Portable Oxygen Systems




After the air goes through the oxygen concentrator, it comes out as 90 to 95 percent pure oxygen and 5 to 10 percent nitrogen. 


albury vital air - portable oxygen systems

What Makes Up a Portable Oxygen Concentrator?

There are many parts that make up a portable oxygen concentrator. A compressor and sieve bed filter are a couple of the main parts. The compressor compresses air that is filtered into the concentrator, then delivers the air in a continuous stream. The compressed air moves to the sieve bed filters. The sieve bed filter plays an important role, as it is the device that removes the nitrogen from the air. A material called Zeolite, which is a six-sided microscopic cube with holes on each side, is in the sieve bed and this is what removes the nitrogen from the air. Two sieve beds are located in the concentrator. After air is first compressed in the concentrator, it is forced into the first sieve bed. Oxygen is sent into the product tank. The first sieve bed then gets filled up with nitrogen. Next, the gas flow is switched and the compressed air is moved to the second sieve bed. The first sieve bed’s compressor is sent to the outside room and the air from the product tank goes back into the first sieve bed. The drop in pressure from the first sieve bed and the weakening of oxygen makes the Zeolite release nitrogen. The Oxygen and Nitrogen come back together and are released in the room as regular air. The air is then compressed and sent to the second sieve where Oxygen is moved through it to the Product Tank. The whole cycle starts over again with the first sieve after a few seconds. Other important parts are the cooling system that keeps the portable oxygen concentrator from overheating, and the nasal cannula that delivers the purified oxygen after the oxygen has been passed through all the sieve bed filters. The cannula helps improve oxygen absorption.

Is A Portable Oxygen Concentrator Good for Day and Night?

Portable oxygen concentrators have two settings for receiving oxygen: pulse dose and continuous flow. The pulse dose mode is usually used for daytime use, as it delivers air via the cannula when you inhale. Concentrators with pulse dose technology also are more compact in design and offer a longer battery life. The continuous flow mode delivers a constant air flow via the tubes. For people who need oxygen while they sleep, this mode is the best option.

Rebates and Government Assistance

Individuals in their own homes can claim a rebate to offset the electricity costs of running the machine. Further Information on NSW and Vic rebates can be found here, or via contacting Albury Vital Air