Most sieve analyses are carried out dry. But there are some applications which can only be carried out by wet sieving. This is the case when the sample which has to be analysed is e.g. a suspension which must not be dried; or when the sample is a very fine powder which tends to agglomerate (mostly < 45 µm) – in a dry sieving process this tendency would lead to a clogging of the sieve meshes and this would make a further sieving process impossible. A wet sieving process is set up like a dry process: the sieve stack is clamped onto the sieve shaker and the sample is placed on the top sieve. Above the top sieve a water-spray nozzle is placed which supports the sieving process additionally to the sieving motion. The rinsing is carried out until the liquid which is discharged through the receiver is clear. Sample residues on the sieves have to be dried and weighed. When it comes to wet sieving it is very important not to change the sample in its volume (no swelling, dissolving or reaction with the liquid).