the subject invention results from experimental evidence proving that there is a direct correlation between the percent composite cure of a light curable resin material and the degree of light transmission through the material as measured by the output of the light detector cell . to substantiate this correlation , the output voltage of a light detector cell in a conventional radiometer was measured based solely upon exposing the detector cell to light . fig1 is a graph showing the relationship between the output of a standard light detector cell with time over two sequential 40 second exposures spaced 30 seconds apart . as can be seen from this graph , whenever the detector cell is subjected to the light source under the same conditions by e . g . turning the light source on and off , a voltage ( in millivolts ) is generated by the detector cell with the output over each consecutive time period being essentially identical to one another within very dose tolerances . in the tests performed , the differences between the first and second light exposures was only about 0 . 75 millivolts , and for purposes of the present invention , may be deemed insignificant . fig2 shows the real - time curing profile of a sample of light curable material exposed to a light source ( obtained using ftir spectroscopy ) indicating the change in composite cure of the material with respect to time . as noted previously , 2 successive 40 - second exposures were given , 30 seconds apart as was the case in fig1 as well . note the plateau for maximal exposure of this sample of material is reached in approximately 125 seconds , and that no additional curing occurs from the second exposure , but cure increases slowly as a result from the previous exposure ( a finding well established in the literature ). fig3 shows the change in light detector response as measured by millivoltage response when a sample of uncured composite ( of similar thickness to that as the specimen used for fig2 ) as it is exposed to 2 sequential 40 second exposures spaced 30 seconds apart . the same light source was used for both test conditions . note that during the first exposure , light transmission increases with exposure time resulting in a plateau . upon second exposure , no obvious trend is seen to relate the second exposure with a change in detector output other than that which would be a slow , continuous increase resulting from the first exposure . upon overlapping fig2 and fig3 on the same time scale ( fig4 ), it becomes evident that extent of cure of the light curable material and the change in light transmission , noted as an increase in detector millivoltage response , appear to be related and follow one another very closely with respect to time . although a slight deviation from one wave to the other ( 0 . 75 mv ) was noted during the first two exposures in fig1 , the graph of fig4 shows that light transmission and conversion are essentially identical at the end of the second exposure . accordingly , this deviation can be ignored . eliminating time as a separate axis and correlating composite cure with voltage output at similar time points is shown in fig5 , based upon measured data points taken at fixed intervals in time e . g . once per second . as millivoltage generation increases , the percentage of composite cure also increases and they tended to accumulate in the same portion of the graph , which indicates the existence of a point of diminishing return for either parameter . stated otherwise , the data points tend to accumulate on the right side of the graph corresponding to where composite cure and millivoltage are maximal . this finding indicates that the effect of further light exposure would be insignificant . a curve can be mathematically derived that simulates this relationship based on measured data from which a mathematical model can be predicated with millivoltage generation predicting the extent of composite cure . these results show that an algorithm can be written with an accuracy of up to 99 . 5 %, showing the change of light transmission can be used to accurately predict when the composite cure value will reach a plateau with respect to exposure duration . in this way , the time it takes for any specific light curing composite to approach maximum monomer conversion can be accurately determined . the relationship shown for this specific example in fig5 shows the shape of the algorithm , y =− 15 . 368x 2 + 127 . 273x − 213 . 217 having a coefficient of correlation r 2 of 0 . 995 , where y = percent composite cure and x = detector millivoltage output . the coefficient of correlation , r 2 , is a number between 0 to 1 , with 0 indicating absolutely no correlation between factors , and 1 representing complete correlation . thus , the observed correlation of 0 . 995 shows a great predictability of millivoltage change being an indicator of the level of composite cure . the value actually indicates that of 100 % of the variability seen in the data , the predicted model can explain 99 . 5 %, leaving only 0 . 5 % attributed to unexplained error . the present relationship uses a second order polynomial to describe the correlation between change in optical density ( represented by change in detector millivoltage response ) and change in composite cure ( extent of polymerization ). thus , a change in light detector output can accurately predict a level of composite cure . other mathematical algorithms may be applied as well , such as higher degree polynomial , logarithmic , exponential , power , or a combination of these functions as is well known to those skilled in the art . the radiometer 10 of the present invention is shown in both fig6 and 7 with fig7 representing a block diagram of the internal electronic components of the radiometer 10 . accordingly , the radiometer 10 comprises a conventional detector cell 11 which may represent any conventional light sensor such as a silicon or selenium detector cell for providing either an output voltage or a change in electrical resistance in direct response to the degree of light exposure . in addition the radiometer 10 further comprises a micro - controller ( microprocessor ) 12 , battery 13 , serial input / output port 14 , lcd display 1 , an on / off function switch 2 , a scroll switch 3 , and a mode switch . the function switch 3 permits the radiometer to be scrolled to perform either an “ optical conversion ” function mode , “ power ” function mode , “ energy function ” mode , or a “ calibration ” function mode . when scrolled to optical conversion function mode , the lcd display 1 provides a time display output in seconds that will indicate the shortest exposure time to provide maximal composite cure for a test sample of uncured composite using any type light source as explained hereafter . optical conversion mode : any conventional light curing source ( not shown ) having e . g ., a standard light guide 9 may be used to cure a sample of an uncured dental composite 6 . the thickness of the sample of uncured composite 6 is adjusted by use of different thickness sample holders 7 with each sample holder 7 having a thickness corresponding to a typical depth of a dental restoration . the sample of composite material 6 is placed in a sample holder 7 of appropriate thickness for a given restoration . it is held by grip detail 8 as shown in fig6 c and inserted along a groove or track 5 ( fig6 a ) so that the sample sits directly over the detector window 4 of the light sensor 11 , which is shown in fig7 . the light guide 9 is placed over the sample in line with the detector window 4 so that light may be shined through the uncured dental composite sample 6 . the function switch 3 is then scrolled to “ optical conversion ” mode of operation . the display i will then display time in seconds needed to maximally cure the composite , i . e ., will stop when the display shows a time corresponding to the exposure duration needed to achieve the composite cure for the sample composite that represents a time when the sample is cured in accordance with the algorithm used in programming the micro - controller 12 . in accordance with the present invention , the micro - controller 12 is programmed using an algorithm such as the one explained earlier . the degree of composite cure measured can be determined to be anywhere between 80 % and 99 . 5 % of maximum . it should be understood that , for most composite resin materials , no matter how long the material is exposed to light , the extent of composite cure will plateau at between 45 % to 65 % of the maximum cure value ( 100 %) for that material , and generally at about 50 % as evident from fig2 . thus , for example , using the 2 % preferred change as the basis upon which the micro - controller 12 is programmed , when the display times out , the sample has cured to 98 % of its maximum achievable value . the micro - controller 12 sampling rate is 0 . 1 seconds or less to insure accuracy . power mode : when the function switch 3 is scrolled to “ power mode ” the radiometer 10 will measure the curing light output intensity in watt / cm 2 or milliwatt / cm 2 and the display 1 for this mode of operation is programmed to update for as long as the push button is held . when the push button is released , the radiometer will continue to measure the curing light output intensity but the display will correspond only to peak measurements . energy mode : when the function switch ( 3 ) is scrolled to “ energy mode ”, a momentary push of the function button will set the energy measured in joules or millijoules to zero ( start ) and begin to accumulate values once the intensity is above a preset level . the term “ energy ” is the mathematical product of the power density ( measured in w / cm 2 or mw / cm 2 ) times the exposure duration ( seconds ). thus , as a light exposure continues over time , the accumulated energy delivered to the target also increases and is thus measured by the instrument . the on button ( 2 ) turns the radiometer unit on and it will remain on for two minutes if not used and then will automatically power down to conserve battery life . calibration mode : the radiometer is calibrated at the factory by using a standard lamp and a plastic filter with the same optical transmission characteristics as that of well polymerized dental composite . the user can then compare the exposure time displayed using the calibration filter and the light unit being tested . comparing the standard reading and the actual value will indicate the offset to which the unit is out of calibration . an auto ranging feature of the micro controller will adjust this offset to zero by holding down the on switch ( 2 ) ( in the optical conversion mode ) and the function switch ( 3 ) simultaneously for two seconds ). lcd display : this panel will display real - time light intensity ( power density ), accumulated light energy delivered , or recommended exposure time depending on the mode of operation . the light sensor is a solid - state photo detector with 400 to 500 nm sensitivity , but other ranges such as 300 - 400 nm are possible to measure the intensity of the light coming through the dental composite . mode switch : this switch will allow scrolling through the functions of optical conversion , power and energy . function / calibration switch : this switch is used to calibrate the radiometer using a standard plastic filter as described previously . serial i / o port : this port is configured as rs232c and will allow two - way communication between the radiometer and a computer or remote display . a “ blue tooth ” or usb port can also be used . battery : two alkaline , lithium or rechargeable batteries power the radiometer . either button can be pushed to turn the radiometer on , it will remain on for two minutes after the last button is pushed , and then , for battery life conservation the radiometer will go into a “ sleep ” mode . low battery indication is evidenced by flashing the display . micro controller the radiometer programs are controlled by a microprocessor . inputs include measurement of light , reading mode and function switches . outputs include rs232c or usb communication and display drivers . the plastic filter is designed to simulate the light transmission characteristics of a well - cured dental composite restoration and can be used for calibration . it may come in four different depths ( i . e . 2 , 3 , 4 & amp ; 6 mm ) or any depth that is desired . the plastic is selected from a group of plastic materials that have optical transmission characteristics identical to that of a well cured dental restorative material of a given thickness .