SPR can be used to study the real-time kinetics of molecular interactions. Determining the affinity between two ligands involves establishing the equilibrium dissociation constant, representing the equilibrium value for the product quotient. This constant can be determined using dynamic SPR parameters, calculated as the dissociation rate divided by the association rate.
In this process, a ligand is immobilized on the dextran surface of the SPR crystal. Through a microflow system, a solution with the analyte is injected over the ligand-covered surface. The binding of the analyte to the ligand causes an increase in the SPR signal (expressed in response units, RU). Following the association time, a solution without the analyte (typically a buffer) is introduced into the microfluidics to initiate the dissociation of the bound complex between the ligand and analyte. As the analyte dissociates from the ligand, the SPR signal decreases. From these association (on rate, ) and dissociation rates (off rate, ), the equilibrium dissociation constant (binding constant, ) can be calculated.
The detected SPR signal is a consequence of the electromagnetic coupling of the incident light with the surface plasmon of the gold layer. This interaction is particularly sensitive to the characteristics of the layer at the gold–solution interface, which is usually just a few nanometers thick. When substances bind to the surface, it alters the way light is reflected, causing a change in the reflection angle, which can be measured as a signal in SPR experiments. One common application is measuring the kinetics of antibody-antigen interactions.