Ammeter in both vanes, the vanes repel each other and the moving vane deflects against the restoring force provided by fine helical springs. The deflection of a moving iron meter is proportional to the square of the current. Consequently, such meters would normally have a non linear scale, but the iron parts are usually modified in shape to make the scale fairly linear over most of its range. Moving iron instruments indicate the RMS value of any AC waveform applied. Moving iron ammeters are commonly used to measure current in industrial frequency AC circuits. In a hot-wire ammeter, a current passes
Electrical conductivity meter necessary to develop a specific curve to each measurement point or use commercially available products. The high pressure and temperature of cooking process combined with high concentration of alkali components puts a heavy strain on conductivity sensors that are installed in process. The scaling on the electrodes need to be taken into account, otherwise conductivity measurement drifts causing increased calibration and maintenance need. Electrical conductivity meter An electrical conductivity meter (EC meter) measures the electrical conductivity in a solution. It is commonly used in hydroponics, aquaculture, aquaponics, and freshwater systems to monitor the amount of nutrients, salts or impurities in