targeted reinnervation targeted reinnervation targeted reinnervation enables amputees to control motorized prosthetic devices and to regain sensory feedback. the method was developed by dr. todd kuiken at northwestern university and rehabilitation institute of chicago and dr. gregory dumanian at northwestern university division of plastic surgery. overview targeted reinnervation has an efferent and an afferent component. targeted muscle reinnervation is a method by which a spare muscle (the target muscle) of an amputated patient is denervated (its original nerves cut and/or de-activated), then reinnervated with residual nerves of the amputated limb. the resultant emg signals of the targeted muscle now represent the motor commands to the missing limb, and are used to drive a motorized prosthetic device. targeted sensory reinnervation is a method by which skin near or over the targeted muscle is denervated, then reinnervated with afferent fibers of the remaining hand nerves. therefore, when this piece of skin is touched, it provides the amputee with a sense of the missing arm or hand being touched. motivation several methods exist that seek to achieve advanced control of motorized neural prosthetics. chronic brain implants record neuronal signals from the motor cortex, while methods such as eeg and fmri obtain motor commands non-invasively. the recorded signals are decoded into electrical signals, and input into assistive devices or motorized prosthetics. traditional myoelectric prostheses utilize surface emg signals from the remains of the amputated limb. for example, a patient may flex a shoulder muscle in order to generate emg signals that may be used to send “bend elbow” command to the prosthesis. however, there are shortcomings to all of these methods. chronic implants fail over a period of time because neuronal signal degrade due to tissue immune response to foreign bodies. eeg and fmri do not obtain as strong signals as direct electrode implant. traditional myoelectric prostheses are unable to provide multiple control signals simultaneously, thus only one action can be performed at a time. they are also unnatural to use because the users have to use muscles (such as shoulder) that are not normally involved with lower arm functions to control lower arm functions (such as opening and closing hands). the solution to these problems could include a completely different concept of neural interface. advantages targeted reinnervation does not require any implants. therefore, it does not have the issue of tissue foreign body response as chronic brain implant technology does. the targeted muscle acts as a natural amplifier for the neuronal signals produced by the transferred residual nerves. this is an advantage over technologies like eeg and fmri that utilize weaker signals. with targeted reinnervation, multiple yet independent emg signals can be produced, thus multiple functions of the artificial limb can be controlled simultaneously. for example, the patient would be able to perform actions such as throwing a ball relatively gracefully, exhibiting simultaneous control of elbow and hand. the control is also intuitive to the patient because the emg signals are generated by transferred residual limb nerves, unlike traditional myoelectric prosthetics where emg signals have to be generated by muscles normally not involved in arm or wrist functions. also, existing commercially available myoelectric prostheses, such as powered wrists, elbows can be used. there is no need to develop specific prostheses for targeted reinnervation. by means of nerve transfer, targeted reinnervation can also provide sensory feedback, which has not been achieved by any other form of prosthetics aforementioned. methods targeted muscle reinnervation the goal of targeted muscle reinnervation is to transfer multiple nerves into separate regions of the targeted muscle, record multiple yet independent signals from the muscle regions, and to use the emg signals to control a motorized prosthesis sophisticated enough to process multiple control signals. surgical procedure the requirement to transplant multiple nerves into a muscle region originated from a hypothesis that hyper-reinnervation, by which an excessive amount of motor neurons transferred to a muscle, can increase the reinnervation of muscle fibers hence improving the recovery of paralyzed muscles. the hypothesis was tested on rat skeletal muscles and the result indicated that hyper-reinnervated muscles recovered more muscle mass and strength and more number of motor units were formed. the first surgical patient was a bilateral shoulder disarticulation amputee. both arms were entirely amputated at the shoulder level, with only the shoulder blades remaining. the pectoral muscles were chosen targets because they were close to the shoulder, and they were also biologically non-functional due to detachment from the amputated arm. the pectoral muscles were first denervated by cutting the original nerves that innervate them. the proximal ends of the original nerves were ligated to prevent them from reinnervating the pectoral muscle. then the remnant arm nerves (brachial plexus) were transferred into the pectoral muscles. the musculocutaneous nerve was transferred to the clavicular head of the pectoralis major muscle; the median nerve was transferred to the upper sternal of the pectoralis major muscle; the radial nerve was transferred to the lower sternal head of the pectoralis major muscle. the pectoralis minor muscle was translocated from under the pectorialis major muscle to the lateral chest wall, so that its emg signals would not interfere with those of the pectoralis major muscle, and it is also a fourth muscle target. the ulnar nerve was then transferred to the moved pectoralis minor muscle. the musculocutaneous, median, radial, and ulnar nerves (brachial plexus) were sewn onto the distal ends of the original pectoral muscle nerve fascicles and onto the muscle itself. subcutaneous fat over the pectoral muscle was removed so that the electrodes can be as close to the muscle as possible to obtain optimal emg signals. post-surgical training about 3 months after surgery, the patient had the first twitch in pectoral muscle when he attempted to bend his phantom elbow. five months past surgery, he was able to contract four regions of pectoralis major muscle by attempting different moves. for example, when the patient attempted to bend his elbow, the muscle region beneath the clavicle contracted strongly. this was an indication of successful musculocutaneous nerve transfer because musculocutaneous nerve innervates biceps. the patient was then soon put to a training session and a testing session. during training session, the patient was sitting in an upright position and shown each of the 27 normal movements (such as shoulder adduction/abduction, hand open/close, elbow flexion/extension etc.) on a video. after each demonstration, the patient followed the movement 10 times, exerting a moderate force, held for 2.5 seconds. the patient was given 5 seconds of rest after each attempt. during the test session, the patient performed 5 sets of the 27 movements in random order. he was first shown a video of a movement, then asked to follow the repeated video of the same movement simultaneously after 2 seconds. emg recording and processing a biosemi active ii system (produced by biosemi, amsterdam, netherlands) and a 127-channel electrode array were used to record monopolar emg signals while patient was attempting movements during training and testing sessions. one hundred and fifteen electrodes were used to record emg from the pectoral muscle; two electrodes were used to record from each of the deltoid, latissimus dorsi, supraspinatus, upper trapezius, middle trapezius and lower trapezius muscles. the electrodes were placed at a distance of 15mm from each other. to remove the artifact caused by body movement, the emg signals were preliminarily filtered with a fifth order butterworth high-pass filter set at 5 hz. the major contaminant of the emg signal was the ecg artifact. to remove ecg noise, an ecg template was constructed by averaging ecg complexes recorded when muscles were relaxed. the time between each ecg complex was used to calculate a representative inter-spike-interval. detection of ecg spikes was calculated from the correlations between emg and the ecg template. a threshold was set so that signals exceeding the threshold were marked as possible ecg spikes. the inter-spike-intervals of the possible spikes were then compared to the previously calculated representative inter-spike-interval to determine whether the possible spikes were to be accepted as ecg artifacts. another major task of processing the emg signals is to eliminate crosstalk from other muscles. first, positions of and distance between electrodes are empirically determined to obtain strongest emg thus least crosstalk. setting a threshold above background noise and crosstalk from other muscles also helps eliminating crosstalk. smaller muscle size and subcutaneous fat facilitate crosstalk. with a minimal level of less than 3mm subcutaneous fat, crosstalk is expected to be minimal in an area of 2–3 cm diameter. prosthetic components after surgery, the patient was fitted with his pre-surgery body-powered prosthesis on the right side and an experimental myoelectric prosthesis consisted of a griefer terminal device, a power wrist rotator, a boston digital arm, and an lti-collier shoulder joint on the left side. three strongest emg signals were chosen from the successful nerve transfers: the musculocutaneous nerve, the median nerve and the radial nerve. the emg resulting from contraction of muscle reinnvervated by median nerve was used to control hand closing movement; the emg from musculocutaneous nerve was used to control elbow bending; the emg from radial nerve was used to control wrist rotation and flexion. the performances of these two prostheses were compared with a box-and-blocks test, where the patient was allowed 2 minutes to move one-inch cubes from one box to another, over a short wall. the result was quantified by the total number of blocks moved. to test the terminal device (“hand”), elbow and wrist rotator, the patient was administered a clothes-pin test, where he was asked to pick up clothes pins from a horizontal bar, rotate them, then put them on a higher vertical bar. the time used to move 3 clothes-pins was recorded. both tests were repeated 3 times. the quantified results showed that the myoelectric prosthesis performed 246% better (moved 2.46 times more blocks) in box-and-blocks test, and 26.3% better (used 26.3% less time in moving clothes pins) in clothes-pin test. an experimental six-motor prosthesis was also constructed. the most striking feature of targeted reinnervation compared to traditional myoelectric prosthetics is its ability to provide multiple signals to control multiple functions simultaneously. although current myoelectric prostheses can be used directly, they are designed and aimed at traditional myoelectric control. thus, the only commercially available prosthesis only has powered terminal device (often a hook), wrist rotation and powered elbow. to fully utilize the multiple signals provided by targeted reinnervation, an experimental prosthesis was constructed with added power components: a touchemas shoulder, a humeral rotator, and a hand capable of opening and closing with wrist flexion/extension function. the elbow and hand functions were driven by four nerve transfer signals, and the humeral rotation was driven by emg from latissumus dorsi and deltoids. with this six-motor prosthesis, the patient could control multiple joints at the same time and perform new tasks that could not be accomplished with other prostheses, such as reaching out to pick up objects and putting on a hat. discovery targeted sensory reinnervation was discovered by accident. while receiving an alcohol rub on his chest after the surgery, the patient described a sensation of being touched on the pinky. the explanation for this phenomenon is that, since his subcutaneous fat was removed during surgery, his chest skin was denervated. thus, the afferent nerve fibers regenerated through the pectoral muscle, reinnervating the skin over the muscle. since then, areas of the pectoral muscle have been mapped to parts of arm and hand according to patient's description of touch sensations he felt. when touched in a specific region on the pectoral muscle, the patient would describe where in the phantom limb he felt as being touched. for example, when touched in a region immediately above the nipple, he felt as if his anterior forearm was being touched. surgical procedure with this discovery, the team set out to perform nerve transfer surgery specifically aimed to reinnervate sensory feedback. a piece of skin near or over the targeted muscle was denervated, thus the afferent nerve fibers were allowed to reinnervate the skin. in a case of a woman patient with left arm amputation at the humeral neck, the supraclavicular sensory nerve was cut, the proximal end was ligated to prevent regeneration and reinnervation, and the distal end was coapted end-to-side to the ulnar nerve. the intercostobrachial cutaneous nerve was treated with the same method, with the distal end