Feasibility and technical aspects
spasticity
Concerning the robotic set up, the positioning of the LBR arm on the cart (Fig. For three patients, the arm weighing process did not function as intended. As a result, the default weight of 0.01 kg was shown by the robot and the weight needed to be estimated and corrected manually. These patients suffered pronounced spasticity in the elbow flexors.