human
TASK: Your task is to answer biomedical questions using the given abstract. Only output yes, no, or maybe as answer.
INPUT: ['In 1998, Digital Imaging Communications in Medicine (DICOM) proposed a calibration tool, the grayscale standard display function (GSDF), to obtain output consistency of radiographs. To our knowledge, there have been no previous reports of investigating the relation between perceptual linearity and detectability on a calibration curve.', "To determine a suitable calibration curve for diagnostic liquid crystal display (LCD) monitors, the GSDF and Commission Internationale de l'Eclairage (CIE) curves were compared using psychophysical gradient delta and receiver operating characteristic (ROC) analysis for clinical images.", 'We succeeded in expressing visually recognized contrast directly using delta instead of the just noticeable difference (JND) index of the DICOM standard. As a result, we found that the visually recognized contrast at low luminance areas on the LCD monitor calibrated by the CIE curve is higher than that calibrated by the GSDF curve. On the ROC analysis, there was no significant difference in tumor detectability between GSDF and CIE curves for clinical thoracic images. However, the area parameter Az of the CIE curve is superior to that of the GSDF curve. The detectability of tumor shadows in the thoracic region on clinical images using the CIE curve was superior to that using the GSDF curve owing to the high absolute value of delta in the low luminance range.'] Do psychophysical evaluation of calibration curve for diagnostic LCD monitor?
OUTPUT:

agent
yes
TASK: Your task is to answer biomedical questions using the given abstract. Only output yes, no, or maybe as answer.
INPUT: ['In 1998, Digital Imaging Communications in Medicine (DICOM) proposed a calibration tool, the grayscale standard display function (GSDF), to obtain output consistency of radiographs. To our knowledge, there have been no previous reports of investigating the relation between perceptual linearity and detectability on a calibration curve.', "To determine a suitable calibration curve for diagnostic liquid crystal display (LCD) monitors, the GSDF and Commission Internationale de l'Eclairage (CIE) curves were compared using psychophysical gradient delta and receiver operating characteristic (ROC) analysis for clinical images.", 'We succeeded in expressing visually recognized contrast directly using delta instead of the just noticeable difference (JND) index of the DICOM standard. As a result, we found that the visually recognized contrast at low luminance areas on the LCD monitor calibrated by the CIE curve is higher than that calibrated by the GSDF curve. On the ROC analysis, there was no significant difference in tumor detectability between GSDF and CIE curves for clinical thoracic images. However, the area parameter Az of the CIE curve is superior to that of the GSDF curve. The detectability of tumor shadows in the thoracic region on clinical images using the CIE curve was superior to that using the GSDF curve owing to the high absolute value of delta in the low luminance range.'] Do psychophysical evaluation of calibration curve for diagnostic LCD monitor?
OUTPUT:
