successful organ preservation is an important component of transplantationand ensures the maintenance of organ viability until implantationinto the recipient .
currently , heart preservation for transplantationis limited to 4 to 6 hours of cold ischemic storage , and longerperiods of ischemia are known to adversely affect survival .
this is in contrast to preservation of the liver , kidney , andpancreas , which have been successfully preserved for 24 to 36hours although graft function may be transiently compromised .
ischemia initiates a complex injury process that is characterized by the loss of high energy phosphates , the accumulation of intracellular inosine and hypoxanthine , cessation of the na - k pump , followed by cell swelling , cytosolic calcium increase , and enzymatic degradation . in general , lower organ temperatures ( 0c to 4c ) maintain high energy phosphates more effectively .
however , temperatures below 2c significantly increase the risk of cold injury with some proteins denaturing below 0c . at higher temperatures ( 8c to 12c ) , organ function is maintained to a greater extent [ 11 , 12 ] .
however , at temperatures above 12c the higher metabolic demand for oxygen leads to irreversible hypoxic injury and thus significantly impairs organ function .
ideally , there should be a temperature range at which these two tendencies can be balanced .
the instructions for use of organ preservation solutions state a temperature range between 2 - 8c or 4 - 8c .
hence , the balance seemingly forms the cornerstone of cold ischemic storage which is hypothermia at 4c to 8c with an appropriate cold storage solution .
hypothermiadecelerates metabolism and the ionic constituents of the storage solution facilitaterapid cessation of electrical activity .
the formulation of the preservation solution is based on three principles : ( a ) hypothermic arrest of metabolism ; ( b ) provision of a physical and biochemical environment that maintains viability of the structural components of the tissue during hypothermic metabolic arrest ; and ( c ) minimization of the effects of reperfusion injury .  
hypothermia does not stop metabolism but it slows down biochemicalreaction rates and decreases the rate at which intracellularenzymes degrade essential cellular components necessary fororgan viability .
most enzymes of hypothermic animals showa 1.5 to 2.0-fold decrease in activity for every 10c decreasein temperature .
hypothermia also retards lysis of organelles like lysosomesthat , in turn , release autolytic enzymes that cause cell death . 
the current gold standard for transporting a donor organ to a recipient for transplant involves storing the organ in organ preservation fluid within sequential plastic bags that are placed on ice and transported in a cooler .
some of the potential issues associated with this method of storage include ( a ) variability of packaging materials used ; ( b ) bioincompatibility of the material that touches the organ ; ( c ) tissue injury due to movement / vibration ; and ( d ) potential uneven cooling of the organ tissue . in a multicenter study of 186 organs , recommended packaging guidelines were tested for their ability to maintain organ temperatures at acceptable temperatures .
paragonix technologies , inc . has developed a series of medical devices intended for transport of donor hearts and kidneys to recipients for implant : the sherpa pak  cardiac transport system ( cts ) and the sherpa pak  kidney transport system ( kts ) . 
the sherpa pak  system ( figures 1 and 2 ) is intended to provide a safe , consistent method for cold ischemic storage and transport of donor organs .
a , b , c ) future product design of sherpa pak cts and kts .
the systems are designed to overcome many of the potential issues associated with the current systems used for transport .  
the sherpa pak s cooling mechanism is based on phase - change material ( pcm ) , which is a substance with a high heat of fusion which is capable of storing and releasing large amounts of energy .
the sherpa pak s pcm panels are designed to hold 5c longer than conventional phase change material cold packs ( which undergo phase change at 0c , and have little heat capacity at 5c ) .
in addition , the pcm can be placed in direct contact with temperature sensitive products because there is no risk of freezing ( please note , however , that the cold panels do not come in direct contact with the donor organ ) .
management of preservation in the sherpa pak system is the same for both kidneys and hearts other than that the heart is connected to a connector in the inner hard shell assembly and the kidney is not .
we have performed multiple temperature profile studies for varying periods of time , from 4 hours up to 30 hours of storage ( tables 1 and 2 ) . preconditioning protocol for components of the sherpa pak systems .
performance data for static hypothermic storage between 4 and 8c for up to 30 hours using the sherpa pak technology at ambient temperature .
all protocols were in conformity with the guide for the care and use of laboratory animals published by the national institutes of health 86 - 23 , revised in 1985 .
anesthesia was induced with 4.4 mg / kg telazol , 2.2 mg / kg xylazine and 0.04 mg / kg atropine intramuscularly and maintained with isoflurane after endotracheal intubation .
median sternotomy was performed and heparin ( 300 units / kg ) was given intravenously .
an aortic root cannula was inserted and , after cross - clamping the ascending aorta , 1 l of celsior ( genzyme , cambridge , ma , usa ) cardioplegia was administered through the cannula .
the heart was vented through the left atrial appendage , and superior and inferior vena cava , and donor cardiectomy was performed .
complete sherpa pak cardiac transport systems ( cts ) and sherpa pak kidney transport systems ( kts ) were utilized for the thermal qualification test . prior to initiating the testing
, the components of the sherpa pak systems were preconditioned according to the instructions for use provided with the device ( table 1 ) .
the sherpa pak  system consists of multiple components : 1 ) an outer transport shipper which contains various non - ice - based temperature controlled packaging elements ; 2 ) an inner and outer hard shell assembly ( i.e. sherpa pak/sherpa pak  shell ) which provides a double , rigid barrier shipper in which the organ is immersed and suspended in a cold storage fluid cleared for use in storing and transporting donor organs ; 3 ) an off - the - shelf temperature data logger that monitors and displays the temperature of the organ during transport ; and 4 ) an off - the - shelf timer that displays elapsed time during transport .
once conditioned , the systems were assembled in accordance with the instructions for use with the exception that additional thermocouples were placed to monitor temperature at various locations within the system .
these thermocouples were connected to an external display ( the presence of the thermocouples did not alter the closure of the device ) .
temperature readings were monitored and recorded at regular intervals for the period of the experiment .
temperature measurements were taken every 5 min in studies i.a , i.b and iii.a , and every 10 s in studies ii and iii.b . when not otherwise indicated , the storage solution was saline or water . in order to capture any variations in temperature within the device , thermocouples were placed in various locations of the device .
thermocouple locations are designated as follows : center1 = one probe ( n=1 ) located in the center of the fluid filled sherpa pak. wetted probe = one probe ( n=1 ) located at the top of the fluid filled sherpa pak ( within fluid ) .
this assembly was then placed into a third solution - filled bag and placed onto crushed ice in an ice chest . when not otherwise indicated , the storage solution was saline or water . study design . 
three different ex - vivo studies have been performed with the paragonix sherpa pak  devices : study i : the temperature of the fluid filled device was measured for up to 30 h at an outside temperature set at 22c ( study i.a and study i.b , table 2).study ii : the temperature of the fluid filled device was measured for up to 30 h at extreme outside temperatures set at -8c and 31c ( study ii , table 3).study iii : the temperature of pig hearts attached to the device was measured for up to 12 h ( study iii.a and iii.b ; table 4 ) .
study i : the temperature of the fluid filled device was measured for up to 30 h at an outside temperature set at 22c ( study i.a and study i.b , table 2 ) .
study ii : the temperature of the fluid filled device was measured for up to 30 h at extreme outside temperatures set at -8c and 31c ( study ii , table 3 ) . study iii : the temperature of pig hearts attached to the device was measured for up to 12 h ( study iii.a and iii.b ; table 4 ) .
performance data for static hypothermic storage between 4 and 8c for up to 30 hours using the sherpa pak technology with two 1 hour excursions to extreme hot and cold temperatures .
summary of animal studies of static hypothermic storage between 4 and 8c using the sherpa pak technology .
sherpa pak  can keep the temperature of the preservation solution between 4 - 8c for a period of up to 30 hours .  the first part of study i ( table 2 ) demonstrated in three separate test runs that all temperatures recorded from the individually placed temperature probes remained below the 8c upper limit of the design specification through four hours .
moreover , all recorded temperatures from all individually placed temperature probes remained below the 8c upper limit through 12 hours of experimentation .
most importantly , the experiment verified that the fluid within the sherpa pak ( in which the donor heart is immersed ) was maintained between 4c and 6c ( as measured by the three probes placed within the fluid ; two at center of the container , one against the side wall ) which is within the accepted 4c - 8c .
the second part of study i ( table 2 ) demonstrated that all temperatures recorded from the individually placed temperature probes remained below the 8c upper limit through 30 hours .
most importantly , the experiment verified that the fluid within the sherpa pak ( in which the donor organ is immersed ) was maintained between 4c and 8c ( as measured by the temperature probes placed within the fluid ) .
sherpa pak  consistently maintains fluid temperature at 4 - 8c despite extreme fluctuations in ambient temperature .
results of the four separate test runs ( two test runs each at cold or hot temperatures ) of the experiment ( study ii : table 3 ) demonstrated that the storage solution remained below the 8c upper limit of the design specification through 24 hours .
interestingly , in the second warm temperature test run , the cold storage solution was n't preconditioned according to its instructions for use and the starting temperature of the fluid was 7c ( versus 6c according to manufacturer ifus ) .
even with this slightly elevated temperature , the sherpa pak system was capable of maintaining temperature throughout the 24 hour period .
sherpa pak  can maintain the temperature at 4 - 8c when the system is loaded with a pig heart .
both the sherpa pak  cts as well as the conventional ice storage demonstrated their ability to cool and maintain temperature throughout the experimental period .
however , the  ice system  cooled the heart based on the two thermocouples to temperatures as low as 0c at the apex closest to the ice , and < 1c inside the left ventricle of the pig heart .
the experiment verified that the fluid within the sherpa pak ( in which the donor organ is immersed ) was maintained between 4c and 8c ( study iii , table 4 ) .
temperatures were recorded every 5 min , providing up to 145 readings in study iii.a , and every 10 s , providing up to 10801 readings in study iii.b results of 12 temperature test runs during pig heart storage demonstrated that the sherpa pak cts was able to maintain fluid temperatures within a range of 4 - 8c when fully loaded with a pig heart ( table 4 ) .
the purpose of this study was to verify whether the paragonix sherpa pak device was capable of maintaining the desired temperature for organ storage which lies between 4 and 8c .
several studies were performed during which all system components including the cold storage solution were preconditioned according to the instructions for use .   in certain studies ,
exterior conditions remained constant as the system will always be under the direct supervision of medical personnel during transport and not anticipated to be exposed to uncontrolled temperatures or environmental conditions .
the purpose of study ii , in which exterior conditions were changed to either high temperatures ( 31c ) or low temperatures ( -8c ) for 1 hour twice during 30 hours of monitoring , was to challenge the system at two extreme temperatures .
the results of the testing verified that the design of the sherpa system was able to maintain the fluid in which the donor organ is contained within the clinically acceptable range of 4c - 8c through 24 hours , even when challenged to high and low temperatures . when ambient temperatures were kept constant ( at 22c ) the system was able to maintain temperatures within a range of 4c - 8c throughout 30 hours of monitoring .
based upon the results of the testing , we conclude that the sherpa pak devices are designed such that they can reliably maintain a clinically acceptable storage temperature for the donor organ .
it is commonly accepted that the 4c - 8c temperature range is the best balance for preservation of high energy phosphate stores , minimization of cold injury , and preservation of post - transplant function .
although temperatures below 4c offer superior maintenance of high energy phosphates , temperatures below 2c significantly increase the risk of cold injury and frostbite [ 8 , 9 ] .
the results of the studies demonstrated that the paragonix sherpa pak device was capable of maintaining a very consistent temperature within the 4c - 8c temperature range through twelve hours .
the conventional ice storage was also very effective at cooling and maintaining temperature of the heart through six hours ; however , the data show that temperatures encountered with this system approached 0c for the last few hours of the experiment , thus increasing the potential for issues with the donor heart .
these data compared well to a multicenter study on transport temperature of organs . studying 186 organs ,
the average organ temperature during transportation was below 2c , and after 6 hours below 0c .
based upon these test results , it appears that the paragonix sherpa pak transport system is effective at maintaining the temperature of the organ in the optimal temperature range . while the study investigated ambient temperatures that might be expected in a clinical situation and temperature excursion that may occur during organ transport
, a limitation of the study is that measurements were taken in a research setting , which may not entirely reflect the clinical setting .
the paragonix sherpa pak device may decrease cold injury of donor organs by maintaining the temperature consistently between 4c and 8c and therefore may decrease primary graft failure after organ transplantation .
 introductioncurrently , the gold standard for donor organ preservation in clinical organ transplantation consists of 3 plastic bags and an ice box . 
 the first plastic bag includes the organ itself immersed in preservation solution ( e.g. celsior ) . 
 this bag is put in a second bag filled with saline , and then these two are put in a third bag filled with saline which is then put in the ice box .   
 the disadvantage of this method is that the organ usually gets too cold . 
 it has been shown that the theoretical perfect temperature for organ preservation is 4c - 8c . 
 while higher temperatures lead to hypoxic injury of the organ because the metabolism is not decreased efficiently , lower temperatures than 4c increase the risk of cold injury with protein denaturation . in the current study , we investigated a device that keeps the organ temperature consistently in the desired range of 4c - 8c and can potentially decrease cold injury to donor organs.methodsthree different ex vivo studies were performed with the paragonix sherpa pak devices : 1 ) the temperature of the fluid - filled device was measured for up to 30 hours at an outside temperature set at 22c , 2 ) the temperature of the fluid - filled device was measured for up to 30 hours at extreme outside temperatures set at -8c and 31c , 3 ) the temperature of a pig heart attached to the device was measured up to 12 hours.resultsall studies showed that the paragonix sherpa pak can keep the temperature of the heart consistently between 4 and 8c.conclusionsthe paragonix sherpa pak device may decrease cold injury of donor organs by maintaining the temperature consistently between 4c and 8c and therefore may decrease primary graft failure after organ transplantation . 
Introduction
Methods
Results
Discussion
Conclusion