Samples spotted on a SELDI surface are typically analyzed using time-of-flight mass spectrometry. An irradiating laser ionizes peptides from crystals of the sample/matrix mixture. The matrix absorbs the energy of the laser pulse, preventing destruction of the molecule, and transfers charge to the sample molecules, forming ions. The ions are then briefly accelerated through an electric potential and travel down a field-free flight tube where they are separated by their velocity differences. The mass-to-charge ratio of each ion can be determined from the length of the tube, the kinetic energy given to ions by the electric field, and the velocity of the ions in the tube. The velocity of the ions is inversely proportional to the square root of the mass-to-charge ratio of the ion; ions with low mass-to-charge ratios are detected earlier than ions with high mass-to-charge ratios.