Results
aphakic, diabetic retinopathy, devicesAxis
DIABETIC RETINOPATHY, ENDOPHTHALMITIS, RETINAL DETACHMENT, VITREOUS HEMORRHAGE, DISEASES
Forty SO-filled aphakic eyes of 40 patients (18 males, accounting for 45%) with an average age of 54.9 ± 10.3 years (range, 29–74 years old) were included. Among the primary diseases, 33 eyes had complicated retinal detachment, 6 eyes had diabetic retinopathy with vitreous hemorrhage, and 1 eye had endophthalmitis. All eyes were filled with SO (Oxane 5700 Bausch & Lomb, Kingston-upon-Thames, UK) by the same specialist (GZ). Table 
Summary of results for the measured variables in both tested devicesAxis of the flattestradius (°)Central cornealthickness (µm)CoV, coefficient of variation
The differences of the means, with their SDs, 95% confidence intervals and paired t test results between the OA-2000 and the IOLMaster700Axis of the flattestradius (°)Central cornealthickness (µm)Difference of the means is the value measured with the OA-2000 minus value measured with the IOLMaster 700LoA, limits of agreementThe mean AL obtained with the OA-2000 was 23.57 ± 0.93 mm (range: 21.50 to 25.68 mm), and that obtained with the IOLMaster 700 was 23.69 ± 0.94 mm (range: 21.85 to 25.86 mm), resulting in a mean offset of 0.124 ± 0.125 mm (Figures 
Bland-Altman plot of the axial length measurements using the OA-2000 against IOLMaster 700. The solid line represents the mean difference. The dotted lines on the side represent the upper and lower 95% limits of agreement
Bland-Altman plot of the flattest keratometry measurements using the OA-2000 against IOLMaster 700. The solid line represents the mean difference. The dotted lines on the side represent the upper and lower 95% limits of agreement
Bland-Altman plot of the steepest keratometry measurements using the OA-2000 against IOLMaster 700. The solid line represents the mean difference. The dotted lines on the side represent the upper and lower 95% limits of agreement
Bland-Altman plot of the axis of the flattest keratometry measurements (Ax1) using the OA-2000 against IOLMaster 700. The solid line represents the mean difference. The dotted lines on the side represent the upper and lower 95% limits of agreement
Bland-Altman plot of the central corneal thickness (CCT) measurements using the OA-2000 against IOLMaster 700. The solid line represents the mean difference. The dotted lines on the side represent the upper and lower 95% limits of agreement