Results
In all, 66 participants with a mean age of 46.4 years (±21.7) were enrolled in the study. All repeatability ratios and intra-device variability were less than 1 (anterior corneal curvature: flat = 0.86, steep = 0.85; posterior corneal curvature: flat = 0.43, steep = 0.61; and map: thinnest = 0.22; central = 0.24, 2 mm = 0.26, 4 mm = 0.27, and 6 mm = 0.30). All reproducibility ratios, which measure the inter-device variability, were less than 1 (anterior corneal curvature: flat = 0.58, steep = 0.73; posterior corneal curvature: flat = 0.25, steep = 0.31; and pachymetry map: thinnest = 0.20; central = 0.20; 2 mm = 0.20; 4 mm = 0.19; and 6 mm = 0.22). A ratio of less than 1 indicates that the CASIA SS-1000 has more consistent measurements.