Medical thermometer and displaying temperature became available, electronic thermometers (often called "digital", because they display numeric values) have been used. Many display readings to great precision (0.1 °C or 0.2 °F, sometimes half that), but this should not be taken as a guarantee of accuracy: specified accuracy must be checked in documentation and maintained by periodical recalibration. A typical inexpensive electronic ear thermometer for home use has a displayed resolution of 0.1 °C, but a stated accuracy within ±0.2 °C (±0.35 °F) when new. The first electronic clinical thermometer, invented in 1954, used a flexible probe that contained a Carboloy thermistor. Types
Mercury-in-glass thermometer with fused quartz allows the upper temperature range to be extended to . Mercury cannot be used below the temperature at which it becomes solid, . If the thermometer contains nitrogen, the gas may flow down into the column when the mercury solidifies and be trapped there when the temperature rises, making the thermometer unusable until returned to the factory for reconditioning. To avoid this, some weather services require that all mercury-in-glass thermometers be brought indoors when the temperature falls to . To measure lower meteorological temperatures, a thermometer containing a mercury-thallium alloy which does not solidify until the temperature