The carrier gas is sent through the central channel and into the very hot plasma. The sample is then exposed to radio frequency which converts the gas into a plasma. The high temperature of the plasma is sufficient to cause a very large portion of the sample to form ions. This fraction of ionization can approach 100% for some elements (e.g. sodium), but this is dependent on the ionization potential. A fraction of the formed ions passes through a ~1 mm hole (sampler cone) and then a ~0.4 mm hole (skimmer cone). The purpose of which is to allow a vacuum that is required by the mass spectrometer.
The vacuum is created and maintained by a series of pumps. The first stage is usually based on a roughing pump, most commonly a standard rotary vane pump. This removes most of the gas and typically reaches a pressure of around 133 Pa. Later stages have their vacuum generated by more powerful vacuum systems, most often turbomolecular pumps. Older instruments may have used oil diffusion pumps for high vacuum regions.