Ammeter permanent magnet causes the coil to move. The modern form of this instrument was developed by Edward Weston, and uses two spiral springs to provide the restoring force. The uniform air gap between the iron core and the permanent magnet poles make the deflection of the meter linearly proportional to current. These meters have linear scales. Basic meter movements can have full-scale deflection for currents from about 25 microamperes to 10 milliamperes. Because the magnetic field is polarised, the meter needle acts in opposite directions for each direction of current. A DC ammeter is thus sensitive to which way round
Electrical conductivity meter necessary to develop a specific curve to each measurement point or use commercially available products. The high pressure and temperature of cooking process combined with high concentration of alkali components puts a heavy strain on conductivity sensors that are installed in process. The scaling on the electrodes need to be taken into account, otherwise conductivity measurement drifts causing increased calibration and maintenance need. Electrical conductivity meter An electrical conductivity meter (EC meter) measures the electrical conductivity in a solution. It is commonly used in hydroponics, aquaculture, aquaponics, and freshwater systems to monitor the amount of nutrients, salts or impurities in