Instead of light, this method employs ultrasound for collecting information on the particles that are dispersed in fluid. Dispersed particles absorb and scatter ultrasound similarly to light. This has been known since Lord Rayleigh developed the first theory of ultrasound scattering and published a book "The Theory of Sound" in 1878. There have been hundreds of papers studying ultrasound propagation through fluid particulates in the 20th century. It turns out that instead of measuring scattered energy versus angle, as with light, in the case of ultrasound, measuring the transmitted energy versus frequency is a better choice. The resulting ultrasound attenuation frequency spectra are the raw data for calculating particle size distribution. It can be measured for any fluid system with no dilution or other sample preparation. This is a big advantage of this method. Calculation of particle size distribution is based on theoretical models that are well verified for up to 50% by volume of dispersed particles on micron and nanometer scales. However, as concentration increases and the particle sizes approach the nanoscale, conventional modelling gives way to the necessity to include shear-wave re-conversion effects in order for the models to accurately reflect the real attenuation spectra.