Heat flux sensor

A heat flux sensor is a transducer that generates an electrical signal proportional to the total heat rate applied to the surface of the sensor. The measured heat rate is divided by the surface area of the sensor to determine the heat flux.
The heat flux can have different origins; in principle convective, radiative as well as conductive heat can be measured. Heat flux sensors are known under different names, such as heat flux transducers, heat flux gauges, heat flux plates. Some instruments are actually single-purpose heat flux sensors, like pyranometers for solar radiation measurement. Other heat flux sensors include Gardon gauges (also known as a circular-foil gauge), thin-film thermopiles, and Schmidt-Boelter gauges. In SI units, the heat rate is measured in Watts, and the heat flux is computed in Watts per meter squared.
Usage.
Heat flux sensors are used for a variety of applications. Common applications are studies of building envelope thermal resistance, studies of the effect of fire and flames or laser power measurements. More exotic applications include estimation of fouling on boiler surfaces, temperature measurement of moving foil material, etc.
The total heat flux is composed of a conductive, convective and radiative part. Depending on the application, one might want to measure all three of these quantities or single one out.
An example of measurement of conductive heat flux is a heat flux plate incorporated into a wall.
An example of measurement of radiative heat flux density is a pyranometer for measurement of solar radiation.
An example of a sensor sensitive to radiative as well as convective heat flux is a Gardon or Schmidt–Boelter gauge, used for studies of fire and flames. The Gardon must measure convection perpendicular to the face of the sensor to be accurate due to the circular-foil construction, while the wire-wound geometry of the Schmidt-Boelter gauge can measure both perpendicular and parallel flows. In this case the sensor is mounted on a water-cooled body. Such sensors are used in fire resistance testing to put the fire to which samples are exposed to the right intensity level.
There are various examples of sensors that internally use heat flux sensors examples are laser power meters, pyranometers, etc.
We will discuss three large fields of application in what follows.
Applications in meteorology and agriculture.
Soil heat flux is a most important parameter in agro-meteorological studies, since it allows one to study the amount of energy stored in the soil as a function of time.
Typically, two or three sensors are buried in the ground around a meteorological station at a depth of around 4 cm below the surface. The problems that are encountered in soil are threefold:
The result of all this is the quality of the data in soil heat flux measurement is not under control; the measurement of soil heat flux is considered to be extremely difficult.
Applications in building physics.
In a world ever more concerned with saving energy, studying the thermal properties of buildings has become a growing field of interest. One of the starting points in these studies is the mounting of heat flux sensors on walls in existing buildings or structures built especially for this type of research. Heat flux sensors mounted to building walls or envelope component can monitor the amount of heat energy loss/gain through that component and/or can be used to measure the envelope thermal resistance, R-value, or thermal transmittance, U-value.
The measurement of heat flux in walls is comparable to that in soil in many respects. Two major differences however are the fact that the thermal properties of a wall generally do not change (provided its moisture content does not change) and that it is not always possible to insert the heat flux sensor in the wall, so that it has to be mounted on its inner or outer surface.
When the heat flux sensor has to be mounted on the surface of the wall, one has to take care that the added thermal resistance is not too large. Also, the spectral properties should be matching those of the wall as closely as possible. If the sensor is exposed to solar radiation, this is especially important. In this case one should consider painting the sensor in the same color as the wall. Also, in walls the use of self-calibrating heat flux sensors should be considered.
Applications in medical studies.
The measurement of the heat exchange of human beings is of importance for medical studies, and when designing clothing, immersion suits and sleeping bags.
A difficulty during this measurement is that the human skin is not particularly suitable for the mounting of heat flux sensors. Also, the sensor has to be thin: the skin essentially is a constant temperature heat sink, so added thermal resistance has to be avoided. Another problem is that test persons might be moving. The contact between the test person and the sensor can be lost. For this reason, whenever a high level of quality assurance of the measurement is required, it can be recommended to use a self-calibrating sensor.
Applications in industry.
Heat flux sensors are also used in industrial environments, where temperature and heat flux may be much higher. Examples of these environments are aluminium smelting, solar concentrators, coal fired boilers, blast furnaces, flare systems, fluidized beds, cokers...
Applications in aerospace and explosive research.
Special heat flux solutions are used in highly transient temperatures changes. These gauges called Thermocouple MCT, allow the measurement of highly transient surface temperatures. For example, they are typical for testing wind tunnel models in impulse facilities, the change of the cylinder wall temperature during one cycle of a combustion engine, all types of industrial applications, and research-oriented work where the registration of highly transient temperatures is of importance. The response time of the gauges has been proven to be in the range of a few microseconds.
The output of all gauges represents the time-dependent temperature of its measuring part which in this case may significantly deviate from the temperature of the gauge-surrounding heating or cooling environment. For example, in a piston engine a flush wall-mounted temperature gauge registers with its typical response time the variation of the cylinder wall temperature and not the variation of the average gas temperature within the cylinder. The measured time-dependent surface temperature of the gauge and its known thermal properties allow to recalculate the time-dependent heat flux from the heating environment onto the gauge which caused the temperature change of the gauge. This is accomplished by the theory of heat conduction into a semi-infinite body. The design of the gauges is such that during a typical time period of about 10 ms, the requirements of a body of semi-infinite thickness are fulfilled. The direction of the deduced heat flux is perpendicular to the measuring surface of the gauge.
Properties.
A heat flux sensor should measure the local heat flux density in one direction. The result is expressed in watts per square meter. The calculation is done according to:
formula_1
Where formula_2 is the sensor output and formula_3 is the calibration constant, specific for the sensor.
As shown before in the figure to the left, heat flux sensors generally have the shape of a flat plate and a sensitivity in the direction perpendicular to the sensor surface.
Usually, a number of thermocouples connected in series called thermopiles are used. General advantages of thermopiles are their stability, low ohmic value (which implies little pickup of electromagnetic disturbances), good signal-noise ratio and the fact that zero input gives zero output. Disadvantageous is the low sensitivity.
For better understanding of heat flux sensor behavior, it can be modeled as a simple electrical circuit consisting of a resistance, formula_4, and a capacitor, formula_5. In this way it can be seen that one can attribute a thermal resistance formula_6, a thermal capacity formula_7 and also a response time formula_8 to the sensor.
Usually, the thermal resistance and the thermal capacity of the entire heat flux sensor are equal to those of the filling material. Stretching the analogy with the electric circuit further, one arrives at the following expression for the response time:
formula_9
In which formula_10 is the sensor thickness, formula_11 the density, formula_12 the specific heat capacity and formula_13 the thermal conductivity. From this formula one can conclude that material properties of the filling material and dimensions are determining the response time.
As a rule of thumb, the response time is proportional to the thickness to the power of two.
Other parameters that are determining sensor properties are the electrical characteristics of the thermocouple. The temperature dependence of the thermocouple causes the temperature dependence and the non-linearity of the heat flux sensor. The non-linearity at a certain temperature is in fact the derivative of the temperature dependence at that temperature.
However, a well-designed sensor may have a lower temperature dependence and better linearity than expected. There are two ways of achieving this:
Another factor that determines heat flux sensor behavior, is the construction of the sensor. In particular some designs have a strongly nonuniform sensitivity. Others even exhibit a sensitivity to lateral fluxes. The sensor schematically given in the above figure would for example also be sensitive to heat flows from left to right. This type of behavior will not cause problems as long as fluxes are uniform and in one direction only.
To promote uniformity of sensitivity, a so-called sandwich construction as shown in the figure to the left can be used. The purpose of the plates, which have a high conductivity, is to promote the transport of heat across the whole sensitive surface.
It is difficult to quantify non-uniformity and sensitivity to lateral fluxes. Some sensors are equipped with an extra electrical lead, splitting the sensor into two parts. If during application, there is non-uniform behavior of the sensor or the flux, this will result in different outputs of the two parts.
Summarizing: The intrinsic specifications that can be attributed to heat flux sensors are thermal conductivity, total thermal resistance, heat capacity, response time, non-linearity, stability, temperature dependence of sensitivity, uniformity of sensitivity and sensitivity to lateral fluxes. For the latter two specifications, a good method for quantification is not known.
Calibration of thin heat flux transducers.
In order to do in-situ measurements, the user must be provided with the correct calibration constant formula_14. This constant is also called "sensitivity". The sensitivity is primarily determined by the sensor construction and operation temperatures, but also by the geometry and material properties of the object that is measured. Therefore, the sensor should be calibrated under conditions that are close to the conditions of the intended application. The calibration set-up should also be properly shielded to limit external influences.
Preparation.
To do a calibration measurement, one needs a voltmeter or datalogger with resolution of ±2μV or better. One should avoid air gaps between layers in the test stack. These can be filled with filling materials, like toothpaste, caulk or putty. If need be, thermally conductive gel can be used to improve contact between layers. A temperature sensor should be placed on or near the sensor and connected to a readout device.
Measuring.
The calibration is done by applying a controlled heat flux through the sensor. By varying the hot and cold sides of the stack, and measuring the voltages of the heat flux sensor and temperature sensor, the correct sensitivity can be determined with:
formula_15
where formula_16 is the sensor output and formula_17 is the known heat flux through the sensor.
If the sensor is mounted onto a surface and is exposed to convection and radiation during the expected applications, the same conditions should be taken into account during calibration.
Doing measurements at different temperatures allows for determining sensitivity as a function of the temperature.
In-situ calibration.
While heat flux sensors are typically supplied with a sensitivity by the manufacturer, there are times and situations that call for a re-calibration of the sensor. Especially in building walls or envelopes the heat flux sensors cannot be removed after the initial installation or may be very difficult to reach. In order to calibrate the sensor, some come with an integrated heater with specified characteristics. By applying a known voltage on and current through the heater, a controlled heat flux is provided which can be used to calculate the new sensitivity.
Error sources.
The interpretation of measurement results of heat flux sensors is often done assuming that the phenomenon that is studied, is quasi-static and taking place in a direction transversal to the sensor surface.
Dynamic effects and lateral fluxes are possible error sources.
Dynamic effects.
The assumption that conditions are quasi-static should be related to the response time of the detector.
The case that the heat flux sensor is used as a radiation detector (see figure to the left) will serve to illustrate the effect of changing fluxes. Assuming that the cold joints of the sensor are at a constant temperature, and an energy flows from formula_18, the sensor response is:
formula_19
This shows that one should expect a false reading during a period that equals several response times, formula_8. Generally, heat flux sensors are quite slow and will need several minutes to reach 95% response. This is the reason why one prefers to work with values that are integrated over a long period; during this period the sensor signal will go up and down. The assumption is that errors due to long response times will cancel. The upgoing signal will give an error, the downgoing signal will produce an equally large error with a different sign. This will be valid only if periods with stable heat flow prevail.
In order to avoid errors caused by long response times, one should use sensors with low value of formula_21, since this product determines the response time. In other words: sensors with low mass or small thickness.
The sensor response time equation above holds as long as the cold joints are at a constant temperature. An unexpected result shows when the temperature of the sensor changes.
Assuming that the sensor temperature starts changing at the cold joints, at a rate of formula_22, starting at formula_23, formula_8 is the sensor response time, the reaction to this is: