the figures are described in general and the attached list of components forms part of the description . additional benefits and features of the invention , as well as additional preferred formations , arise from the description of the figures . the functionality and design of the preferred sample design can be seen in the block diagram in fig4 : on the input side you will see the gas connections — 1 c compressor input , 1 a compressed air input , 1 b ventilation gas input , e . g . oxygen input and 27 therapy gas input , e . g . for no . the inputs are sighted on a block 12 , which is found within the housing 7 . comprising compressor input 1 c leads to a compressor 6 , which is connected via a flexible pipe 10 with a rigid pipe 11 a , which forms a compressed air pipe 4 to a gas supply container 8 . pipe 11 a is interrupted by a valve 2 a , which either feeds or blocks the compressed air from the compressor or from the compressed air connection 1 a to the gas supply container 8 . a similar ventilation gas valve 2 b feeds or blocks the ventilation gas passage to pipe 11 a from ventilation gas connection 1 b ( in this case oxygen , for example ). parallel to the pipe 11 a a rigid pipe 11 c , is fed through block 12 or is designed in the block wall in particular . pipe 11 c connects the therapy gas connection 27 to the therapy gas output 28 . pipe 11 c is divided by a controllable therapy gas valve 25 . this design was preferable to the known superstructures , where the therapy gas pipes and therapy gas control valves were housed separately to the ventilator and therefore needed additional expenditure on equipment . the advantages over the known design include not only a reduced amount of housing and reduced structural volume , but it is also simpler to operate and clearer in use . the gas supply container 8 is equipped with an oxygen sensor 26 . on the output side it is connected to the ventilator tube 5 via a rigid pipe 11 b . the rigid pipe 11 b has a controllable inspiration valve 2 c . the pressure and / or flow is measured before as well as after the gas supply container 8 via integrated sensors 22 a and 22 b . a proximal flow sensor 23 permits measurement of the patient &# 39 ; s own breathing performance . to administer medication and humidity a nebulizer pipe 36 is provided , which has a nebulizer valve 34 and nebulizer chamber 37 . various check valves 24 prevent loss of pressure if the connection tubes are disconnected . an expiration valve 2 d terminates the expiration pipe 29 . a ventilation tube 5 , as well as a therapy gas output 28 if needed , run to the patient parallel to the expiration pipe 29 . the housing 7 also contains a control system or electronics 3 , which are symbolically depicted and are especially connected to the electronically controlled valves and sensors . it also contains a battery or accumulator 9 , which is powered from the charging equipment 15 . a novel and preferential design of the charging equipment 15 allows it to connect to the dc power supply via a dc connection 16 on the one side and be charged with ac voltage of between 80 and 270 v via an ac connection 39 on the other side , without having to be set manually . appropriate electronics are integrated in the charging equipment 15 . the connection between battery 9 and charging equipment 15 is designed so that the battery receives a permanent charge all the time the charging equipment 15 is connected to the mains . if the external power supply fails , the system automatically switches to the internal battery . in this sample design the block is divided into two — a lower section 12 b and an upper section 12 a . this is for assembly reasons and not essential . in the same way single piece or multi - piece blocks could be used for the invention . as can be seen in fig3 the lower section 12 b includes an oxygen block 33 with oxygen sensor 26 and expiration valve 2 d . in service the lower section 12 b is screwed tightly to the upper section 12 a to form the gas supply container 8 on the inside . in this invention the shell walls of the upper and lower sections 12 a and 12 b contain rigid pipes which represent the gas routes according to the block diagram in fig4 . the gas supply pipes are at least 100 mm in length . a special pipe is represented by a broken line : gas deliverer 14 is a pipe which is connected to the compressor output 13 . compressed air leaving the compressor is warm when it is expelled . as this warm air has to travel a relatively long way through gas deliverer 14 , this heat is released to block 12 . this reduces the risk of condensation building up in the gas supply container 8 . an alternative to the gas deliverer 14 shown here would be to use a longer , perhaps spirally rolled pipe inside the gas supply container 8 to access the compressor output 13 . included in the upper section 12 a are the compressed air and ventilator gas connections 1 a and 1 b , as well as therapy gas connection 27 . it also carries a tank pressure control valve 32 , the inspiration valve 2 c , the therapy gas valve 25 , and a safety block 35 , which has a patient pressure relief valve and a patient suction relief valve . the last valves mentioned are equally located on valve block 38 , which is designed as an integrated block and which incorporates the emergency valves and the devices for flow and pressure measurement in particular . the upper section 12 a additionally carries a nebulizer valve 34 and the front connections 30 . fig2 primarily shows the design in fig3 when it is installed in housing 7 , which is illustrated in two sections ( 7 a , 7 b ) in this sample design . the compressor 6 with its flexible pipe 10 is symbolically illustrated . in the lower section of the housing 7 b you can see a compartment with integrated battery 9 , the electronics 3 and the location of the dc connection 16 , an ac connection 39 , as well as an interface 21 for a wide variety of connections , such as rs232 , nurse call , etc . the grips on the side 40 are used for transportation , as is the stand 31 , which is shown in fig1 . in fig1 you can also see a display 17 and a keyboard 18 with push buttons 20 and a preferentially used push - and - turn knob 19 . this makes it particularly easy to make the menu - controlled device settings , as mentioned in the introduction to the description .