ELSDs analyze solutes eluting out of the chromatographic column, both in LC and SFC. As the eluent exits the columns outlet into the detector inlet, it is mixed with an inert carrier gas (usually nitrogen) and forced through a nebulizer, which separates the liquid into fine aerosolized droplets. These droplets then pass into a heated drift tube, where the mobile phase solvent is evaporated off. As the mobile phase evaporates, the droplets become smaller and smaller until all that is left is minute particles of dried analyte. These particles are pushed through the drift tube by the carrier gas to the detection region. In this region, a beam of light crosses the column of analyte and the scattering of light is measured by a photodiode or photomultiplier tube. The detectors output is non-linear across more than one order of magnitude and proper calibration is required for quantitative analysis.