Mercury-in-glass thermometer and public health by decreasing the amount of mercury waste released. The HPA had, in 2007, released a guide to dealing with small spills of mercury. In the United States, both the American Academy of Pediatrics and the United States Environmental Protection Agency recommend that alternative thermometers be used in the home. Mercury-in-glass thermometer The mercury-in-glass or mercury thermometer was invented by physicist Daniel Gabriel Fahrenheit in Amsterdam (1714). It consists of a bulb containing mercury attached to a glass tube of narrow diameter; the volume of mercury in the tube is much less than the volume in the bulb.
Mercury-in-glass thermometer with fused quartz allows the upper temperature range to be extended to . Mercury cannot be used below the temperature at which it becomes solid, . If the thermometer contains nitrogen, the gas may flow down into the column when the mercury solidifies and be trapped there when the temperature rises, making the thermometer unusable until returned to the factory for reconditioning. To avoid this, some weather services require that all mercury-in-glass thermometers be brought indoors when the temperature falls to . To measure lower meteorological temperatures, a thermometer containing a mercury-thallium alloy which does not solidify until the temperature