Mercury-in-glass thermometer the thermometer, the bulb is made to reach thermal equilibrium with a temperature standard such as an ice/water mixture, and then with another standard such as water/vapour, and the tube is divided into regular intervals between the fixed points. In principle, thermometers made of different material (e.g., coloured alcohol thermometers) might be expected to give different intermediate readings due to different expansion properties; in practice the substances used are chosen to have reasonably linear expansion characteristics as a function of true thermodynamic temperature, and so give similar results. The application of mercury (1714) and Fahrenheit scale (1724) for liquid-in-glass thermometers
Beckmann thermometer (R on diagram) at the upper end of the tube, by means of which the quantity of mercury in the bulb can be increased or diminished so that the instrument can be set to measure temperature differences at either high or low temperature values. In contrast, the range of a typical mercury-in-glass thermometer is fixed, being set by the calibration marks etched on the glass or the marks on the printed scale. In setting the thermometer, a sufficient amount of mercury must be left in the bulb and stem to give readings between the required temperatures. First, the thermometer is