Gear meters differ from oval gear meters in that the measurement chambers are made up of the gaps between the teeth of the gears. These openings divide up the fluid stream and as the gears rotate away from the inlet port, the meter's inner wall closes off the chamber to hold the fixed amount of fluid. The outlet port is located in the area where the gears are coming back together.      The fluid is forced out of the meter as the gear teeth mesh and reduce the available pockets to nearly zero volume.