Measurement of VOCs from the indoor air is done with sorption tubes e. g. Tenax (for VOCs and SVOCs) or DNPH-cartridges (for carbonyl-compounds) or air detector. The VOCs adsorb on these materials and are afterwards desorbed either thermally (Tenax) or by elution (DNPH) and then analyzed by GC-MS/FID or HPLC. Reference gas mixtures are required for quality control of these VOC-measurements. Furthermore, VOC emitting products used indoors, e.g. building products and furniture, are investigated in emission test chambers under controlled climatic conditions. For quality control of these measurements round robin tests are carried out, therefore reproducibly emitting reference materials are ideally required. Other methods have used proprietary Silcosteel-coated canisters with constant flow inlets to collect samples over several days. These methods are not limited by the adsorbing properties of materials like Tenax.