The principle of thermal conductivity of gases can also be used to measure the concentration of a gas in a binary mixture of gases.
Working: if the same gas is present around all the Wheatstone bridge filaments, then the same temperature is maintained in all the filaments and hence same resistances are also maintained; resulting in a balanced Wheatstone bridge. However, If the dissimilar gas sample (or gas mixture) is passed over one set of two filaments and the reference gas on the other set of two filaments, then the Wheatstone bridge becomes unbalanced. And the resulting net voltage output of the circuit will be correlated with the database to identify the constituents of the sample gas.
Using this technique many unknown gas samples can be identified by comparing their thermal conductivity with other reference gas of known thermal conductivity. The most commonly used reference gas is nitrogen; as the thermal conductivity of most common gases (except hydrogen and helium) are similar to that of nitrogen.