Results
The mean difference between sphygmomanometers was 3.9mmHg for systolic blood pressure (95% Confidence Interval (CI):2.5,5.2) and 1.4mmHg for diastolic blood pressure (95% CI:0.3,2.4), with the Omron HEM-907 measuring higher. For maximum grip strength, the mean difference when either one of the electronic dynamometers was compared with either the hydraulic or spring-gauge device was 4-5kg, with the electronic devices measuring higher. The differences were small when comparing the two electronic devices (difference = 0.3kg, 95% CI:-0.9,1.4), and when comparing the hydraulic and spring-gauge devices (difference = 0.2kg, 95% CI:-0.8,1.3). In all cases limits of agreement were wide. The mean difference in FEV