Ammeter in both vanes, the vanes repel each other and the moving vane deflects against the restoring force provided by fine helical springs. The deflection of a moving iron meter is proportional to the square of the current. Consequently, such meters would normally have a non linear scale, but the iron parts are usually modified in shape to make the scale fairly linear over most of its range. Moving iron instruments indicate the RMS value of any AC waveform applied. Moving iron ammeters are commonly used to measure current in industrial frequency AC circuits. In a hot-wire ammeter, a current passes
Electrical cardiometry Electrical cardiometry Electrical cardiometry is a method based on the model of "Electrical Velocimetry", and non-invasively measures stroke volume (SV), cardiac output (CO), and other hemodynamic parameters through the use of 4 surface ECG electrodes. Electrical cardiometry is a method trademarked by Cardiotronic, Inc., and is U.S. FDA approved for use on adults, children, and neonates. The measured bioimpedance over time can be expressed as the superposition of three components: where Z is the quasi-static portion of the electrical impedance (base impedance), ΔZ are the changes of impedance due to the respiratory cycle, and ΔZ are the changes of impedance