Instrumentation of supercritical fluid chromatography SFC has a similar setup to an HPLC instrument. The stationary phases are similar, and are packed inside similar column types. However, there are special features in these systems, because of the need to keep the mobile phase at supercritical fluidic state over the entire system. Temperature is critical to keep the fluids in a supercritical state, so there should be a heat control tool in the system, similar to that of GC. Also, there should be a precise pressure control mechanism, a restrictor to keep the pressure above a certain point, because pressure is another essential parameter to keep the mobile phase in a supercritical fluid state, so it is kept at the required minimal level. A microprocessor mechanism is placed in the instrument for SFC. This unit collects data for pressure, oven temperature, and detector performance to control the related pieces of the instrument.
CO utilized in carbon dioxide  dedicated pumps, which require that the incoming CO and pump heads be kept cold, in order to maintain the carbon dioxide at a temperature and pressure fit for supercritical fluidic state, where it can be effectively metered at a specified flow rate range.  The CO subsequently becomes supercritical fluid throughout the injector and the column oven, when the temperature and pressure it is subjected to, are raised above the critical point of the liquid, thus the supercritical state is achieved. 
Supercritical fluids combine useful properties of gas and liquid phases, as it can behave like both a gas and a liquid in various aspects. A supercritical fluid provides a gas-like characteristic when it fills a container and it takes the shape of the container. The motion and kinetics of the molecules are quite similar to gas molecules. On the other hand, a supercritical fluid behaves like a liquid because its density property is near liquid; thus, a supercritical fluid shows a similarity to the dissolving effect of a liquid. The result is that one can load masses, similar to those used in HPLC, on column per injection, and still maintain a high chromatographic efficiency similar to those attained in GC. Typically, gradient elution is employed in analytical SFC using a polar co-solvent such as methanol, possibly with a weak acid or base at low concentrations ~1%. The apparent plate count per analysis can be observed to exceed 500K plates per meter routinely with 5 um stationary phases. The operator uses software to set mobile phase flow rate, co-solvent composition, system back pressure and column oven temperature, which must exceed 40 °C for supercritical conditions needed to be achieved with CO. In addition, SFC provides an additional control parameter – pressure – by using an automated static and dynamic back pressure regulator. From an operational standpoint, SFC is as simple and robust as HPLC, but fraction collection is more convenient because the primary mobile phase evaporates leaving only the analyte and a small volume of polar co-solvent. If the outlet CO is captured, it can be re-compressed and recycled, allowing for >90% reuse of CO.
Similar to HPLC, SFC uses a variety of detection methods including UV/VIS, mass spectrometry, FID (unlike HPLC) and evaporative light scattering.