Here a throwing motion acts on the sample. The vertical throwing motion is overlaid with a slight circular motion which results in distribution of the sample amount over the whole sieving surface. The particles are accelerated in the vertical direction (are thrown upwards). In the air they carry out free rotations and interact with the openings in the mesh of the sieve when they fall back. If the particles are smaller than the openings, they pass through the sieve. If they are larger, they are thrown. The rotating motion while suspended increases the probability that the particles present a different orientation to the mesh when they fall back again, and thus might eventually pass through the mesh.
Modern sieve shakers work with an electro-magnetic drive which moves a spring-mass system and transfers the resulting oscillation to the sieve stack. Amplitude and sieving time are set digitally and are continuously observed by an integrated control-unit. Therefore, sieving results are reproducible and precise (an important precondition for a significant analysis). Adjustment of parameters like amplitude and sieving time serves to optimize the sieving for different types of material. This method is the most common in the laboratory sector.