Does new magnetic resonance imaging technology provide better geometrical accuracy during stereotactic imaging?
Larger distortions were observed when using the SYMPHONY 1.5T unit than those with the EXPERT 1T unit. Typical average distortion in EXPERT 1T was not more than 0.6 mm and 0.9 mm for axial and coronal images, respectively. Typical mean distortion for SYMPHONY 1.5T was not more than 1 mm and 1.3 mm for axial and coronal images, respectively. The image sequence affected the distortions in both units. Coronal T2-weighted spin-echo images performed in subthalamic imaging produced the largest distortions of 2.6 mm and 3 mm in the EXPERT 1T and SYMPHONY 1.5T, respectively. Larger distortions were observed in coronal slices than in axial slices in both units, and this effect was more pronounced in SYMPHONY 1.5T. Noncentrally located slice positions in the investigated volume of the phantom were associated with larger distortions.
We conclude that induction of iNOS in atheroma of high-cholesterol-fed ApoE-/-/iNOS+/+ mice leads to increased production and activation of MMPs, with a subsequent decrease in collagen content, affording fragile plaque.