The evaluation of the performance cannot be considered complete without considering the limit of detection, the range of linear response and the signal reproducibility. EI is known as a low-efficiency ionization technique. Because less than 1/10 000 of the gas-phase sample molecules are ionized, impressive detection limits cannot be expected. However, the efficient interfacing mechanism of this interface allows picogram-level detection limit in selected ion monitoring (SIM) for most substances. On the other hand, soft ionization techniques such as ESI are, in some cases, far more efficient but generate fewer fragment ions. The cost of this attitude is paid in terms of structural information so that a second analyzer to generate MS/MS spectra is an obligation. A typical EI spectrum, in general, has extensive structural information, and a cheaper, single-stage mass spectrometer might be sufficient for analyte characterization or identification. As a rule of the thumb, nanogram-level sensitivity is obtained in full-scan mode for most substances. Linearity and reproducibility are two point of strength of the interface. Up to four orders of magnitude linearity with RSD lower than 10% are common values in many applications.