Faraday force magnetometry uses the fact that a spatial magnetic field gradient produces force that acts on a magnetized object, F = (M⋅∇)B. In Faraday force magnetometry the force on the sample can be measured by a scale (hanging the sample from a sensitive balance), or by detecting the displacement against a spring. Commonly a capacitive load cell or cantilever is used because of its sensitivity, size, and lack of mechanical parts. Faraday force magnetometry is approximately one order of magnitude less sensitive than a SQUID. The biggest drawback to Faraday force magnetometry is that it requires some means of not only producing a magnetic field, but also producing a magnetic field gradient. While this can be accomplished by using a set of special pole faces, a much better result can be achieved by using set of gradient coils. A major advantage to Faraday force magnetometry is that it is small and reasonably tolerant to noise, and thus can be implemented in a wide range of environments, including a dilution refrigerator. Faraday force magnetometry can also be complicated by the presence of torque (see previous technique). This can be circumvented by varying the gradient field independently of the applied DC field so the torque and the Faraday force contribution can be separated, and/or by designing a Faraday force magnetometer that prevents the sample from being rotated.