A "strong" ion exchanger will not lose the charge on its matrix once the column is equilibrated and so a wide range of pH buffers can be used. "Weak" ion exchangers have a range of pH values in which they will maintain their charge. If the pH of the buffer used for a weak ion exchange column goes out of the capacity range of the matrix, the column will lose its charge distribution and the molecule of interest may be lost. Despite the smaller pH range of weak ion exchangers, they are often used over strong ion exchangers due to their having greater specificity. In some experiments, the retention times of weak ion exchangers are just long enough to obtain desired data at a high specificity.
Resins (often termed beads) of ion exchange columns may include functional groups such as weak/strong acids and weak/strong bases. There are also special columns that have resins with amphoteric functional groups that can exchange both cations and anions. Some examples of functional groups of strong ion exchange resins are quaternary ammonium cation (Q), which is an anion exchanger, and sulfonic acid (S, -SOOH), which is a cation exchanger. These types of exchangers can maintain their charge density over a pH range of 0–14. Examples of functional groups of Weak ion exchange resins include diethylaminoethyl (DEAE, -CHN(CHH)), which is an anion exchanger, and carboxymethyl (CM,  -CH-COOH), which is a cation exchanger. These two types of exchangers can maintain the charge density of their columns over a pH range of 5–9.
In ion chromatography, the interaction of the solute ions and the stationary phase based on their charges determines which ions will bind and to what degree. When the stationary phase features positive groups which attracts anions, it is called an anion exchanger; when there are negative groups on the stationary phase, cations are attracted and it is a cation exchanger. The attraction between ions and stationary phase also depends on the resin, organic particles used as ion exchangers.
Each resin features relative selectivity which varies based on the solute ions present who will compete to bind to the resin group on the stationary phase. The selectivity coefficient, the equivalent to the equilibrium constant,  is determined via a ratio of the concentrations between the resin and each ion, however, the general trend is that ion exchangers prefer binding to the ion with a higher charge, smaller hydrated radius, and higher polarizability, or the ability for the electron cloud of an ion to be disrupted by other charges. Despite this selectivity, excess amounts of an ion with a lower selectivity introduced to the column would cause the lesser ion to bind more to the stationary phase as the selectivity coefficient allows fluctuations in the binding reaction that takes place during ion exchange chromatography.
Following table shows the commonly used ion exchangers