Ammeter in both vanes, the vanes repel each other and the moving vane deflects against the restoring force provided by fine helical springs. The deflection of a moving iron meter is proportional to the square of the current. Consequently, such meters would normally have a non linear scale, but the iron parts are usually modified in shape to make the scale fairly linear over most of its range. Moving iron instruments indicate the RMS value of any AC waveform applied. Moving iron ammeters are commonly used to measure current in industrial frequency AC circuits. In a hot-wire ammeter, a current passes
Network analyzer (electrical) Network analyzer (electrical) A network analyzer is an instrument that measures the network parameters of electrical networks. Today, network analyzers commonly measure s–parameters because reflection and transmission of electrical networks are easy to measure at high frequencies, but there are other network parameter sets such as y-parameters, z-parameters, and h-parameters. Network analyzers are often used to characterize two-port networks such as amplifiers and filters, but they can be used on networks with an arbitrary number of ports. Network analyzers are used mostly at high frequencies; operating frequencies can range from 5 Hz to 1.05 THz. Special types of network analyzers