Results
Bland-Altman plots showed a bias in slope (95% confidence intervals) between ABG and TCM4-measurements of 0.05mmHg/min (-0.05 to 0.15), and limits of agreement were -0.88mmHg/min (-1.06 to -0.70) and 0.98mmHg/min (0.81 to 1.16). Bias between ABG and TCM5 was -0.14mmHg/min (-0.23 to -0.04), and limits of agreement were -0.98mmHg/min (-1.14 to -0.83) and 0.71mmHg/min (0.55 to 0.87). A linear mixed model (predicting the CO