Mercury-in-glass thermometer the thermometer, the bulb is made to reach thermal equilibrium with a temperature standard such as an ice/water mixture, and then with another standard such as water/vapour, and the tube is divided into regular intervals between the fixed points. In principle, thermometers made of different material (e.g., coloured alcohol thermometers) might be expected to give different intermediate readings due to different expansion properties; in practice the substances used are chosen to have reasonably linear expansion characteristics as a function of true thermodynamic temperature, and so give similar results. The application of mercury (1714) and Fahrenheit scale (1724) for liquid-in-glass thermometers
Mercury-in-glass thermometer with fused quartz allows the upper temperature range to be extended to . Mercury cannot be used below the temperature at which it becomes solid, . If the thermometer contains nitrogen, the gas may flow down into the column when the mercury solidifies and be trapped there when the temperature rises, making the thermometer unusable until returned to the factory for reconditioning. To avoid this, some weather services require that all mercury-in-glass thermometers be brought indoors when the temperature falls to . To measure lower meteorological temperatures, a thermometer containing a mercury-thallium alloy which does not solidify until the temperature