Download Agilent Technologies 6834B User`s guide

4 - Programming Examples
Step 4
Determine the number of times the list is repeated before it completes. For example, to
repeat a list 10 times use:
Entering INFinity makes the list repeat indefinitely. At *RST, the count is set to 1.
Step 5
Determines how the list sequencing responds to triggers. For a closely controlled
sequence of output levels, you can use a dwell-paced list. To cause the list to be paced by
dwell time use:
As each dwell time elapses, the next point is immediately output. This is the *RST setting.
If you need the output to closely follow asynchronous events, then a trigger-paced list is
more appropriate. In a trigger-paced list, the list advances one point for each trigger
received. To enable trigger-paced lists use:
The dwell time associated with each point determines the minimum time that the output
remains at that point. If a trigger is received before the previous dwell time completes, the
trigger is ignored. Therefore, to ensure that no triggers are lost, program the dwell time to
Step 6
Use the transient trigger system to trigger the list. This is described in detail under
"Triggering Output Changes".
Triggering Output Changes
The ac source has two independent trigger systems. One is used for generating output changes, and the
other is used for triggering measurements. This section describes the output trigger system. The
measurement trigger system is described under "Triggering Measurements".
The basic components of both systems are the same, but the transient trigger system has additional delay
and phase synchronization features that the measurement trigger system does not have. The following
transient trigger sources can be selected:
generates a trigger when the trigger system is initiated.
selects GPIB bus triggers.
selects the external Trigger In BNC connector.
SCPI Triggering Nomenclature
In SCPI terms, trigger systems are called sequences. When more than one trigger system exists, they are
differentiated by naming them SEQuence1, SEQuence2, ... etc. In the ac source, SEQuence1 is the
transient trigger system, SEQuence2 is the phase synchronization trigger system, and SEQuence3 is the
measurement trigger system.
The ac source uses aliases with more descriptive names for these sequences. These aliases can be used
instead of the sequence forms.
Sequence Form