ffp standards ffp standards ffp standards refer to the filtering half mask classification by en 149, a european standard of testing and marking requirements for filtering half masks. ffp standard masks (where ffp stands for filtering facepiece) cover the nose, mouth and chin and may have inhalation and/or exhalation valves. this article is about the european standard only. for information on mechanical air filters in general, and the types of respirators which meet the en 149 standard, see mechanical filter respirator. for more detailed information on items made to the en 149 standard, see filtering facepiece respirator and elastomeric respirator. en 149 defines three classes of such particle half masks, called ffp1, ffp2 and ffp3, according to their filtering efficiency. it also classifies masks into "single shift use only" (not re-usable, marked nr) or "re-usable (more than one shift)" (marked r), while an additional marking letter d indicates that a mask has passed an optional clogging test using dolomite dust. such mechanical filter respirators protect against the inhalation of particulates such as dust particles, droplets, and aerosols. almost identical tests (but different markings) are used in australia, new zealand, korea and brazil. similar standards are used in the united states, china and japan. for example, en 149 ffp2 masks have similar performance requirements to n95 masks in the united states and kn95 filters of china, and en 149 ffp3 masks have similar performance requirements to n99 masks in the united states. however en 149 test requirements differ somewhat from the u.s./chinese/japanese standards: en 149 requires an additional paraffin oil (paraffinum perliquidum) aerosol test and it tests at a range of different flow rates and defines several associated and permissible pressure drop levels. classification the en 149 standard defines performance requirements for three classes of particle-filtering half masks: ffp1, ffp2 and ffp3. the protection provided by an ffp2 (or ffp3) mask includes the protection provided by a mask of the lower-numbered classes. a mask conforming to the standard must have its class written on it, along with the name of the standard and its year of publication, as well as any applicable option codes, e.g. “en 149:2001 ffp1 nr d”. some manufacturers use in addition the colour of the elastic band to identify the mask class, however, the en 149 standard does not specify any such colour coding and different manufacturers have used different colour schemes. class filter penetration limit (at 95 l/min air flow) inward leakage typical elastic band ffp1 filters at least 80% of airborne particles <22% yellow ffp2 filters at least 94% of airborne particles <8% blue or white ffp3 filters at least 99% of airborne particles <2% red ffp1 mask ffp1 face mask it is the least filtering mask of the three. aerosol filtration percentage: 80% minimum internal leak rate: maximum 22% it is mainly used as a dust mask (for example for diy jobs). dust can cause lung diseases, such as silicosis, anthracosis, siderosis and asbestosis (in particular dust from silica, coal, iron ore, zinc, aluminium or cement are common particulate risks). ffp2 mask ffp2 face masks with and without exhalation valve aerosol filtration percentage: 94% minimum internal leak rate: maximum 8% this mask offers protection in various areas such as the glass industry, foundry, construction, pharmaceutical industry and agriculture. it effectively stops powdered chemicals. this mask can also serve as protection against respiratory viruses such as avian influenza or severe acute respiratory syndrome associated with the coronavirus (sars), as well as against the bacteria of pneumonic plague and tuberculosis. it is similar to the us-standard n95 respirator. ffp3 mask ffp3 face mask aerosol filtration percentage: 99% minimum internal leak rate: maximum 2% the ffp3 mask is the most filtering of the ffp masks. it protects against very fine particles such as asbestos and ceramic. it does not protect against gases and in particular oxides of nitrogen. requirements en 149 defines laboratory tests, field tests and certain requirements to ensure the conformity of the masks. the following points are analyzed: packaging materials: resistance to manipulation practical performance test leakage: total leakage inwards and penetration of the filter material there are some european organizations that issue an examination certificate confirming conformity and specifying the characteristics of the products: inrs then apave in france inspec in great britain fachausschuss in germany institute for occupational safety and health of the german social accident insurance in germany ciop-pib in poland 2009 version with the publication of the 2009 version of the standard, the designation of the respiratory protection mask is now "particulate filtering half mask". the abbreviation nr or r is added after ffp1, ffp2, ffp3: nr (not reusable): if the use of the filtering half-face mask is limited to one working day. it is not reusable. r (reusable): if the filtering half-face mask can be used for more than one working day, so it is reusable. additional suffixes include: anti-clogging nr masks (d). when the half mask has passed the dolomite dust test, a letter d can be added to indicate that the service life may exceed 8 h. example: ffp3 nr d. valve. the presence of a valve can be indicated by the letter v. particulate type. the letters s or l respectively specify the filtration of solid dust (nacl-only) or liquid mist (paraffin oil). example: ffp3 slv. masks manufactured before the new standard was taken into account may still have the old marking. marking ffp respirators are a type of personal protective equipment (ppe). here is the notice that should appear on every mask : manufacturer's name mask ce number of the certification body (only ffp3) + en 149: 2009 + the mask class (ffp1, ffp2 or ffp3) + acronym (nr or r) (note: in case of ffp1 the conformity assessment procedures are done by the manufacturer itself as per annex iv (module a) the marking must comply with european union directive 89/686 / eec on ppe. if any of these entries are missing, the mask will be considered non-compliant. medical use en 149 tests the ability of masks to protect the wearers against inhaling liquid and dry aerosols. it makes no statement about, and does not specifically test the suitability of such masks for, infection control against airborne transmission of pathogens through respiratory droplets. nevertheless, ffp2 and ffp3 masks are commonly used for this purpose in medical practice. a researcher at the mrc biostatistics unit at cambridge university claimed in 2021 that switching to ffp3 in addenbrooke hospital in cambridge may have cut ward-based covid infection of staff to zero. similar standards main article: mechanical filter respirator several regions use standards based on nearly identical tests and thresholds as those in en 149, but with different markings: russia (gost r 12.4.191-2011): identical. south korea (kmoel - 2017–64): considered identical to pre-2009 version. ffp1 is called "2nd grade" or kf80, ffp2 "1st grade" or kf94, and ffp3 "special grade" or kf99. australia and new zealand (as/nz 1716:2012): similar grades with a different test agent. grades are written simply "p". brazil (abnt/nbr 13698:2011): identical to pre-2009 version. grades are written as "pff". other regions use similar tests that (in parts) resemble more closely the 42 cfr 84 requirements in the united states: japan (jmhlw-notification 214, 2018): similar grades with a different coding scheme for nr/r and s/l types. written with a two-letter prefix d/r and s/l that maps to nr/r and s/l respectively. leakage requirement not present. taiwan (cns 14755): d1/d2/d3 grades for 80/95/99 efficiency. no requirements for inward leakage.