Mercury-in-glass thermometer related to the rapid cooling (and contraction) of the glass. When Celsius decided to use his own temperature scale, he originally defined his scale "upside-down", i.e. he chose to set the boiling point of pure water at 0 °C (212 °F) and the freezing point at -100 °C (-32 °F). One year later, Frenchman Jean-Pierre Christin proposed to invert the scale with the freezing point at and the boiling point at . He named it Centigrade (100 grades). Finally, Celsius proposed a method of calibrating a thermometer: These points are adequate for approximate calibration but both vary with atmospheric pressure.
Thermometer different from the change in resistance of a platinum resistance thermometer, so these two will disagree slightly at around 50 °C. There may be other causes due to imperfections in the instrument, e.g. in a liquid-in-glass thermometer if the capillary tube varies in diameter. For many purposes reproducibility is important. That is, does the same thermometer give the same reading for the same temperature (or do replacement or multiple thermometers give the same reading)? Reproducible temperature measurement means that comparisons are valid in scientific experiments and industrial processes are consistent. Thus if the same type of thermometer is calibrated in