Hardness test is Mohs scale, which is used in mineralogy. One tool to make this measurement is the sclerometer. Another tool used to make these tests is the pocket hardness tester. This tool consists of a scale arm with graduated markings attached to a four-wheeled carriage. A scratch tool with a sharp rim is mounted at a predetermined angle to the testing surface. In order to use it a weight of known mass is added to the scale arm at one of the graduated markings, the tool is then drawn across the test surface. The use of the weight and markings
Leeb rebound hardness test a Brinell indentation on the same material is ~3 mm (hardness value ~400 HBW 10/3000), with a standard-compliant measuring time of ~15 seconds plus the time for measuring the indentation. Depending on the probe (“impact device”) and indenter (“impact body”) types that vary by geometry, size, weight, material and spring force, diverse impact devices and hardness units are distinguished, e.g.: Generally, impact device types are optimized for certain application fields. This is similar to using various indenter geometries and test loads in Rockwell (e.g. HRA, HRB, HRC), Brinell and Vickers. Leeb hardness results in HL are often converted to the