Pressure sensor TruStability™ SSC seriesfor anesthesia equipmentfor respiratory devicesfor sleep apnea therapy devices

pressure sensor
pressure sensor
Add to favorites
Compare this product
indicative price


Evaluation type
for anesthesia equipment, for respiratory devices, for sleep apnea therapy devices, for the medical industry


Pressure Ranges ±1.6 mbar to ±10 bar | ±160 Pa to ±1 MPa | ±0.5 inH20 to ±150 psiThe TruStability® Standard Accuracy Silicon Ceramic (SSC) Series is a piezoresistive silicon pressure sensor offering a ratiometric analog or digital output for reading pressure over the specified full scale pressure span and temperature range.The SSC Series is fully calibrated and temperature compensated for sensor offset, sensitivity, temperature effects, and nonlinearity using an on-board Application Specific Integrated Circuit (ASIC). Calibrated output values for pressure are updated at approximately 1 kHz for analog and 2 kHz for digital.The SSC Series is calibrated over the temperature range of -20 °C to 85 °C [-4 °F to 185 °F]. The sensor is characterized for operation from a single power supply of either 3.3 Vdc or 5.0 Vdc.These sensors measure absolute, gage, or differential pressures. The absolute versions have an internal vacuum reference and an output value proportional to absolute pressure. Gage versions are referenced to atmospheric pressure and provide an output proportional to pressure variations from atmosphere. Differential versions allow measurement of pressure between two pressure ports.

Other Honeywell products

Sensing and Internet of Things Products - Medical: Respiratory, Patient Monitoring, Laboratory Equipment
