Precession electron diffraction is typically conducted using accelerating voltages between 100-400 kV. Patterns can be formed under parallel or convergent beam conditions. Most modern TEMs can achieve a tilt angle, φ, ranging from 0-3°. Precession frequencies can be varied from Hz to kHz, but in standard cases 60 Hz has been used. In choosing a precession rate, it is important to ensure that many revolutions of the beam occur over the relevant exposure time used to record the diffraction pattern. This ensures adequate averaging over the excitation error of each reflection. Beam sensitive samples may dictate shorter exposure times and thus, motivate the use of higher precession frequencies.
One of the most significant parameters affecting the diffraction pattern obtained is the precession angle, φ. In general, larger precession angles result in more kinematical diffraction patterns, but both the capabilities of the beam tilt coils in the microscope and the requirements on the probe size limit how large this angle can become in practice. Because PED takes the beam off of the optic axis by design, it accentuates the effect of the spherical aberrations within the probe forming lens. For a given spherical aberration, C, the probe diameter, d, varies with convergence angle, α, and precession angle, φ, as
Thus, if the specimen of interest is quite small, the maximum precession angle will be restrained. This is most significant for conditions of convergent beam illumination. 50 nm is a general lower limit on probe size for standard TEMs operating at high precession angles (>30 mrad), but can be surpassed in C corrected instruments.  In principle the minimum precessed probe can reach approximately the full-width-half-max (FWHM) of the converged un-precessed probe in any instrument, however in practice the effective precessed probe is typically ~10-50x larger due to uncontrolled aberrations present at high angles of tilt.  For example, a 2 nm precessed probe with >40 mrad precession angle was demonstrated in an aberration-corrected Nion UltraSTEM with native sub-Å probe (aberrations corrected to ~35 mrad half-angle).
If the precession angle is made too large, further complications due to the overlap of the ZOLZ and HOLZ reflections in the projected pattern can occur. This complicates the indexing of the diffraction pattern and can corrupt the measured intensities of reflections near the overlap region, thereby reducing the effectiveness of the collected pattern for direct methods calculations.