Mercury-in-glass thermometer the thermometer, the bulb is made to reach thermal equilibrium with a temperature standard such as an ice/water mixture, and then with another standard such as water/vapour, and the tube is divided into regular intervals between the fixed points. In principle, thermometers made of different material (e.g., coloured alcohol thermometers) might be expected to give different intermediate readings due to different expansion properties; in practice the substances used are chosen to have reasonably linear expansion characteristics as a function of true thermodynamic temperature, and so give similar results. The application of mercury (1714) and Fahrenheit scale (1724) for liquid-in-glass thermometers
Thermometer instruments can give readings to one thousandth of a degree. However, this precision does not mean the reading is true or accurate, it only means that very small changes can be observed. A thermometer calibrated to a known fixed point is accurate (i.e. gives a true reading) at that point. Most thermometers are originally calibrated to a constant-volume gas thermometer. In between fixed calibration points, interpolation is used, usually linear. This may give significant differences between different types of thermometer at points far away from the fixed points. For example, the expansion of mercury in a glass thermometer is slightly