The turbine meter is available in various manufacturers configurations of a common theme;  turbine blades and rotor configured devices. These devices are designed such that when a gas stream passes through them they will spin proportionally to the amount of gas passing over the blades in a repeatable fashion. Accuracy is then ensured by completion of a calibration, indicating the relationship between rotational speed and volume, at various Reynolds Numbers. The fundamental difference between the orifice meter and the turbine meter is the flow equation derivation. The orifice meter flow calculation is based on fluid flow fundamentals (a 1st Law of Thermodynamics derivation utilizing the pipe diameter and vena contracta diameters for the continuity equation). Deviations from theoretical expectation can be assumed under the Coefficient of Discharge. Thus, one can manufacture an orifice meter of known uncertainty with only the measurement standard in hand and access to a machine shop. The need for flow conditioning, and hence, a fully developed velocity flow profile is driven from the original determination of Cd which utilized fully developed or reference profiles' as explained above.
Conversely, the turbine meter operation is not rooted deeply in fundamentals of thermodynamics. This is not to say that the turbine meter is in any way an inferior device. There are sound engineering principles providing theoretical background. It is essentially an extremely repeatable device that is then assured accuracy via calibration. The calibration provides the accuracy. It is carried out in good flow conditions (flow conditions free of swirl and a uniform velocity flow profile) this is carried out for every meter manufactured. Deviations from the as-calibrated conditions would be considered installation effects, and the sensitivity of the turbine meter to these installation effects is of interest. The need for flow conditioning is driven from the sensitivity of the meter to deviations from as calibrated conditions of swirl and velocity profile.
Generally, recent research indicates that turbine meters are sensitive to swirl but not to the shape of the velocity profile. A uniform velocity profile is recommended, but no strict requirements for fully developed flow profiles are indicated. Also, no significant errors are evident when installing single or dual rotor turbine meters downstream of two elbows out-of-plane without flow conditioning devices.