Anemometer is blowing on the back of the cup on the opposite end of the cross. Since a hollow hemisphere has a drag coefficient of .38 on the spherical side and 1.42 on the hollow side, more force is generated on the cup that is presenting its hollow side to the wind. Because of this asymmetrical force, torque is generated on the axis of the anemometer, causing it to spin. Theoretically, the speed of rotation of the anemometer should be proportional to the wind speed, because the force produced on an object is proportional to the speed of the fluid flowing
Anemometer which follows the same concept, but uses two pins or strings to monitor the variation in temperature. The strings contain fine wires, but encasing the wires makes them much more durable and capable of accurately measuring air, gas, and emissions flow in pipes, ducts, and stacks. Industrial applications often contain dirt that will damage the classic hot-wire anemometer. In laser Doppler velocimetry, laser Doppler anemometers use a beam of light from a laser that is divided into two beams, with one propagated out of the anemometer. Particulates (or deliberately introduced seed material) flowing along with air molecules near where the