Anemometer mounted on a vertical shaft. The air flow past the cups in any horizontal direction turned the shaft at a rate that was roughly proportional to the wind speed. Therefore, counting the turns of the shaft over a set time interval produced a value proportional to the average wind speed for a wide range of speeds. It is also called a rotational anemometer. On an anemometer with four cups, it is easy to see that since the cups are arranged symmetrically on the end of the arms, the wind always has the hollow of one cup presented to it and
Anemometer past it, but other factors influence the rotational speed, including turbulence produced by the apparatus, increasing drag in opposition to the torque that is produced by the cups and support arms, and friction of the mount point. When Robinson first designed his anemometer, he asserted that the cups moved one-third of the speed of the wind, unaffected by the cup size or arm length. This was apparently confirmed by some early independent experiments, but it was incorrect. Instead, the ratio of the speed of the wind and that of the cups, the "anemometer factor", depends on the dimensions of the