Alcohol thermometer to give a background for reading the scale. The range of usefulness of the thermometer is set by the boiling point of the liquid used. In the case of the ethanol-filled thermometer, the upper limit for measurement is 78 °C (172.4 °F), which makes it useful for measuring daytime, nighttime and body temperatures, although not for anything much hotter than these. Ethanol-filled thermometers are used in preference to mercury for meteorological measurements of minimum temperatures and can be used down to −70 °C (-94 °F). The physical limitation of the ability of a thermometer to measure low temperature is the
Beckmann thermometer (R on diagram) at the upper end of the tube, by means of which the quantity of mercury in the bulb can be increased or diminished so that the instrument can be set to measure temperature differences at either high or low temperature values. In contrast, the range of a typical mercury-in-glass thermometer is fixed, being set by the calibration marks etched on the glass or the marks on the printed scale. In setting the thermometer, a sufficient amount of mercury must be left in the bulb and stem to give readings between the required temperatures. First, the thermometer is