Potentiometric sensors measure a potential or charge accumulation of an electrochemical cell. The transducer typically comprises an ion selective electrode (ISE) and a reference electrode.  The ISE features a membrane that selectively interacts with the charged ion of interest, causing the accumulation of a charge potential compared to the reference electrode. The reference electrode provides a constant half-cell potential that is unaffected by analyte concentration. A high impedance voltmeter is used to measure the electromotive force or potential between the two electrodes when zero or no significant current flows between them. The potentiometric response is governed by the Nernst equation in that the potential is proportional to the logarithm of the concentration of the analyte.