It has already been mentioned that in HR-CS AAS lamp flicker noise is eliminated using correction pixels. In fact, any increase or decrease in radiation intensity that is observed to the same extent at all pixels chosen for correction is eliminated by the correction algorithm. This obviously also includes a reduction of the measured intensity due to radiation scattering or molecular absorption, which is corrected in the same way. As measurement of total and background absorption, and correction for the latter, are strictly simultaneous (in contrast to LS AAS), even the fastest changes of background absorption, as they may be observed in ET AAS, do not cause any problem. In addition, as the same algorithm is used for background correction and elimination of lamp noise, the background corrected signals show a much better signal-to-noise ratio compared to the uncorrected signals, which is also in contrast to LS AAS.