relev harmonis standard moment refer publish addit experi qualiti manag system audit team possess experi relev field metrolog instrument technolog knowledg applic requir direct evalu procedur includ inspect visit manufactur premis decis notifi manufactur notif conclus examin reason assess decis manufactur undertak fulfil oblig aris qualiti system approv maintain remain adequ effici manufactur notifi bodi approv qualiti system period inform intend chang qualiti system notifi bodi evalu propos decid chang qualiti system satisfi requir refer paragraph assess requir notifi decis manufactur notif conclus examin reason assess decis surveil respons notifi bodi purpos surveil make manufactur fulfil oblig aris approv qualiti system manufactur notifi bodi entranc inspect purpos locat manufactur inspect test storag provid inform qualiti system document technic document refer paragraph qualiti record inspect report test data calibr data qualif report personnel concern notifi bodi carri period audit make manufactur maintain appli qualiti system provid audit report manufactur addition notifi bodi pai unexpect visit manufactur visit notifi bodi carri product test carri verifi qualiti system function correctli provid manufactur visit report test carri test report written declar conform manufactur affix quot ce quot mark supplementari metrolog mark respons notifi bodi refer paragraph identif number measur instrument satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi model instrument drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user manufactur year instrument manufactur dispos nation author document refer paragraph indent chang refer paragraph approv decis report notifi bodi refer paragraph notifi bodi period make member state design list qualiti system approv issu refus immedi inform member state design withdraw qualiti system approv authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres annex declar conform type base qualiti assur final product inspect test quot declar conform type base qualiti assur final product inspect test quot part conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern conform type ec type examin certif satisfi requir direct manufactur manufactur oper approv qualiti system paragraph final product inspect test measur instrument concern subject surveil paragraph qualiti system manufactur lodg applic assess qualiti system notifi bodi choic applic includ relev inform instrument categori envisag document qualiti system technic document approv type copi ec type examin certif qualiti system ensur complianc instrument type ec type examin certif requir direct element requir provis adopt manufactur document systemat orderli manner form written polici procedur instruct qualiti system document permit consist interpret qualiti programm plan manual record adequ descript qualiti object organis structur respons power manag regard product qualiti examin test carri manufactur qualiti record inspect report test data calibr data qualif report personnel concern mean monitor effect oper qualiti system notifi bodi assess qualiti system determin satisfi requir refer paragraph presum conform requir respect qualiti system compli specif nation standard implement relev harmonis standard moment refer publish addit experi qualiti manag system audit team possess experi relev field metrolog instrument technolog knowledg applic requir direct evalu procedur includ inspect visit manufactur premis decis notifi manufactur notif conclus examin reason assess decis manufactur undertak fulfil oblig aris qualiti system approv maintain remain adequ effici manufactur notifi bodi approv qualiti system inform intend chang qualiti system notifi bodi evalu propos decid chang qualiti system satisfi requir refer paragraph assess requir notifi decis manufactur notif conclus examin reason assess decis surveil respons notifi bodi purpos surveil make manufactur fulfil oblig aris approv qualiti system manufactur notifi bodi entranc inspect purpos locat inspect test storag provid inform qualiti system document qualiti record inspect report test data calibr data qualif report personnel concern notifi bodi carri period audit make manufactur maintain appli qualiti system provid audit report manufactur addition notifi bodi pai unexpect visit manufactur visit notifi bodi carri product test carri verifi qualiti system function correctli provid manufactur visit report test carri test report written declar conform manufactur affix quot ce quot mark supplementari metrolog mark respons notifi bodi refer paragraph identif number measur instrument conform type ec type examin certif satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi instrument model drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user manufactur year instrument manufactur dispos nation author document refer indent paragraph chang refer subparagraph paragraph approv decis report notifi bodi refer paragraph final subparagraph paragraph paragraph notifi bodi period make member state design list qualiti system approv issu refus immedi inform member state design withdraw qualiti system approv authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres annex declar conform base qualiti assur final product inspect test quot declar conform base qualiti assur final product inspect test quot conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern conform requir direct technic document manufactur establish technic document articl document enabl assess conform instrument requir direct relev assess cover design manufactur oper instrument manufactur technic document dispos nation author year instrument manufactur manufactur manufactur oper approv qualiti system final product inspect test measur instrument concern paragraph subject surveil paragraph qualiti system manufactur lodg applic assess qualiti system notifi bodi choic applic includ relev inform instrument categori envisag document qualiti system technic document refer paragraph qualiti system ensur complianc instrument requir direct element requir provis adopt manufactur document systemat orderli manner form written polici procedur instruct qualiti system document permit consist interpret qualiti programm plan manual record document adequ descript qualiti object organis structur respons power manag regard product qualiti examin test carri manufactur qualiti record inspect report test data calibr data qualif report personnel concern mean monitor effect oper qualiti system notifi bodi assess qualiti system determin satisfi requir refer paragraph presum conform requir respect qualiti system compli specif nation standard implement relev harmonis standard moment refer publish addit experi qualiti manag system audit team possess experi relev field metrolog instrument technolog knowledg applic requir direct evalu procedur includ inspect visit manufactur premis decis notifi manufactur notif conclus examin reason assess decis manufactur undertak fulfil oblig aris qualiti system approv maintain remain adequ effici manufactur notifi bodi approv qualiti system inform intend chang qualiti system notifi bodi evalu propos decid chang qualiti system satisfi requir refer paragraph assess requir notifi decis manufactur notif conclus examin reason assess decis surveil respons notifi bodi purpos make manufactur fulfil oblig aris approv qualiti system manufactur notifi bodi entranc inspect purpos locat inspect test storag provid inform qualiti system document technic document refer paragraph qualiti record inspect report test data calibr data qualif report personnel concern notifi bodi carri period audit make manufactur maintain appli qualiti system provid audit report manufactur addition notifi bodi pai unexpect visit manufactur visit notifi bodi carri product test carri verifi qualiti system function correctli provid manufactur visit report test carri test report written declar conform manufactur affix quot ce quot mark supplementari metrolog mark respons notifi bodi refer paragraph identif number measur instrument satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi model instrument drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user manufactur year instrument manufactur dispos nation author document refer paragraph indent chang refer paragraph approv decis report notifi bodi refer paragraph notifi bodi period make member state design list qualiti system approv issu refus immedi inform member state design withdraw qualiti system approv authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres annex declar conform type base product verif quot declar conform type base product verif quot part conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument subject provis paragraph conform type ec type examin certif satisfi requir direct manufactur manufactur measur ensur conform manufactur instrument approv type ec type examin certif requir direct verif notifi bodi chosen manufactur carri examin test carri check conform instrument type ec type examin certif requir direct examin test check conform metrolog requir carri choic manufactur examin test instrument paragraph examin test instrument statist basi paragraph verif conform metrolog requir examin test instrument instrument individu examin test set relev document refer articl equival test carri verifi conform metrolog requir appli absenc relev document notifi bodi concern decid test carri notifi bodi issu certif conform respect examin test carri affix identif number approv instrument affix respons manufactur certif conform inspect nation author year certif instrument statist verif conform metrolog requir manufactur measur order manufactur process ensur homogen lot produc present instrument verif form homogen lot random sampl drawn lot requir paragraph instrument sampl individu examin test set relev document refer articl equival test establish conform metrolog requir appli carri determin lot accept reject absenc relev document notifi bodi concern decid test carri statist procedur meet requir statist control base attribut sampl system ensur level qualiti probabl accept conform limit qualiti probabl accept conform lot accept instrument lot approv instrument sampl found satisfi test notifi bodi issu certif conform respect examin test carri affix identif number approv instrument affix respons manufactur certif conform inspect nation author year certif instrument lot reject notifi bodi measur prevent place market lot event frequent reject lot notifi bodi suspend statist verif measur written declar conform manufactur affix quot ce quot mark supplementari metrolog mark measur instrument conform approv type satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi model instrument drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user agre notifi bodi refer paragraph manufactur affix notifi bodi identif number measur instrument notifi bodi respons manufactur agre notifi bodi respons affix notifi bodi identif number measur instrument manufactur process authoris repres manufactur oblig fulfil behalf respons authoris repres oblig contain paragraph annex declar conform base product verif quot declar conform base product verif quot conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument subject provis paragraph conform requir direct technic document manufactur establish technic document articl document enabl assess conform instrument requir direct relev assess cover design manufactur oper instrument manufactur technic document dispos nation author year instrument manufactur manufactur manufactur measur ensur conform manufactur instrument requir direct verif notifi bodi chosen manufactur carri examin test carri check conform instrument requir direct examin test check conform metrolog requir carri choic manufactur examin test instrument paragraph examin test instrument statist basi paragraph verif conform metrolog requir examin test instrument instrument individu examin test set relev document refer articl equival test carri verifi conform metrolog requir appli absenc relev document notifi bodi concern decid test carri notifi bodi issu certif conform respect examin test carri affix identif number approv instrument affix respons manufactur certif conform inspect nation author year certif instrument statist verif conform metrolog requir manufactur measur order manufactur process ensur homogen lot produc present instrument verif form homogen lot random sampl drawn lot requir paragraph instrument sampl individu examin test set relev document refer articl equival test establish conform metrolog requir appli carri determin lot accept reject absenc relev document notifi bodi concern decid test carri statist procedur meet requir statist control base attribut sampl system ensur level qualiti probabl accept conform limit qualiti probabl accept conform lot accept instrument lot approv instrument sampl found satisfi test notifi bodi issu certif conform respect examin test carri affix identif number approv instrument affix respons manufactur certif conform inspect nation author year certif instrument lot reject notifi bodi measur prevent place market lot event frequent reject lot notifi bodi suspend statist verif measur written declar conform manufactur affix quot ce quot mark supplementari metrolog mark measur instrument satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi model instrument drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user agre notifi bodi refer paragraph manufactur affix notifi bodi identif number measur instrument notifi bodi respons manufactur agre notifi bodi respons affix notifi bodi identif number measur instrument manufactur process authoris repres manufactur oblig fulfil behalf respons authoris repres oblig contain paragraph annex declar conform base unit verif quot declar conform base unit verif quot conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument subject provis paragraph conform requir direct technic document manufactur establish technic document articl make notifi bodi refer paragraph technic document enabl assess conform instrument requir direct relev assess cover design manufactur oper instrument manufactur technic document dispos nation author ten year manufactur manufactur measur ensur conform manufactur instrument requir direct verif notifi bodi chosen manufactur carri examin test set relev document refer articl equival test check conform instrument requir direct carri absenc relev document notifi bodi concern decid test carri notifi bodi issu certif conform respect examin test carri affix identif number approv instrument affix respons manufactur certif conform inspect nation author year certif instrument written declar conform manufactur affix quot ce quot mark supplementari metrolog mark respons notifi bodi refer paragraph identif number measur instrument satisfi requir direct declar conform drawn dispos nation author year instrument manufactur identifi instrument drawn copi declar suppli measur instrument authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres annex declar conform base full qualiti assur quot declar conform base full qualiti assur quot conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern satisfi requir direct manufactur manufactur oper approv qualiti system design manufactur final product inspect test measur instrument concern paragraph subject surveil paragraph qualiti system manufactur lodg applic assess qualiti system notifi bodi choic applic includ relev inform instrument categori envisag document qualiti system qualiti system ensur complianc instrument requir direct element requir provis adopt manufactur document systemat orderli manner form written polici procedur instruct qualiti system document permit consist interpret qualiti programm plan manual record adequ descript qualiti object organis structur respons power manag regard design product qualiti technic design specif includ standard appli relev document refer articl appli full mean ensur essenti requir direct appli instrument met design control design verif techniqu process systemat action design instrument pertain instrument categori cover manufactur qualiti control qualiti assur techniqu process systemat action examin test carri manufactur frequenc qualiti record inspect report test data calibr data qualif report personnel concern mean monitor achiev requir design product qualiti effect oper qualiti system notifi bodi assess qualiti system determin satisfi requir refer paragraph presum conform requir respect qualiti system compli specif nation standard implement relev harmonis standard moment refer publish addit experi qualiti manag system audit team possess experi relev field metrolog instrument technolog knowledg applic requir direct evalu procedur includ inspect visit manufactur premis decis notifi manufactur notif conclus examin reason assess decis manufactur undertak fulfil oblig aris qualiti system approv maintain remain adequ effici manufactur notifi bodi approv qualiti system inform intend chang qualiti system notifi bodi evalu propos decid chang qualiti system satisfi requir refer paragraph assess requir notifi decis manufactur notif conclus examin reason assess decis surveil respons notifi bodi purpos surveil make manufactur fulfil oblig aris approv qualiti system manufactur notifi bodi entranc inspect purpos locat manufactur inspect test storag provid inform qualiti system document qualiti record foreseen design part qualiti system result analys calcul test qualiti record foreseen manufactur part qualiti system inspect report test data calibr data qualif report personnel concern notifi bodi carri period audit make manufactur maintain appli qualiti system provid audit report manufactur addition notifi bodi pai unexpect visit manufactur visit notifi bodi carri product test carri respons verifi qualiti system function correctli provid manufactur visit report test carri test report written declar conform manufactur affix quot ce quot mark supplementari metrolog mark respons notifi bodi refer paragraph identif number measur instrument satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi instrument model drawn copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user manufactur year instrument manufactur dispos nation author document qualiti system refer paragraph indent chang refer paragraph approv decis report notifi bodi refer paragraph notifi bodi period make member state design list qualiti system approv issu refus immedi inform member state design withdraw qualiti system approv authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres annex declar conform base full qualiti assur design examin quot declar conform base full qualiti assur design examin quot conform assess procedur manufactur fulfil oblig laid annex ensur declar measur instrument concern satisfi requir direct manufactur manufactur oper approv qualiti system design manufactur final product inspect test measur instrument concern paragraph subject surveil paragraph adequaci technic design measur instrument examin provis paragraph qualiti system manufactur lodg applic assess qualiti system notifi bodi choic applic includ relev inform instrument categori envisag document qualiti system qualiti system ensur complianc instrument requir direct element requir provis adopt manufactur document systemat orderli manner form written polici procedur instruct qualiti system document permit consist interpret qualiti programm plan manual record adequ descript qualiti object organis structur respons power manag regard design product qualiti technic design specif includ standard appli relev document refer articl appli full mean ensur essenti requir direct appli instrument met design control design verif techniqu process systemat action design instrument pertain instrument categori cover manufactur qualiti control qualiti assur techniqu process systemat action examin test carri manufactur frequenc carri qualiti record inspect report test data calibr data qualif report personnel concern mean monitor achiev requir design product qualiti effect oper qualiti system notifi bodi assess qualiti system determin satisfi requir refer paragraph presum conform requir respect qualiti system compli specif nation standard implement relev harmonis standard moment refer publish offici journal addit experi qualiti manag system audit team possess experi relev field metrolog instrument technolog knowledg applic requir direct evalu procedur includ inspect visit manufactur premis decis notifi manufactur notif conclus examin reason assess decis manufactur undertak fulfil oblig aris qualiti system approv maintain remain adequ effici manufactur notifi bodi approv qualiti system inform intend chang qualiti system notifi bodi evalu propos decid chang qualiti system satisfi requir refer paragraph assess requir notifi decis manufactur notif conclus examin reason assess decis notifi bodi period make member state design list qualiti system approv issu refus immedi inform member state design withdraw qualiti system approv design examin manufactur lodg applic examin design notifi bodi refer paragraph applic enabl understand design manufactur oper instrument enabl assess conform requir direct includ address manufactur written declar applic lodg notifi bodi technic document articl document enabl assess conform instrument requir direct relev assess cover design oper instrument support evid adequaci technic design evid mention document appli relev document refer articl appli full includ result test carri laboratori manufactur test laboratori behalf respons notifi bodi examin applic design meet provis direct appli measur instrument issu ec design examin certif manufactur certif address manufactur conclus examin condit valid data identif approv instrument relev part technic document annex certif certif annex relev inform conform evalu servic control evalu conform manufactur instrument examin design reproduc metrolog perform properli adjust mean includ metrolog characterist design instrument measur requir ensur integr instrument seal identif softwar inform element identif instrument check visual extern conform design specif inform verifi characterist manufactur instrument case assembl inform ensur compat assembl measur instrument notifi bodi establish evalu report regard dispos member state design prejudic articl notifi bodi releas content report full part agreement manufactur certif valid ten year date issu renew subsequ period ten year manufactur deni design examin certif notifi bodi provid detail reason denial manufactur notifi bodi issu quot ec quot design examin certif inform fundament modif approv design modif approv design receiv addit approv notifi bodi issu quot ec quot design examin certif affect conform essenti requir direct condit valid certif prescrib condit instrument addit approv form addit origin quot ec quot design examin certif notifi bodi period make member state design quot ec quot design examin certif annex issu addit amend relat certif issu notifi bodi immedi inform member state design withdraw ec design examin certif manufactur authoris repres copi quot ec quot design examin certif annex addit technic document year measur instrument manufactur manufactur authoris repres establish commun oblig make technic document request respons person design manufactur surveil respons notifi bodi purpos surveil make manufactur fulfil oblig aris approv qualiti system manufactur notifi bodi entranc inspect purpos locat design manufactur inspect test storag provid inform qualiti system document qualiti record foreseen design part qualiti system result analys calcul test qualiti record foreseen manufactur part qualiti system inspect report test data calibr data qualif report personnel concern notifi bodi carri period audit make manufactur maintain appli qualiti system provid audit report manufactur addition notifi bodi pai unexpect visit manufactur visit notifi bodi carri product test carri respons verifi qualiti system function correctli provid manufactur visit report test carri test report written declar conform manufactur affix quot ce quot mark supplementari metrolog mark respons notifi bodi refer paragraph identif number measur instrument satisfi requir direct declar conform drawn instrument model dispos nation author year instrument manufactur identifi model instrument drawn mention number design examin certif copi declar suppli measur instrument market requir interpret appli batch consign individu instrument case larg number instrument deliv singl user manufactur year instrument manufactur dispos nation author document refer indent chang refer paragraph approv decis report notifi bodi refer paragraph authoris repres manufactur oblig contain paragraph fulfil behalf respons authoris repres annex mi water meter relev requir annex specif requir annex conform assess procedur list annex appli water meter intend measur volum clean cold heat water residenti commerci light industri definit water meter instrument design measur memoris displai volum meter condit water pass measur transduc minimum flowrat lowest flowrat water meter indic satisfi requir maximum permiss error mpe transit flowrat transit flowrat flowrat occur perman minimum flowrat flowrat rang divid zone quot upper zone quot quot lower zone quot zone characterist mpe perman flowrat highest flowrat water meter oper satisfactori manner normal condit steadi intermitt flow condit overload flowrat overload flowrat highest flowrat meter oper satisfactori manner short period time deterior specif requir rate oper condit manufactur rate oper condit instrument flowrat rang water valu flowrat rang fulfil condit amp gt year date entri forc direct ratio temperatur rang water valu temperatur rang fulfil condit meter design oper rang rel pressur rang water rang bar bar power suppli nomin ac voltag suppli limit dc suppli mpe mpe posit neg volum deliv flowrat transit flowrat includ overload flowrat water temperatur amp lt water temperatur amp gt mpe posit neg volum deliv flowrat minimum flowrat transit flowrat exclud water temperatur permiss effect disturb electromagnet immun effect electromagnet disturb water meter chang measur result greater critic chang defin indic measur result interpret valid result momentari variat interpret memoris transmit measur result undergo electromagnet disturb water meter recov oper mpe measur function safeguard recoveri measur data present disturb critic chang smaller valu volum half magnitud mpe upper zone measur volum volum mpe volum minut flowrat durabl test take account period time estim manufactur perform criteria satisfi variat measur result durabl test compar initi measur result exce meter volum includ exclud meter volum includ includ error indic volum meter durabl test exce meter volum includ exclud meter volum includ includ water meter intend meter water temperatur meter volum includ includ water meter intend meter water temperatur suitabl meter instal oper posit mark manufactur meter design measur revers flow case revers flow volum subtract cumul volum separ record mpe appli forward revers flow water meter design measur revers flow prevent revers flow withstand accident revers flow deterior chang metrolog properti unit measur meter volum displai cubic metr put member state ensur requir determin distributor person legal design instal meter meter accur measur consumpt foreseen forese conform assess conform assess procedur refer articl manufactur choos annex mi ga meter volum convers devic relev requir annex specif requir annex conform assess procedur list annex appli ga meter volum convers devic defin intend residenti commerci light industri definit ga meter instrument design measur memoris displai quantiti fuel ga volum mass pass convers devic devic fit ga meter automat convert quantiti measur meter condit quantiti base condit minimum flowrat qmin lowest flowrat ga meter indic satisfi requir maximum permiss error mpe maximum flowrat qmax highest flowrat ga meter indic satisfi requir mpe transit flowrat qt transit flowrat flowrat occur maximum minimum flowrat flowrat rang divid zone quot upper zone quot quot lower zone quot zone characterist mpe overload flowrat qr overload flowrat highest flowrat meter oper short period time deterior base condit condit measur quantiti fluid convert part specif requir ga meter rate oper condit manufactur rate oper condit ga meter take account flowrat rang ga fulfil condit gt tabl gt temperatur rang ga minimum rang fuel ga relat condit ga meter design rang gase suppli pressur countri destin manufactur ga famili group maximum oper pressur minimum temperatur rang climat environ nomin ac voltag suppli limit dc suppli maximum permiss error mpe ga meter indic volum meter condit mass tabl gt tabl gt error qt qmax sign exce class class ga meter temperatur convers convert volum mpe meter increas rang extend symmetr temperatur manufactur li rang addit increas permit interv permiss effect disturb electromagnet immun effect electromagnet disturb ga meter volum convers devic chang measur result greater critic chang defin indic measur result interpret valid result momentari variat interpret memoris transmit measur result undergo disturb ga meter recov oper mpe measur function safeguard recoveri measur data present disturb critic chang smaller valu quantiti half magnitud mpe upper zone measur volum quantiti mpe quantiti minut maximum flowrat effect upstream downstream flow disturb instal condit manufactur effect flow disturb exce mpe durabl test take account period time estim manufactur perform criteria satisfi class meter variat measur result durabl test compar initi measur result flow rate rang qt qmax exce measur result error indic durabl test exce mpe paragraph class meter variat measur result durabl test compar initi measur result exce mpe paragraph error indic durabl test exce mpe paragraph suitabl ga meter power main ac dc provid emerg power suppli devic mean ensur failur princip power sourc measur function safeguard dedic power sourc lifetim year lifetim warn shown indic devic suffici number digit ensur quantiti pass hour qmax return digit initi valu ga meter instal oper posit declar manufactur instal instruct ga meter test element enabl test carri reason time ga meter respect mpe flow direct flow direct mark unit meter quantiti displai cubic metr kilogram part ii specif requir volum convers devic volum convers devic constitut assembl articl definit indent volum convers devic essenti requir ga meter appli applic addit requir appli base condit convert quantiti manufactur base condit convert quantiti mpe ambient temperatur ambient humid nomin valu power suppli temperatur convers devic rate oper condit convers devic rate oper condit note error ga meter account suitabl electron convers devic capabl detect oper oper rang state manufactur paramet relev measur accuraci case convers devic stop integr convert quantiti totalis separ convert quantiti time oper oper rang electron convers devic capabl displai relev data measur addit equip part iii put conform assess put member state impos measur residenti measur perform mean class meter class meter qmax qmin ratio equal greater member state impos measur commerci light industri measur perform class meter requir paragraph member state ensur properti determin distributor person legal design instal meter meter accur measur consumpt foreseen forese conform assess conform assess procedur refer articl manufactur choos oder annex mi activ electr energi meter relev requir annex specif requir annex conform assess procedur list annex appli activ electr energi meter intend residenti commerci light industri note electr energi meter combin extern instrument transform depend measur techniqu appli annex cover electr energi meter instrument transform definit activ electr energi meter devic measur activ electr energi consum circuit electr current flow meter refer current transform oper meter design ist lowest declar meter regist activ electr energi uniti power factor polyphas meter balanc load imin error li maximum permiss error mpe polyphas meter balanc load itr error li smallest mpe class index meter imax maximum error li mpe voltag electr suppli meter refer voltag frequenc voltag suppli meter fn refer frequenc pf power factor cosφ cosin phase differ specif requir accuraci manufactur class index meter class indic defin class rate oper condit manufactur rate oper condit meter valu fn ist imin itr imax appli meter current valu meter satisfi condit tabl tabl gt tabl gt voltag frequenc power factor rang meter satisfi mpe requir tabl rang recognis typic characterist electr suppli public distribut system voltag frequenc rang amp lt amp lt fn amp lt amp lt fn power factor rang cosö induct cosö capacit mpe effect measurand influenc quantiti evalu separ measurand influenc quantiti constant refer valu error measur exce mpe state tabl calcul error measur gt refer graphic gt meter oper vari load current percentag error exce limit tabl tabl mpe percent rate oper condit defin load current level oper temperatur gt tabl gt electromechan polyphas meter current rang singl phase load limit itr amp lt amp lt imax meter oper temperatur rang relev mpe valu appli permiss effect disturb gener electr energi meter directli connect main suppli main current measurand special electromagnet environ electr meter meter compli electromagnet environ addit requir electromagnet environ permiss effect reflect situat disturb long durat affect accuraci critic chang valu transient disturb temporari degrad loss function perform meter recov affect accuraci critic chang valu forese high risk due lightn overhead suppli network predomin metrolog characterist meter protect effect disturb long durat tabl critic chang valu disturb long durat gt tabl gt permiss effect transient electromagnet phenomena effect electromagnet disturb electr energi meter immedi disturb output intend test accuraci meter produc puls signal energi critic chang reason time disturb meter recov oper mpe limit measur function safeguard recoveri measur data present prior disturb chang regist energi critic chang critic chang kwh imax number measur element meter volt imax amp overcurr critic chang suitabl rate oper voltag posit error meter exce displai total energi suffici number digit ensur meter oper hour full load imax pf indic return initi reset event loss electr circuit amount electr energi measur remain read period month run load voltag appli current flow current circuit current circuit open circuit meter regist energi voltag start meter start continu regist pf polyphas meter balanc load current equal ist unit electr energi measur displai kilowatt hour megawatt hour put member state impos measur residenti measur perform mean class meter purpos member state authoris requir class meter member state impos measur commerci light industri measur perform class meter purpos member state authoris requir class meter member state ensur current rang determin distributor person legal design instal meter meter accur measur consumpt foreseen forese conform assess conform assess procedur refer articl manufactur choos annex mi heat meter relev requir annex specif requir conform assess procedur list annex appli heat meter defin intend residenti commerci light industri definit heat meter instrument design measur heat heat exchang circuit liquid call heat convei liquid heat meter complet instrument combin instrument consist assembl flow sensor temperatur sensor pair calcul defin articl combin thereof temperatur heat convei liquid θin inlet heat exchang circuit θout outlet heat exchang circuit temperatur differ θin θout amp gt θmax upper limit heat meter function correctli mpe θmin lower limit heat meter function correctli mpe δθmax upper limit heat meter function correctli mpe δθmin lower limit heat meter function correctli mpe flow rate heat convei liquid qs highest permit short period time heat meter function correctli qp highest permit perman heat meter function correctli qi lowest permit heat meter function correctli thermal power heat exchang ps upper limit permit heat meter function correctli specif requir rate oper condit valu rate oper condit manufactur temperatur liquid θmax θmin temperatur differ δθmax δθmin subject restrict δθmax δθmin amp gt δθmin pressur liquid maximum posit intern pressur heat meter withstand perman upper limit temperatur flow rate liquid qs qp qi valu qp qi subject restrict qp qi amp gt thermal power ps accuraci class accuraci class defin heat meter mpe applic complet heat meter maximum permiss rel error applic complet heat meter express percent true accuraci class class ef ec ef ec paragraph class ef ec ef ec paragraph class ef ec ef ec paragraph permiss influenc electromagnet disturb instrument influenc static magnet field electromagnet field main frequenc influenc electromagnet disturb chang measur result greater critic chang laid requir indic measur result interpret valid result critic chang complet heat meter equal absolut mpe applic heat meter paragraph durabl test take account period time estim manufactur perform criteria satisfi flow sensor variat measur result durabl test compar initi measur result exce critic chang temperatur sensor variat measur result durabl test compar initi measur result exce inscript heat meter accuraci class limit flow rate limit temperatur limit temperatur differ place flow sensor instal flow return indic direct flow assembl provis assembl appli assembl manufactur manufactur heat meter consist assembl essenti requir heat meter appli assembl relev addit appli rel mpe flow sensor express accuraci class class ef qp class ef qp class ef qp error ef relat true relationship flow sensor output signal mass volum rel mpe temperatur sensor pair express δθmin error relat true relationship temperatur sensor pair output temperatur differ rel mpe calcul express ec δθmin error ec relat heat true heat critic chang assembl heat meter equal respect absolut mpe applic assembl paragraph inscript assembl gt tabl gt put member state impos measur residenti measur perform mean class meter member state impos measur commerci light industri authoris requir class meter requir paragraph member state ensur properti determin distributor person legal design instal meter meter accur measur consumpt foreseen forese conform assess conform assess procedur refer articl manufactur choos annex mi measur system continu dynam measur quantiti liquid water relev essenti requir annex specif requir annex conform assess procedur list annex appli measur system intend continu dynam measur quantiti volum mass liquid water term quot volum quot annex read quot mass kg quot definit meter instrument design measur continu memoris displai quantiti meter condit liquid flow measur transduc close fulli charg conduit calcul part meter receiv output signal measur transduc possibl measur instrument displai measur result measur instrument instrument connect calcul measur quantiti characterist liquid view make correct convers convers devic part calcul take account characterist liquid temperatur densiti measur measur instrument store memori automat convert volum liquid measur meter condit volum base condit mass mass liquid measur meter condit volum meter condit volum base condit note convers devic includ relev measur instrument base condit condit measur quantiti liquid meter condit convert measur system system compris meter devic requir ensur correct measur intend facilit measur oper fuel dispens measur system intend refuel motor vehicl small boat small aircraft servic arrang arrang custom measur system purpos obtain liquid servic devic specif devic part servic arrang measur system perform servic arrang minimum measur quantiti mmq smallest quantiti liquid measur metrolog accept measur system direct indic indic volum mass measur meter physic capabl measur note direct indic convert quantiti convers devic interrupt interrupt measur system consid interrupt interrupt liquid flow stop easili rapidli flowrat rang rang minimum flowrat qmin maximum flowrat qmax specif requir rate oper condit manufactur rate oper condit instrument flowrat rang flowrat rang subject condit flowrat rang measur system flowrat rang element meter ii meter measur system tabl gt tabl gt properti liquid measur instrument type liquid relev characterist temperatur rang pressur rang densiti rang viscos rang nomin ac voltag suppli limit dc voltag suppli base condit convert valu note paragraph prejudic member state oblig requir temperatur accord articl council direct eec octob harmonis structur excis duti miner oil heavi fuel oil lpg methan temperatur pursuant articl direct accuraci classif maximum permiss error mpe quantiti equal greater litr mpe indic tabl gt tabl gt quantiti litr mpe indic tabl gt tabl gt matter measur quantiti magnitud mpe greater valu absolut mpe tabl tabl absolut mpe minimum measur quantiti emin minimum measur quantiti greater equal litr condit appli condit emin fulfil condit emin amp gt smallest scale interv indic devic condit emin formula emin mmq mmq minimum measur quantiti numer line tabl minimum measur quantiti litr mention condit appli emin tabl relat line tabl convert indic case convert indic mpe line tabl convers devic mpe convert indic due convers devic equal valu tabl part convers devic test separ calcul mpe quantiti liquid indic applic calcul posit neg equal tenth mpe defin line tabl measur instrument measur instrument accuraci good valu tabl tabl gt tabl gt valu appli indic characterist quantiti liquid displai convers devic accuraci calcul function mpe calcul characterist quantiti liquid posit neg equal fifth fix requir paragraph appli calcul convers maximum permiss effect disturb effect electromagnet disturb measur system chang measur result greater critic chang defin paragraph indic measur result show momentari variat interpret memoris transmit measur result case interrupt system imposs perform measur chang measur result greater critic chang case measur system permit retriev measur result critic chang occur cut flow critic chang greater mpe measur quantiti emin durabl test take account period time estim manufactur perform criterion satisfi variat measur result durabl test compar initi measur result exce meter line tabl suitabl measur quantiti relat measur indic provid devic deviat scale interv devic scale interv case devic scale interv deviat greatest scale interv case servic arrang scale interv main indic devic measur system scale interv servic devic result measur deviat divert measur quantiti normal condit readili appar percentag air ga easili detect liquid lead variat error greater liquid potabl liquid liquid viscos exceed mpa potabl liquid liquid viscos exceed mpa allow variat smaller mmq appli case air ga pocket instrument direct sale measur system direct sale provid mean reset displai divert measur quantiti displai quantiti transact base perman parti transact accept measur result measur system direct sale interrupt percentag air ga liquid lead variat error greater valu paragraph fuel dispens displai fuel dispens capabl reset measur start measur inhibit displai reset measur system fit price displai differ price price calcul unit price quantiti exce price emin differ smallest monetari power suppli failur measur system provid emerg power suppli devic safeguard measur function failur main power suppli devic equip mean save displai data present order permit conclus transact progress mean stop flow moment failur main power suppli devic put tabl gt tabl gt note manufactur accuraci type measur system unit measur meter quantiti displai millilitr cubic centimetr litr cubic metr gram kilogram tonn conform assess conform assess procedur refer articl manufactur choos oj direct abolish direct ec oj annex mi automat weigh instrument relev essenti requir annex specif requir annex conform assess procedur list chapter annex appli automat weigh instrument defin intend determin mass bodi action graviti bodi definit automat weigh instrument instrument determin mass product intervent oper predetermin programm automat process characterist instrument automat catchweigh automat weigh instrument determin mass pre assembl discret load prepackag singl load loos materi automat checkweigh automat catchweigh subdivid articl mass subgroup differ mass nomin set point weight label automat catchweigh label individu articl weight weight price label automat catchweigh label individu articl weight price inform automat gravimetr fill instrument automat weigh instrument fill contain predetermin virtual constant mass product bulk discontinu totalis totalis hopper weigher automat weigh instrument determin mass bulk product divid discret load mass discret load determin sequenc sum discret load deliv bulk continu totalis automat weigh instrument continu determin mass bulk product conveyor belt systemat subdivis product interrupt movement conveyor belt rail weighbridg automat weigh instrument load receptor inclus rail convei railwai vehicl specif requir chapter requir common type automat weigh instrument rate oper condit manufactur rate oper condit instrument measurand measur rang term maximum minimum capac electr suppli influenc quantiti case ac voltag suppli nomin ac voltag suppli ac voltag limit case dc voltag suppli nomin minimum dc voltag suppli dc voltag limit mechan climat influenc quantiti minimum temperatur rang chapter annex mechan environ class annex paragraph applic instrument special mechan strain instrument incorpor vehicl manufactur defin mechan condit influenc quantiti applic rate oper characterist product weigh permiss effect disturb electromagnet environ requir perform critic chang relev chapter annex type instrument suitabl mean provid limit effect tilt load rate oper maximum permiss error mpe exceed normal oper adequ materi handl facil provid enabl instrument respect mpe normal oper oper control interfac clear effect integr displai present verifi oper adequ set capabl provid enabl instrument respect mpe normal oper result measur rang identifi printout conform assess conform assess procedur refer articl manufactur choos mechan system electromechan instrument electron system system softwar chapter ii automat catchweigh accuraci class instrument divid primari categori design manufactur primari categori divid accuraci class xi xii xiii amp xiv ii amp manufactur categori instrument categori appli instrument check prepackag made accord requir council direct eec decemb approxim law member state relat make volum prepackag liquid council direct eec januari approxim law member state relat make weight volum prepackag product applic prepackag accuraci class supplement factor quantifi maximum permiss standard deviat paragraph manufactur factor amp lt form neg number categori instrument categori appli automat catchweigh mpe error categori mpe categori instrument tabl gt tabl gt standard deviat maximum permiss standard deviat class instrument result multipl factor tabl tabl gt tabl gt class xi xii class xiii greater class xiv greater verif scale interv singl interv instrument tabl gt tabl gt verif scale interv multi interv instrument tabl gt tabl gt partial weigh rang total number partial rang measur rang measur rang class instrument manufactur account minimum capac class class ii amp lt amp lt amp lt class class scale grade postal scale garbag weigher dynam set dynam set facil oper load rang manufactur fit dynam set facil compens dynam effect load motion inhibit oper load rang capabl secur perform influenc factor electromagnet disturb mpe due influenc factor categori instrument automat oper tabl static weigh automat oper tabl categori instrument load automat oper tabl static weigh automat oper categori tabl critic chang due disturb verif scale interv temperatur rang class xi minimum rang class xii ii minimum rang chapter iii automat gravimetr fill instrument accuraci class manufactur refer accuraci class ref oper accuraci class es instrument type design refer accuraci class ref accuraci instrument type instal individu instrument design oper accuraci class account specif product weigh class design factor amp lt form neg number refer accuraci class ref applic static load oper accuraci class regim relat accuraci load weight multipli limit error class mpe static weigh error static load rate oper condit mpe refer accuraci class ref maximum permiss deviat fill averag tabl multipli class design factor instrument fill made load cumul select combin weigher mpe static load accuraci requir fill sum maximum permiss deviat individu load deviat averag fill tabl gt tabl gt note calcul deviat fill averag adjust account effect materi particl size error rel pre set set error instrument pre set fill weight maximum differ pre set averag mass fill exce maximum permiss deviat fill averag tabl perform influenc factor electromagnet disturb mpe due influenc factor paragraph critic chang due disturb chang static weight indic equal mpe paragraph calcul rate minimum fill chang give equival effect fill case instrument fill consist multipl load calcul critic chang round higher scale interv manufactur rate minimum fill chapter iv discontinu totalis accuraci class instrument divid accuraci class mpe tabl gt tabl gt totalis scale interv totalis scale interv dt rang max amp lt dt amp lt max minimum totalis load ómin minimum totalis load ómin load mpe equal totalis scale interv dt minimum load manufactur set instrument tare weigh discharg set devic automat oper inhibit indic vari dt instrument automat set devic dt instrument semi automat automat set devic oper interfac oper adjust reset function inhibit automat oper printout instrument equip print devic reset total inhibit total print printout total occur automat oper interrupt perform influenc factor electromagnet disturb mpe due influenc factor tabl tabl gt tabl gt critic chang due disturb totalis scale interv weight indic store total chapter continu totalis accuraci class instrument divid accuraci class measur rang manufactur measur rang ratio minimum net load weigh unit maximum capac minimum totalis load minimum totalis load σmin class class class totalis scale interv gener totalis devic mpe tabl gt tabl gt speed belt speed belt manufactur singl speed beltweigh variabl speed beltweigh manual speed set control speed vari nomin product speed speed belt gener totalis devic reset gener totalis devic perform influenc factor electromagnet disturb mpe due influenc factor load σmin time tabl round nearest totalis scale interv critic chang due disturb time tabl load equal σmin design class beltweigh round higher totalis scale interv chapter vi automat rail weighbridg accuraci class instrument divid accuraci class mpe mpe weigh motion singl wagon total train shown tabl tabl gt tabl gt mpe weight coupl uncoupl wagon weigh motion valu whichev greatest calcul tabl round nearest scale interv calcul tabl round nearest scale interv weight equal maximum wagon weight inscrib descript mark scale interv mpe weight train weigh motion valu whichev greatest calcul tabl round nearest scale interv calcul tabl weight singl wagon equal maximum wagon weight inscrib descript mark multipli number refer wagon exceed train round nearest scale interv scale interv wagon train exceed weigh coupl wagon error weigh result pass train exce mpe paragraph exce mpe scale interv relationship accuraci class scale interv tabl tabl gt tabl gt measur rang minimum capac greater result minimum wagon weight divid number partial weigh minimum wagon weight perform influenc factor electromagnet disturb mpe due influenc factor tabl tabl gt tabl gt critic chang due disturb scale interv oj direct amend direct eec oj oj direct amend eea agreement annex mi taximet relev requir annex specif requir annex conform assess procedur list annex appli taximet definit taximet devic work signal gener make measur instrument devic measur durat calcul distanc basi signal deliv distanc signal gener addition calcul displai fare paid trip basi calcul distanc measur durat trip fare total amount monei due trip base fix initi hire fee length durat trip fare includ supplement charg extra servic cross speed speed found divis time tariff distanc tariff normal calcul mode singl applic tariff fare calcul base applic time tariff cross speed applic distanc tariff cross speed normal calcul mode doubl applic tariff fare calcul base simultan applic time tariff distanc tariff trip oper posit mode taximet fulfil part function oper posit distinguish indic quot hire quot oper posit fare calcul disabl quot hire quot oper posit fare calcul take place basi initi charg tariff distanc travel time trip quot stop quot oper posit fare due trip fare calcul base time disabl design requir taximet design calcul distanc measur durat trip taximet design calcul displai fare increment step equal resolut fix member state oper posit quot hire quot taximet design displai final trip oper posit quot stop quot taximet appli normal calcul mode choos calcul mode secur set taximet suppli data secur interfac oper posit quot hire quot quot hire quot quot stop quot totalis data paragraph gener inform constant distanc signal gener date secur taxi identifi real time identif tariff fare inform trip total charg fare calcul fare supplement charg date start time finish time distanc travel tariff inform paramet tariff nation legisl requir devic connect interfac taximet devic requir secur set inhibit automat oper taximet reason presenc improp function requir devic relev adjust taximet constant distanc signal gener connect secur adjust rate oper condit mechan environ class appli manufactur rate oper condit instrument minimum temperatur rang climat environ limit dc power suppli instrument design maximum permiss error mpe mpe exclud error due applic taximet taxi time elaps minimum mpe distanc travel minimum mpe calcul fare minimum includ round signific digit fare indic permiss effect disturb electromagnet immun electromagnet class appli mpe laid paragraph respect presenc electromagnet disturb power suppli failur case reduct voltag suppli lower oper limit manufactur taximet continu work correctli resum correct function loss data voltag drop voltag drop temporari due restart engin abort exist measur return posit quot hire quot voltag drop longer period requir condit compat taximet distanc signal gener manufactur taximet supplement charg extra servic enter driver manual command exclud fare displai case taximet displai temporarili fare includ supplementari charg fare calcul calcul mode taximet addit displai mode total distanc durat trip displai real time valu displai passeng suitabl identifi valu identif readabl daylight night condit fare paid measur fraudul affect choic function pre program set free data set secur instrument set data enter secur possibl taximet separ secur set provis paragraph annex appli tariff taximet fit resett totalis valu total distanc travel taxi total distanc travel hire total number hire total amount monei charg supplement total amount monei charg fare totalis valu includ valu save paragraph condit loss power suppli disconnect power taximet totalis valu store year purpos read valu taximet medium adequ measur prevent displai totalis valu deceiv passeng automat chang tariff allow due distanc trip durat trip time dai date dai week properti taxi import correct taximet taximet provid mean secur connect taximet taxi instal purpos test instal taximet equip possibl test separ accuraci time distanc measur accuraci calcul taximet instal instruct manufactur instal manufactur instruct fraudul alter measur signal repres distanc travel suffici exclud gener essenti requir deal fraudul fulfil interest custom driver driver employ fiscal author protect taximet design respect mpe adjust period year normal taximet equip real timeclock mean time dai date automat chang tariff requir real time clock timekeep accuraci correct possibl clock minut week correct summer wintertim perform automat correct automat manual trip prevent valu distanc travel time elaps displai print accord direct unit distanc travel unit kingdom ireland date fix member state articl direct eec kilometr mile member state kilometr time elaps second minut hour suitabl keep mind resolut prevent misunderstand conform assess conform assess procedur refer articl manufactur choos distanc signal gener scope direct annex mi materi measur chapter materi measur length relev essenti requir annex specif requir annex conform assess procedur list chapter appli materi measur length defin requir suppli copi declar conform interpret appli batch consign individu instrument definit materi measur length instrument compris scale mark distanc legal unit length specif requir refer condit tape length equal greater metr maximum permiss error mpe met tractiv forc fifti newton forc valu manufactur mark tape case rigid semi rigid measur tractiv forc need appli refer temperatur manufactur mark measur mpe mpe posit neg mm consecut scale mark bl length round metr tabl termin interv bound surfac mpe distanc begin point increas tabl tabl gt tabl gt dip tape class ii case length scale mark sinker tape mpe mm applic formula mm mpe length consecut scale mark maximum permiss differ consecut interv tabl tabl gt tabl gt rule fold type joint error supplementari exceed mm class ii mm class iii materi materi materi measur length variat due temperatur excurs refer temperatur exce mpe appli class class measur manufactur intend thermal expans correct appli observ read measur made materi dimens alter materi subject wide rang rel humid includ class ii iii mark nomin mark measur millimetr scale number centimetr measur scale interv greater cm scale mark number conform assess conform assess procedur refer articl manufactur choos chapter ii capac serv measur relev essenti requir annex specif requir conform assess procedur list chapter appli capac serv measur defin requir suppli copi declar conform interpret appli batch consign individu instrument requir instrument bear inform respect accuraci appli definit capac serv measur capac measur drink glass jug thimbl measur design determin volum liquid pharmaceut product sold consumpt line measur capac serv measur mark line nomin capac brim measur capac serv measur intern volum equal nomin capac transfer measur capac serv measur intend liquid decant prior consumpt capac capac intern volum brim measur intern volum fill mark line measur specif requir refer condit temperatur refer temperatur measur capac posit correct indic free stand level surfac mpe tabl gt tabl gt materi capac serv measur made materi suffici rigid dimension stabl maintain capac mpe shape transfer measur design chang content equal mpe chang level mm brim fill mark transfer measur design complet discharg liquid measur imped mark nomin capac declar indel mark measur capac serv measur mark distinguish capac lead confus fill mark suffici clear durabl ensur mpe exceed conform assess conform assess procedur refer articl manufactur choos annex mi dimension measur instrument relev essenti requir annex specif requir annex conform assess procedur list annex appli dimension measur instrument type defin definit length measur instrument length measur instrument serv determin length rope type materi textil band cabl feed motion product measur area measur instrument area measur instrument serv determin area irregular shape object leather multi dimension measur instrument multi dimension measur instrument serv determin edg length length height width smallest enclos rectangular parallelepip product chapter equir common dimension measur instrument electromagnet immun effect electromagnet disturb dimension measur instrument chang measur result greater critic chang defin paragraph imposs perform measur momentari variat measur result interpret memoris transmit measur result variat measur result sever notic interest measur result critic chang equal scale interv conform assess conform assess procedur refer articl manufactur choos mechan electromechan instrument electron instrument instrument softwar chapter ii length measur instrument characterist product measur textil characteris characterist factor factor take stretchabl forc unit area product measur account defin formula ga rel elong cloth specimen wide tensil forc ga weight forc unit area cloth specimen oper condit rang dimens factor applic rang manufactur instrument rang factor tabl tabl gt tabl gt measur object transport measur instrument speed rang manufactur instrument measur result depend thick surfac condit kind deliveri big roll pile limit manufactur mpe instrument tabl gt tabl gt lm minimum measur length smallest length manufactur instrument intend true length type materi measur suitabl instrument tape length materi measur laid suitabl underlai suitabl tabl straight unstretch requir instrument ensur product measur unstretch intend stretchabl instrument design chapter iii area measur instrument oper condit rang dimens rang manufactur instrument condit product manufactur limit instrument due speed thick surfac condit relev product mpe instrument mpe dm requir present product case pull back stop product error measur displai blank scale interv instrument scale interv dm addit scale interv dm test purpos chapter iv multidimension measur instrument oper condit rang dimens rang manufactur instrument minimum dimens lower limit minimum dimens valu scale interv tabl tabl gt tabl gt speed product speed rang manufactur instrument mpe instrument mpe annex mi exhaust ga analys relev requir annex specif requir annex conform assess procedur list annex appli exhaust ga analys defin intend inspect profession mainten motor vehicl definit exhaust ga analys exhaust ga analys measur instrument serv determin volum fraction compon exhaust ga motor vehicl engin spark ignit moistur level sampl analys ga compon carbon monoxid carbon dioxid oxygen hydrocarbon hc content hydrocarbon express concentr hexan measur infrar absorpt techniqu volum fraction ga compon express percentag vol part million ppm vol exhaust ga analys calcul lambda volum fraction compon exhaust ga lambda lambda dimensionless repres burn effici engin term air fuel ratio exhaust gase determin refer standardis formula specif requir instrument class class defin exhaust ga analys relev minimum measur rang class shown tabl tabl class measur rang gt tabl gt rate oper condit valu oper condit manufactur climat mechan influenc quantiti minimum temperatur rang climat environ mechan environ class appli electr power influenc quantiti voltag frequenc rang ac voltag suppli limit dc voltag suppli ambient pressur minimum maximum valu ambient pressur class pmin amp lt hpa pmax amp gt hpa maximum permiss error mpe mpe defin fraction measur maximum error permit rate oper condit paragraph annex greater valu shown tabl absolut valu express vol ppm vol percentag valu percent true tabl mpe gt tabl gt mpe lambda calcul convent true calcul formula defin point annex direct ec ep council relat measur air pollut emiss motor vehicl amend council direct eec purpos valu displai instrument calcul permiss effect disturb volum fraction measur instrument critic chang equal mpe paramet concern effect electromagnet disturb chang measur result greater critic chang laid paragraph present measur result valid result requir resolut equal order magnitud higher valu shown tabl tabl resolut gt tabl gt lambda displai resolut standard deviat measur greater modulu mpe applic ga volum fraction measur hc instrument includ ga handl system final determin calibr gase second chang ga content fresh air measur instrument similar condit differ vol second chang fresh air oxygen free ga compon exhaust ga compon valu subject measur affect measur result half modulu mpe compon present maximum volum fraction vol vol vol vol vol ppm vol hc hexan water vapor satur exhaust ga analys adjust facil oper set ga calibr intern adjust adjust facil set intern adjust automat automat semi automat adjust facil instrument unabl make measur long adjust made exhaust ga analys detect hydrocarbon residu ga handl system carri measur hydrocarbon residu present measur exce ppm vol exhaust ga analys devic automat recognis malfunct sensor oxygen channel due wear break connect line exhaust ga analys capabl oper fuel petrol liquefi ga possibl select suitabl coeffici lambda calcul ambigu formula conform assess conform assess procedur refer articl manufactur choos oj