PurposeWe present a methodological overview of a respiration chamber at the Experimental and Clinical Research Center in Berlin, Germany. Since 2010, we investigated 750 healthy subjects and patients with various diseases. We routinely measure resting energy expenditure (REE), dietary-induced thermogenesis, and activity energy expenditure.MethodsThe chamber is a pull calorimeter with a total volume of 11,000 L. The majority of measurements is done with a flow rate of 120 L/min, yielding a favorable time constant of 1.53 h. The gas analysis system consists of two paramagnetic O2sensors and two infrared CO2sensors, one for incoming and one for outgoing air samples. O2and CO2sensors are calibrated simultaneously before each measurement with a 6 min calibration routine. To verify the accuracy of the whole the calorimetric system, it is validated every 2 weeks by 2 h acetone burning tests.ResultsValidation factors (calculated/measured) of 20 representative 2 h acetone burning tests were 1.03 ± 0.03 for\documentclass[12pt]{minimal}
				\usepackage{amsmath}
				\usepackage{wasysym} 
				\usepackage{amsfonts} 
				\usepackage{amssymb} 
				\usepackage{amsbsy}
				\usepackage{mathrsfs}
				\usepackage{upgreek}
				\setlength{\oddsidemargin}{-69pt}
				\begin{document}$$\dot{V}{\text{O}}_{2}$$\end{document}V˙O2, 1.02 ± 0.02 for\documentclass[12pt]{minimal}
				\usepackage{amsmath}
				\usepackage{wasysym} 
				\usepackage{amsfonts} 
				\usepackage{amssymb} 
				\usepackage{amsbsy}
				\usepackage{mathrsfs}
				\usepackage{upgreek}
				\setlength{\oddsidemargin}{-69pt}
				\begin{document}$$\dot{V}{\text{CO}}_{{2}}$$\end{document}V˙CO2, 0.99 ± 0.02 for RER, and 1.03 ± 0.03 for EE. Four repeated 60 min REE measurements of a healthy woman showed variabilities of 231.9 ± 4.8 ml/min for\documentclass[12pt]{minimal}
				\usepackage{amsmath}
				\usepackage{wasysym} 
				\usepackage{amsfonts} 
				\usepackage{amssymb} 
				\usepackage{amsbsy}
				\usepackage{mathrsfs}
				\usepackage{upgreek}
				\setlength{\oddsidemargin}{-69pt}
				\begin{document}$$\dot{V}{\text{O}}_{2}$$\end{document}V˙O2(CV 2.1%), 166.0 ± 6.3 ml/min for\documentclass[12pt]{minimal}
				\usepackage{amsmath}
				\usepackage{wasysym} 
				\usepackage{amsfonts} 
				\usepackage{amssymb} 
				\usepackage{amsbsy}
				\usepackage{mathrsfs}
				\usepackage{upgreek}
				\setlength{\oddsidemargin}{-69pt}
				\begin{document}$$\dot{V}{\text{CO}}_{{2}}$$\end{document}V˙CO2(CV 3.8%), 0.73 ± 0.03 for RER (CV 4.6%), and 4.55 ± 0.07 kJ/min for EE (CV 1.6%).ConclusionsThe data presented show that our respiration chamber produces precise and valid EE measurements with an exceptionally fast responsiveness.Graphical Abstract