Background:There is insufficient information regarding the optimal guidewire for managing malignant hilar biliary obstruction (MHBO). Therefore, a newly designed 0.025-inch guidewire was compared with the conventional 0.035-inch guidewire for selective cannulation of both intrahepatic ducts (IHDs) in patients with MHBO.Methods:Patients were randomly enrolled into the curved type newly designed 0.025-inch guidewire group (0.025 group) or the curved type conventional 0.035-inch guidewire group (0.035 group). The primary outcome was the selective cannulation rate of IHD. If the assigned guidewire failed to pass the stricture within 5 min, the crossover guidewire was selected. If the crossover guidewire failed to cross the stricture within the next 5 min, it was judged as a failed selective cannulation of both IHDs.Results:A total of 90 patients were enrolled (0.025 group,n= 47; 0.035 group,n= 43). There was no significant difference in baseline characteristics between the groups regarding sex, age, BMI, obstruction level, and clinical presentation. Four patients (8.5%) in the 0.025 group the cannulation of the IHD failed and the conventional 0.035-inch guidewire was substituted in a second attempt; the 0.035-inch guidewire failed to cross the stricture in all four patients. In the 0.035 group, eleven patients (25.6%) failed to achieve selective cannulation of IHD, and the 0.025-inch guidewire was substituted; the newly designed 0.025-inch guidewire crossed the stricture in ten of these (10/11, 90.9%). The selective cannulation rate of IHD was significantly higher in the 0.025 group (95.1% vs. 85.5%,p= 0.043).Conclusions:The 0.025 group exhibited a higher success rate for selective cannulation of both IHDs in MHBO than did the 0.035 group.