Indentation hardness test, performed with either 3 gf or 9 gf loads, preceded the development of microhardness testers using traditional indenters. In 1925, Smith and Sandland of the UK developed an indentation test that employed a square-based pyramidal indenter made from diamond. They chose the pyramidal shape with an angle of 136° between opposite faces in order to obtain hardness numbers that would be as close as possible to Brinell hardness numbers for the specimen. The Vickers test has a great advantage of using one hardness scale to test all materials.The first reference to the Vickers indenter with low loads was made
Indentation hardness Knoop hardness is calculated based on the projected area of the indent divided by the applied force, also giving test units in kgf/mm². The Vickers microindentation test is carried out in a similar manner welling to the Vickers macroindentation tests, using the same pyramid. The Knoop test uses an elongated pyramid to indent material samples. This elongated pyramid creates a shallow impression, which is beneficial for measuring the hardness of brittle materials or thin components. Both the Knoop and Vickers indenters require prepolishing of the surface to achieve accurate results. Scratch tests at low loads, such as the Bierbaum microcharacter