The following are additional guidance specific for each technique: For fully automated office oscillometric manometer readings: Preprogram the manometer to wait five minutes before inflation begins and to take and average 2-3 readings at least 30 seconds apart; the most common intervals are 30 seconds, one minute, and two minutes Position the patient and place the proper sized cuff on the upper arm before initiating the wait time and blood pressure readings Turn the manometer on and set controls to take inflate to proper level (above sensed SBP) automatically and measure and average multiple readings; press the button to initiate wait period and automated readings (fully automated) Patient should remain quiet and not use electronic devices (e.g., phones) during the rest period and readings; person measuring the blood pressure may remain in the room (attended) or leave the patient alone in the room (unattended), but no one should interact or speak with the patient during the rest period and readings After the rest and measurements are completed, record the average blood pressure reading displayed; manometer should display the average of the 2-3 readings (as preset) For standard technique with automated oscillometric device: Use an upper-arm cuff oscillometric device that has been validated Position the patient and place the proper sized cuff on the upper arm before initiating the wait time and blood pressure readings Turn the manometer on and set controls to take/inflate to proper level (above sensed SBP) automatically Patient should remain quiet and not use electronic devices (e.g.