SELDI is often criticized for its reproducibility due to differences in the mass spectra obtained when using different batches of chip surfaces. While the method has been successful with analyzing low molecular weight proteins, consistent results have not been obtained when analyzing high molecular weight proteins. There also exists a potential for sample bias, as nonspecific absorption matrices favor the binding of analytes with higher abundances in the sample at the expense of less abundant analytes. While SELDI-TOF-MS has detection limits in the femtomolar range, the baseline signal in the spectra varies and noise due to the matrix is maximal below 2000 Da, with Ciphergen Biosystems suggesting to ignore spectral peaks below 2000 Da.