PurposeTo evaluate a new non-contact instrument (OA-2000) measuring the ocular biometry parameters of silicone oil (SO)-filled aphakic eyes, as compared with IOLMaster 700.MethodsForty SO-filled aphakic eyes of 40 patients were enrolled in this cross-sectional clinical trial. The axial length (AL), central corneal thickness (CCT), keratometry ((flattest keratometry) Kf and (steep keratometry, 90° apart from Kf) Ks), and axis of the Kf (Ax1) were measured with OA-2000 and IOLMaster 700. The coefficient of variation (CoV) was calculated to assess the repeatability. The correlation was evaluated by the Pearson coefficient. Bland-Altman analysis and pairedttest were used to analyze the agreements and differences of parameters measured by the two devices, respectively.ResultsThe mean AL obtained with the OA-2000 was 23.57 ± 0.93 mm (range: 21.50 to 25.68 mm), and that obtained with the IOLMaster 700 was 23.69 ± 0.94 mm (range: 21.85 to 25.86 mm), resulting in a mean offset of 0.124 ± 0.125 mm (p< 0.001). The mean offset of CCT measured by OA-2000 and IOLMaster 700 was 14.6 ± 7.5 μm (p< 0.001). However, the Kf, Ks and Ax1 values from the two devices were comparable (p> 0.05). All the measured parameters of the two devices showed strong linear correlations (all r ≥ 0.966). The Bland-Altman analysis showed a narrow 95% limits of agreement (LoA) of Kf, Ks and AL, but 95%LoA of CCT and Ax1 was wide, which were − 29.3 ~ 0.1 μm and-25.9 ~ 30.7°respectively. The CoVs of the biometric parameters obtained with OA-2000 were lower than 1%.ConclusionIn SO-filled aphakic eyes, the ocular parameters (including AL, Kf, Ks, Ax1, and CCT) measured by the OA-2000 and IOLMaster 700 had a good correlation. Two devices had an excellent agreement on ocular biometric measurements of Kf, Ks and AL. The OA-2000 provided excellent repeatability of ocular parameters in SO-filled aphakic eyes.