determinar a influncia de trs sistemas distintos de instrumentos rotatrios para alargamento cervical na determinao do comprimento real de trabalho .
material e mtodos : trinta primeiros molares inferiores foram submetidos ao acesso endodntico convencional e odontometria inicial pelo mtodo visual , utilizando - se para irrigao / aspirao da cmara pulpar o hipoclorito de sdio a 5% .
procedeu - se a ampliao cervical dos canais msio - vestibulares empregando - se diferentes alargadores cervicais .
de acordo com o sistema de ampliao cervical empregado , as 30 amostras foram divididas aleatoriamente em 3 grupos de 10 dentes cada .
no grupo 1 as brocas de gates - glidden foram utilizadas para a ampliao cervical ; no grupo 2 , orifice openers foram empregados ; e , no grupo 3 , foi usado o sistema la axxess .
foram utilizadas em ordem crescente , duas numeraes do sistema rotatrio correspondente a cada grupo e , aps , realizou - se a odontometria final . a leitura das duas tomadas odontomtricas foi realizada com paqumetro digital , em milmetros , considerando - se duas casas decimais para averiguar a provvel discrepncia entre ambas as tomadas .
a anlise de varincia ( anova ) e o teste de tukey ( p 0,05 ) demonstraram diferena estatstica entre o grupo 1 e os grupos 2 e 3 .
todos os grupos apresentaram encurtamento do comprimento de trabalho aps alargamento cervical ; os grupos preparados com instrumentos la axxes e orifice opener apresentaram os melhores resultados .
during the last years , endodontics has been searching for new instruments , instrumentation techniques , irrigants and drugs to achieve the best cleaning and disinfection of root canals12 . within this context
philippas16 mentioned that physiological or pathological continuous and progressive dentin formation in the pulp chamber floor created interferences that reduced the root canal diameter , especially at the cervical third , and thus it should be removed during endodontic treatment .
leeb7 investigated the effect of cervical flaring on the establishment of apical diameter of different groups of teeth and observed that the use of a single gates - glidden bur removed the cervical constriction allowing larger files to reach the apical portion .
lopes and costa filho10 described that , in addition to the advantages of using specific instruments to prepare the root canal openings , this clinical step also provides better action of irrigation and suction , since auxiliary chemical substances may be placed close to the apical third , thereby allowing elimination of a larger amount of smear layer .
estrela , et al.5 mentioned the main advantages of cervical flaring in endodontics : reducing pressure during irrigation due to the possibility of overflow ; deeper penetration of irrigant ; effective removal of cervical contamination ; reduced possibility of ledges , apical deviation or instrument fracture ; better control of the active tip of the instrument , thus overcoming the influence from apical curvature by a straighter access ; and an improvement of the obturation steps , especially lateral condensation .
the anatomical complexity of molars , especially the mandibular molar and its mesial root , increases the difficulty of routine endodontic procedures , due to the number of root canals , curvatures or even the apical anatomical configuration with several isthmi.21 
  travassos , et al.19 concluded that preparation of the cervical third of curved root canals before the use of endodontic instruments allowed straighter access to the apical region , reducing the possibility of interferences or apical deviation .
tan and messer18 reported that continuous dentin formation reduces the volume of the pulp chamber in different types of teeth due to the progressive narrowing of root canals , besides the appearance of cervical constrictions .
these constrictions should be removed by accomplishment of cervical preparation , thus allowing more accurate working length determination and more precise establishment of the initial apical instrument diameter ( iai ) .
lopes and siqueira jr.11 stated that cervical flaring allowed the introduction of endodontic files without locking into the root canal on the pulp chamber , until the first root canal curvature is reached , thus eliminating cervical interferences to allow better establishment of the initial apical instrument ( iai ) .
tan and messer18 , investigated the effect of instrument type and the influence of cervical flaring on the establishment of the iai of molars and premolars , which pose the greatest challenge in endodontic treatment .
these authors concluded that the removal of cervical dentin projections improved the insertion of larger files to the working length , thus providing criteria that should allow the achievement of a adequate biomechanical preparation .
macedo , et al.13 evaluated the wear of the cervical third of mesiobuccal canals of maxillary molars submitted to different procedures for preparation of the root canal openings and concluded that this clinical step provides excellent configuration of the cervical root canal surface of these roots , thus allowing better uprighting of the endodontic instrument for root canal shaping .
barroso , et al.3 investigated the influence of different methods of cervical flaring on the establishment of the iai in maxillary premolars and concluded that larger instruments may be introduced to the real working length after cervical flaring .
pcora , et al.15 analyzed the influence of cervical preflaring on establishment of the iai and concluded that this step allowed the introduction of larger files to the real working length ( rwl ) .
therefore , the objective of this study was to investigate the influence of three different rotary systems for cervical flaring on establishment of the rwl , testing the hypothesis that there is no difference between the systems studied .
the present study was approved by the ethics committee of the university of passo fundo ( upf ) .
the study used 30 permanent lower first molars obtained from the tooth bank of the dental school ( fo - upf ) .
, the teeth were washed in tap water for 24 hours to remove any remnants of thymol solution .
conventional endodontic access was performed removing and contouring the pulp chamber roof and using irrigation / suction with 5% sodium hypochlorite .
the working length was measured by introducing a number 10 k file into the mesiobuccal root canal of each teeth until the apical foramen , followed by 1-mm stepback to establish the real working length ( rwl ) .
teeth were randomly divided into 3 groups ( n= 10 ) and the cervical flaring of the mesiobuccal canal was performed using three different rotary instruments .
group 1 ( g1 ) was prepared with gates - glidden burs in a conventional low - speed handpiece at 5,000 rpm .
preparation was initiated with number 1 bur , which was introduced 3 times until resistance was found and then removed .
irrigation / suction was performed with 5% sodium hypochlorite , followed by utilization of number 2 bur as used for number 1 bur .
group 2 ( g2 ) was prepared using nickel - titanium cervical flaring instruments ( orifice openers , sybronendo , glendora , usa ) in a tc 3000 handpiece ( nuvag , tcm endo , goldach , switzerland ) set at 315 rpm , at configurations 25:08 and 25:10 , in increasing order of introduction , until resistance was found .
cervical flaring in group 3 ( g3 ) was performed with nickel - titanium la axxes instruments ( sybronendo , glendora , usa ) , connected to a low - speed handpiece at 5,000 rpm . the system was also used in increasing order , at configurations 20:06 and 35:06 , until resistance was found .
the working length measurements were performed as previously mentioned . a digital calyper ( mitutoyo , tokyo , japan )
was used to measure the working length , in millimeters , investigating a possible discrepancy between initial and final measurements .
the differences between both working length measurements were calculated and statistically analyzed using analysis of variance ( anova ) and tukey 's multiple range tests with a significance level of 5% ( =0.05 ) .
analysis of variance ( anova ) revealed statistically significant differences , which were identified by the tukey 's multiple range test ( p0.05 ) and are presented in table 1 .
no significant differences were found between the mean values of g2 and g3 , which were different from the mean values of g1 .
this procedure aims to remove cervical interferences from the root canal openings , which represent an obstacle to free access of endodontic instruments to the apical portions of the root canal .
removal of these anatomical interferences allows free access , thus enhancing root canal shaping at the apical third2 .
root curvatures , especially at the apical third , represent an additional difficulty to the professional , since they may be difficult to observe .
the curvatures in thin , flattened root canals , as the mesial roots of molars , which represent a remarkable difficulty for preparation and several different techniques have been advocating in an attempt to solve this problem .
therefore , this study was conducted on the mesiobuccal root of the mandibular first molar because it presents difficult access in the oral cavity and it may have more than one canal20 .
teeth storage using 0.1% thymol at 9c maintained the specimens hydrated and structurally stabilized , with no need of tissue fixation and absence of bacterial proliferation , thus simulating the clinical conditions and enhancing the intervention1 .
this study investigated different systems used for cervical flaring to examine the effect of cervical interference removal on the establishment of the anatomical diameter of the root canal .
thus , rotary instruments were used for cervical flaring because they allow root canal preparation with larger instruments and reduce the risk of failures as ledges , perforations and zips17 .
in addition , the advantages of nickel - titanium instruments ( ni - ti ) and rotary instrumentation include more effective removal of debris due to the possibility of overflow and continuous rotation of instruments , faster preparation and reduction of root canal transportation6 .
the instruments selection criteria for this study include the following : 1- gates - glidden ( n. 1 and 2 ) are widely used as rotary instruments for cervical flaring8 ; 2- orifice opener ( sybronendo , glendora , usa ) ( 25.08 , 25.10 ) has recognized performance for cervical preparation of root canals9 ; and 3- la axxess ( 20.06 , 35.06 ) , has recently been introduced into the market with almost no literature available and , therefore , should be investigated .
the rotary instruments used during cervical preparation following the aforementioned sequence steps , revealed to be an adequate , fast and safe procedure , as also observed by aun , et al2 .
cervical flaring was performed following the " free tip preparation " technique , using an instrumentation sequence from the lowest to the highest taper .
thus , the instrument tip is free and often act as a guide , significantly reducing the torsional fracture and providing more efficient action of the instrument , in agreement with pcora , et al.14 , yet in disagreement with tan and messer18 , who performed cervical pre - flaring based on the " crown - down " technique , adapted from manual instrumentation .
neither root perforation nor instrument fracture occurred in this study , which can be explain by the linear limit of instrument penetration , instrument introduction until resistance was found , observation of manufacturers ' instructions , experience of the operator and standardization of the degree of curvature of each canal .
therefore , the specimens submitted to pre - flaring with la axxess instruments ( g3 ) produced the highest mean value difference between the first and second working length measurements , which was not statistical different form the mean value of g2 ( p>0.05 ) , in accordance with pcora , et al.15 .
this finding may be due to the characteristics of la axxess instruments , such as the metal alloy properties , the taper ( 0.06 ) , the inactive tip and the flute design of the active tip that allows the removal of cervical interferences without occurrence of deviations or perforations .
along with other research results , this study contributes with the improvement of the endodontic therapy , considering the importance of a proper apical flaring of root canals .
all groups exhibited shorter working length after cervical flaring ; considering the methods of cervical flaring to establish of the working length investigated in this study , the specimens prepared with instruments la axxes ( g3 ) and orifice opener ( g2 ) presented the best results .
 objective : to investigate the influence of three different rotary systems for cervical flaring on establishment of the real working length . 
 material and methods : thirty mandibular first molars were submitted to conventional endodontic access and initial working length measurement , followed by irrigation / suction of the pulp chamber with 5% sodium hypochlorite . 
 teeth were randomly divided into 3 groups ( n=10 ) and cervical flaring of the mesiobuccal canals were performed using one of the following instrument systems : group 1  gates - glidden burs ; group 2  orifice openers ; group 3  la axxess system . 
 two subsequent numbers of instruments of each rotary system were used and the final working length was recorded . 
 a digital calyper was used to record the working length , in millimeters , to investigate a possible discrepancy between initial and final measurements.results:analysis of variance ( anova ) na tukey test revealed statistical difference between group 1 and groups 2 and 3 ( p 0.05).conclusions : all groups presented shorter working length after cervical flaring ; groups prepared with instruments la axxes and orifice opener presented the best results among the systems studied . 
Objetivo:
Resultados:
Concluses:
INTRODUCTION
MATERIAL AND METHODS
RESULTS
DISCUSSION
CONCLUSIONS